2021/01/19 13:44:40.3 AcquisitionEngine: DLL_PROCESS_ATTACH - V11.1.0.1
2021/01/19 13:44:40.3 DataSecurity: DLL_PROCESS_ATTACH - V1.0.0.1
2021/01/19 13:44:40.3 MultiWellPlate: DLL_PROCESS_ATTACH - V1.0.0.6
2021/01/19 13:44:40.4 
2021/01/19 13:44:40.4 Octet Data Acquisition 11.1.2.24 (debug build)
2021/01/19 13:44:40.4 
2021/01/19 13:44:40.6 CreateToolbar()
2021/01/19 13:44:40.7 Instrument started...
2021/01/19 13:44:40.9 Opening instrument: Octet RED96e
2021/01/19 13:44:43.2 ReadInfo()
2021/01/19 13:44:43.7 Serial = FB-90326
2021/01/19 13:44:43.7 GALIL: XY steps/mm = 251.969, Z steps/mm = 1259.84
2021/01/19 13:44:44.4 GetSamplePos1_1 = (3.7179,-106.452,-22.2073)
2021/01/19 13:44:44.4 GetSamplePos1_9 = (3.7179,-36.4881,-39.3672)
2021/01/19 13:44:44.4 m_dPickRowSpacingMM = 9
2021/01/19 13:44:44.4 m_dPickColSpacingMM = -9
2021/01/19 13:44:44.4 m_ySample1RowSpacingMM = 8.74555
2021/01/19 13:44:44.4 m_zSample1RowSpacingMM = -2.14499
2021/01/19 13:44:44.4 m_ySample2RowSpacingMM = 9
2021/01/19 13:44:44.4 m_zSample2RowSpacingMM = 0
2021/01/19 13:44:44.4 
2021/01/19 13:44:44.4 gmcController polling at 6.7 Hz
2021/01/19 13:44:44.4 
2021/01/19 13:44:44.4 Galil Controller - Program Version: 4301
2021/01/19 13:44:44.5 
2021/01/19 13:44:44.5 Opened Octet RED96e with firmware 1.0.0.0, API 1.5
2021/01/19 13:44:44.5 
2021/01/19 13:44:44.5 Set Lamp[65535]=on
2021/01/19 13:44:44.5 Set Lamp[65535]=on
2021/01/19 13:44:44.6 SetHeaterTemperature 30.0
2021/01/19 13:44:44.6 Old State:24- New State:ENGN_State_Launched
2021/01/19 13:44:44.6 Temperature monitor ENABLE
2021/01/19 13:44:44.6 Set Heater[65535] @ 30.000000[C]
2021/01/19 13:44:44.7 Set Lamp[65535]=on
2021/01/19 13:44:44.9 
2021/01/19 13:44:44.9 
2021/01/19 13:44:44.9 ** Initializing Instrument **
2021/01/19 13:44:45.0 Old State:25-ENGN_State_Launched New State:ENGN_State_WaitInitialize
2021/01/19 13:44:45.0 StopDevice[0006]
2021/01/19 13:44:45.2 Opening instrument: Octet RED96e
2021/01/19 13:44:47.4 ReadInfo()
2021/01/19 13:44:47.9 Serial = FB-90326
2021/01/19 13:44:47.9 GALIL: XY steps/mm = 251.969, Z steps/mm = 1259.84
2021/01/19 13:44:48.7 GetSamplePos1_1 = (3.7179,-106.452,-22.2073)
2021/01/19 13:44:48.7 GetSamplePos1_9 = (3.7179,-36.4881,-39.3672)
2021/01/19 13:44:48.7 m_dPickRowSpacingMM = 9
2021/01/19 13:44:48.7 m_dPickColSpacingMM = -9
2021/01/19 13:44:48.7 m_ySample1RowSpacingMM = 8.74555
2021/01/19 13:44:48.7 m_zSample1RowSpacingMM = -2.14499
2021/01/19 13:44:48.7 m_ySample2RowSpacingMM = 9
2021/01/19 13:44:48.7 m_zSample2RowSpacingMM = 0
2021/01/19 13:44:48.7 
2021/01/19 13:44:48.7 gmcController polling at 6.7 Hz
2021/01/19 13:44:48.7 
2021/01/19 13:44:48.7 Galil Controller - Program Version: 4301
2021/01/19 13:44:48.7 
2021/01/19 13:44:48.7 Opened Octet RED96e with firmware 1.0.0.0, API 1.5
2021/01/19 13:44:48.7 
2021/01/19 13:44:48.7 Set Lamp[65535]=on
2021/01/19 13:44:48.8 Set Lamp[65535]=on
2021/01/19 13:44:48.8 Set Heater[65535] @ 30.000000[C]
2021/01/19 13:44:48.9 Looking for 8 devices with timeout of 60000 ms
2021/01/19 13:44:48.9 Found 8 devices in 0 ms
2021/01/19 13:44:48.9 Couldn't open spectrometer factory for type 'SPEC_TYPE_OCEANOPTICS'
2021/01/19 13:44:48.9 EOS EOS_B0735 API = 0x0100
2021/01/19 13:44:48.9 EOS EOS_B0814 API = 0x0100
2021/01/19 13:44:48.9 EOS EOS_B0808 API = 0x0100
2021/01/19 13:44:48.9 EOS EOS_B0732 API = 0x0100
2021/01/19 13:44:48.9 EOS EOS_B0734 API = 0x0100
2021/01/19 13:44:48.9 EOS EOS_B0712 API = 0x0100
2021/01/19 13:44:48.9 EOS EOS_B0817 API = 0x0100
2021/01/19 13:44:48.9 EOS EOS_B0733 API = 0x0100
2021/01/19 13:44:48.9 EOS EOS_B0712 API = 0x0100
2021/01/19 13:44:48.9 EOS EOS_B0734 API = 0x0100
2021/01/19 13:44:48.9 EOS EOS_B0814 API = 0x0100
2021/01/19 13:44:48.9 EOS EOS_B0735 API = 0x0100
2021/01/19 13:44:48.9 EOS EOS_B0732 API = 0x0100
2021/01/19 13:44:48.9 EOS EOS_B0808 API = 0x0100
2021/01/19 13:44:48.9 EOS EOS_B0733 API = 0x0100
2021/01/19 13:44:48.9 EOS EOS_B0817 API = 0x0100
2021/01/19 13:44:48.9 
2021/01/19 13:44:48.9 Calibration coefficients for EOS_B0712
2021/01/19 13:44:48.9 C0 = 5.724263e+02  C1 = 1.047210e-01  C2 = -4.525783e-06  C3 = -1.311609e-10  
2021/01/19 13:44:48.9 
2021/01/19 13:44:48.9 Linearity correction coefficients:
2021/01/19 13:44:48.9 L0 = 9.770590e-01 L1 = 1.598670e-06 L2 = 4.803710e-12 L3 = -5.086290e-15 
2021/01/19 13:44:48.9 L4 = 2.741280e-19 L5 = -6.802670e-24 L6 = 8.120710e-29 L7 = -3.786440e-34 
2021/01/19 13:44:48.9 
2021/01/19 13:44:48.9 Linearity correction order: 7
2021/01/19 13:44:48.9 PCA Serial number: 06 000 2113
2021/01/19 13:44:48.9 Stray light const: 0.000000e+00
2021/01/19 13:44:48.9 Configuration info: TS4 0
2021/01/19 13:44:48.9 Scale factor: 1.023984
2021/01/19 13:44:48.9 
2021/01/19 13:44:48.9 Calibration coefficients for EOS_B0734
2021/01/19 13:44:48.9 C0 = 5.724263e+02  C1 = 1.047210e-01  C2 = -4.525783e-06  C3 = -1.311609e-10  
2021/01/19 13:44:48.9 
2021/01/19 13:44:48.9 Linearity correction coefficients:
2021/01/19 13:44:48.9 L0 = 9.770590e-01 L1 = 1.598670e-06 L2 = 4.803710e-12 L3 = -5.086290e-15 
2021/01/19 13:44:48.9 L4 = 2.741280e-19 L5 = -6.802670e-24 L6 = 8.120710e-29 L7 = -3.786440e-34 
2021/01/19 13:44:48.9 
2021/01/19 13:44:48.9 Linearity correction order: 7
2021/01/19 13:44:48.9 PCA Serial number: 06 000 2172
2021/01/19 13:44:48.9 Stray light const: 0.000000e+00
2021/01/19 13:44:48.9 Configuration info: TS4 0
2021/01/19 13:44:48.9 Scale factor: 1.023984
2021/01/19 13:44:48.9 
2021/01/19 13:44:48.9 Calibration coefficients for EOS_B0814
2021/01/19 13:44:48.9 C0 = 5.724263e+02  C1 = 1.047210e-01  C2 = -4.525783e-06  C3 = -1.311609e-10  
2021/01/19 13:44:48.9 
2021/01/19 13:44:48.9 Linearity correction coefficients:
2021/01/19 13:44:48.9 L0 = 9.770590e-01 L1 = 1.598670e-06 L2 = 4.803710e-12 L3 = -5.086290e-15 
2021/01/19 13:44:48.9 L4 = 2.741280e-19 L5 = -6.802670e-24 L6 = 8.120710e-29 L7 = -3.786440e-34 
2021/01/19 13:44:48.9 
2021/01/19 13:44:48.9 Linearity correction order: 7
2021/01/19 13:44:48.9 PCA Serial number: 06 000 2287
2021/01/19 13:44:48.9 Stray light const: 0.000000e+00
2021/01/19 13:44:48.9 Configuration info: TS4 0
2021/01/19 13:44:48.9 Scale factor: 1.023984
2021/01/19 13:44:48.9 
2021/01/19 13:44:48.9 Calibration coefficients for EOS_B0735
2021/01/19 13:44:48.9 C0 = 5.724263e+02  C1 = 1.047210e-01  C2 = -4.525783e-06  C3 = -1.311609e-10  
2021/01/19 13:44:48.9 
2021/01/19 13:44:48.9 Linearity correction coefficients:
2021/01/19 13:44:48.9 L0 = 9.770590e-01 L1 = 1.598670e-06 L2 = 4.803710e-12 L3 = -5.086290e-15 
2021/01/19 13:44:48.9 L4 = 2.741280e-19 L5 = -6.802670e-24 L6 = 8.120710e-29 L7 = -3.786440e-34 
2021/01/19 13:44:48.9 
2021/01/19 13:44:48.9 Linearity correction order: 7
2021/01/19 13:44:48.9 PCA Serial number: 06 000 2105
2021/01/19 13:44:48.9 Stray light const: 0.000000e+00
2021/01/19 13:44:48.9 Configuration info: TS4 0
2021/01/19 13:44:48.9 Scale factor: 1.023984
2021/01/19 13:44:48.9 
2021/01/19 13:44:48.9 Calibration coefficients for EOS_B0732
2021/01/19 13:44:48.9 C0 = 5.724263e+02  C1 = 1.047210e-01  C2 = -4.525783e-06  C3 = -1.311609e-10  
2021/01/19 13:44:48.9 
2021/01/19 13:44:48.9 Linearity correction coefficients:
2021/01/19 13:44:48.9 L0 = 9.770590e-01 L1 = 1.598670e-06 L2 = 4.803710e-12 L3 = -5.086290e-15 
2021/01/19 13:44:48.9 L4 = 2.741280e-19 L5 = -6.802670e-24 L6 = 8.120710e-29 L7 = -3.786440e-34 
2021/01/19 13:44:48.9 
2021/01/19 13:44:48.9 Linearity correction order: 7
2021/01/19 13:44:48.9 PCA Serial number: 06 000 2214
2021/01/19 13:44:48.9 Stray light const: 0.000000e+00
2021/01/19 13:44:48.9 Configuration info: TS4 0
2021/01/19 13:44:48.9 Scale factor: 1.023984
2021/01/19 13:44:48.9 
2021/01/19 13:44:48.9 Calibration coefficients for EOS_B0808
2021/01/19 13:44:48.9 C0 = 5.724263e+02  C1 = 1.047210e-01  C2 = -4.525783e-06  C3 = -1.311609e-10  
2021/01/19 13:44:48.9 
2021/01/19 13:44:48.9 Linearity correction coefficients:
2021/01/19 13:44:48.9 L0 = 9.770590e-01 L1 = 1.598670e-06 L2 = 4.803710e-12 L3 = -5.086290e-15 
2021/01/19 13:44:48.9 L4 = 2.741280e-19 L5 = -6.802670e-24 L6 = 8.120710e-29 L7 = -3.786440e-34 
2021/01/19 13:44:48.9 
2021/01/19 13:44:48.9 Linearity correction order: 7
2021/01/19 13:44:48.9 PCA Serial number: 06 000 2239
2021/01/19 13:44:48.9 Stray light const: 0.000000e+00
2021/01/19 13:44:48.9 Configuration info: TS4 0
2021/01/19 13:44:48.9 Scale factor: 1.023984
2021/01/19 13:44:48.9 
2021/01/19 13:44:48.9 Calibration coefficients for EOS_B0733
2021/01/19 13:44:48.9 C0 = 5.724263e+02  C1 = 1.047210e-01  C2 = -4.525783e-06  C3 = -1.311609e-10  
2021/01/19 13:44:48.9 
2021/01/19 13:44:48.9 Linearity correction coefficients:
2021/01/19 13:44:48.9 L0 = 9.770590e-01 L1 = 1.598670e-06 L2 = 4.803710e-12 L3 = -5.086290e-15 
2021/01/19 13:44:48.9 L4 = 2.741280e-19 L5 = -6.802670e-24 L6 = 8.120710e-29 L7 = -3.786440e-34 
2021/01/19 13:44:48.9 
2021/01/19 13:44:48.9 Linearity correction order: 7
2021/01/19 13:44:48.9 PCA Serial number: 06 000 2106
2021/01/19 13:44:48.9 Stray light const: 0.000000e+00
2021/01/19 13:44:48.9 Configuration info: TS4 0
2021/01/19 13:44:48.9 Scale factor: 1.023984
2021/01/19 13:44:48.9 
2021/01/19 13:44:48.9 Calibration coefficients for EOS_B0817
2021/01/19 13:44:48.9 C0 = 5.724263e+02  C1 = 1.047210e-01  C2 = -4.525783e-06  C3 = -1.311609e-10  
2021/01/19 13:44:48.9 
2021/01/19 13:44:48.9 Linearity correction coefficients:
2021/01/19 13:44:48.9 L0 = 9.770590e-01 L1 = 1.598670e-06 L2 = 4.803710e-12 L3 = -5.086290e-15 
2021/01/19 13:44:48.9 L4 = 2.741280e-19 L5 = -6.802670e-24 L6 = 8.120710e-29 L7 = -3.786440e-34 
2021/01/19 13:44:48.9 
2021/01/19 13:44:48.9 Linearity correction order: 7
2021/01/19 13:44:48.9 PCA Serial number: 06 000 2286
2021/01/19 13:44:48.9 Stray light const: 0.000000e+00
2021/01/19 13:44:48.9 Configuration info: TS4 0
2021/01/19 13:44:48.9 Scale factor: 1.023984
2021/01/19 13:44:48.9 Opened: EOS_B0712, EOS_B0734, EOS_B0814, EOS_B0735, EOS_B0732, EOS_B0808, EOS_B0733, EOS_B0817, 
2021/01/19 13:44:49.0 
2021/01/19 13:44:49.0 DSP settings (Cross Correlation)
2021/01/19 13:44:49.0  - Transmission
2021/01/19 13:44:49.0 
2021/01/19 13:44:49.0 Ch	IgnoreLow	IgnoreHigh
2021/01/19 13:44:49.0 1	150		200
2021/01/19 13:44:49.0 2	150		200
2021/01/19 13:44:49.0 3	150		200
2021/01/19 13:44:49.0 4	150		200
2021/01/19 13:44:49.0 5	150		200
2021/01/19 13:44:49.0 6	150		200
2021/01/19 13:44:49.0 7	150		200
2021/01/19 13:44:49.0 8	150		200
2021/01/19 13:44:49.4 EJECT: To Chute, position=-2,-1
2021/01/19 13:44:49.5 Old State:1-ENGN_State_WaitInitialize New State:ENGN_State_Initializing
2021/01/19 13:44:49.5 GMC_Interlock
2021/01/19 13:44:49.9 GMC_HomeXYZ
2021/01/19 13:44:49.9 HomeDevice[0006]
2021/01/19 13:44:55.8 GMC_SetPosZ
2021/01/19 13:44:56.1 GMC_ServoHere
2021/01/19 13:44:56.3 GMC_MasterInit
2021/01/19 13:44:56.3 Init(0x0)
2021/01/19 13:45:00.6 GMC_SetPosZ
2021/01/19 13:45:00.8 GMC_ServoHere
2021/01/19 13:45:01.0 GMC_SetPosXY
2021/01/19 13:45:03.9 GMC_SetPosZ
2021/01/19 13:45:08.7 GMC_Eject
2021/01/19 13:45:08.7 EjectDone => EjectEject
2021/01/19 13:45:08.7 SetEjectorPosn(0)
2021/01/19 13:45:11.9 GMC_HomeZ
2021/01/19 13:45:11.9 HomeDevice[0005]
2021/01/19 13:45:18.2 GMC_SetPosZ
2021/01/19 13:45:18.5 GMC_ServoHere
2021/01/19 13:45:18.7 GMC_Interlock
2021/01/19 13:45:19.1 GMC_HomeXYZ
2021/01/19 13:45:19.1 HomeDevice[0006]
2021/01/19 13:45:26.7 GMC_SetPosZ
2021/01/19 13:45:27.0 GMC_ServoHere
2021/01/19 13:45:27.2 GMC_SetPosZ
2021/01/19 13:45:27.4 GMC_ServoHere
2021/01/19 13:45:27.6 GMC_SetPosXY
2021/01/19 13:45:30.6 GMC_SetPosZ
2021/01/19 13:45:30.9 Old State:2-ENGN_State_Initializing New State:ENGN_State_InitializeComplete
2021/01/19 13:45:30.9 Integration Time: 1.00,1.00,1.00,1.00,1.00,1.00,1.00,1.00,
2021/01/19 13:45:30.9 Set Lamp[65535]=on
2021/01/19 13:45:31.0 Set Heater[65535] @ 30.000000[C]
2021/01/19 13:45:31.0 Old State:3-ENGN_State_InitializeComplete New State:ENGN_State_SystemReady
2021/01/19 13:45:31.1 Pickups: 320, Warn: 90, Error: 100
2021/01/19 13:51:56.1 SetHeaterTemperature 21.0
2021/01/19 13:51:56.1 Set Heater[65535] @ 21.000000[C]
2021/01/19 13:51:56.2 Temperature monitor ENABLE
2021/01/19 13:51:56.2 Temp: 2021/01/19 13:51:56, 30.0000, 23.6000
2021/01/19 13:51:56.2 bShakeWhileWaiting=1, nShakerRPM=1000
2021/01/19 13:51:56.2 DELAYED START: Waiting for 600 s
2021/01/19 13:51:56.3 GMC_SetPosZ
2021/01/19 13:51:56.3 Old State:4-ENGN_State_SystemReady New State:ENGN_State_ControllerBusy
2021/01/19 13:51:56.4 GMC_ServoHere
2021/01/19 13:51:56.6 GMC_SetPosXY
2021/01/19 13:51:56.8 GMC_SetPosZ
2021/01/19 13:51:57.2 Old State:29-ENGN_State_ControllerBusy New State:ENGN_State_ControllerDone
2021/01/19 13:51:57.3 StartShaker(1000)
2021/01/19 13:51:57.3 Old State:30-ENGN_State_ControllerDone New State:ENGN_State_SystemReady
2021/01/19 13:52:56.3 Temp: 2021/01/19 13:52:56, 20.8000, 23.7000
2021/01/19 13:53:56.5 Temp: 2021/01/19 13:53:56, 20.6000, 23.8000
2021/01/19 13:54:56.7 Temp: 2021/01/19 13:54:56, 20.8000, 23.9000
2021/01/19 13:55:56.9 Temp: 2021/01/19 13:55:56, 20.9000, 24.0000
2021/01/19 13:56:57.0 Temp: 2021/01/19 13:56:57, 21.0000, 24.0000
2021/01/19 13:57:57.2 Temp: 2021/01/19 13:57:57, 21.0000, 23.9000
2021/01/19 13:58:57.3 Temp: 2021/01/19 13:58:57, 21.0000, 23.8000
2021/01/19 13:59:57.5 Temp: 2021/01/19 13:59:57, 21.0000, 23.9000
2021/01/19 14:00:57.7 Temp: 2021/01/19 14:00:57, 21.0000, 23.8000
2021/01/19 14:01:57.8 Temp: 2021/01/19 14:01:57, 21.0000, 23.8000
2021/01/19 14:01:58.2 DELAYED START: Done.
2021/01/19 14:01:58.2 Spectrometer EOS_B0712: AVRG=20.
2021/01/19 14:01:58.2 Spectrometer EOS_B0734: AVRG=20.
2021/01/19 14:01:58.2 Spectrometer EOS_B0814: AVRG=20.
2021/01/19 14:01:58.2 Spectrometer EOS_B0735: AVRG=20.
2021/01/19 14:01:58.2 Spectrometer EOS_B0732: AVRG=20.
2021/01/19 14:01:58.2 Spectrometer EOS_B0808: AVRG=20.
2021/01/19 14:01:58.2 Spectrometer EOS_B0733: AVRG=20.
2021/01/19 14:01:58.2 Spectrometer EOS_B0817: AVRG=20.
2021/01/19 14:01:58.2 Starting new assay
2021/01/19 14:01:58.5 Plate 1 temperature = 21 C
2021/01/19 14:01:58.5 Old State:4-ENGN_State_SystemReady New State:ENGN_StateRunning_StartNewStep
2021/01/19 14:01:58.6 Spectrometer EOS_B0712: AVRG=20.
2021/01/19 14:01:58.6 Spectrometer EOS_B0734: AVRG=20.
2021/01/19 14:01:58.6 Spectrometer EOS_B0814: AVRG=20.
2021/01/19 14:01:58.6 Spectrometer EOS_B0735: AVRG=20.
2021/01/19 14:01:58.6 Spectrometer EOS_B0732: AVRG=20.
2021/01/19 14:01:58.6 Spectrometer EOS_B0808: AVRG=20.
2021/01/19 14:01:58.6 Spectrometer EOS_B0733: AVRG=20.
2021/01/19 14:01:58.6 Spectrometer EOS_B0817: AVRG=20.
2021/01/19 14:01:58.7 Old State:6-ENGN_StateRunning_StartNewStep New State:ENGN_StateRunning_WaitPickingSensors
2021/01/19 14:01:58.7 Pick(0, 0)
2021/01/19 14:01:58.7 GMC_SetPosZ
2021/01/19 14:01:58.8 Old State:7-ENGN_StateRunning_WaitPickingSensors New State:ENGN_StateRunning_PickingSensors
2021/01/19 14:01:58.8 Temp: 2021/01/19 14:01:58, 21.0000, 23.8000
2021/01/19 14:01:58.9 GMC_ServoHere
2021/01/19 14:01:59.1 GMC_SetPosXY
2021/01/19 14:02:01.3 GMC_SetPosZ
2021/01/19 14:02:06.8 GMC_SetPosZ
2021/01/19 14:02:07.3 Old State:8-ENGN_StateRunning_PickingSensors New State:ENGN_StateRunning_CheckingSensors
2021/01/19 14:02:07.3 Spectrometer EOS_B0712: AVRG=20.
2021/01/19 14:02:07.3 Spectrometer EOS_B0734: AVRG=20.
2021/01/19 14:02:07.3 Spectrometer EOS_B0814: AVRG=20.
2021/01/19 14:02:07.3 Spectrometer EOS_B0735: AVRG=20.
2021/01/19 14:02:07.3 Spectrometer EOS_B0732: AVRG=20.
2021/01/19 14:02:07.3 Spectrometer EOS_B0808: AVRG=20.
2021/01/19 14:02:07.3 Spectrometer EOS_B0733: AVRG=20.
2021/01/19 14:02:07.3 Spectrometer EOS_B0817: AVRG=20.
2021/01/19 14:02:07.5 
2021/01/19 14:02:07.5 Detect Sensors
2021/01/19 14:02:07.5 ==============
2021/01/19 14:02:07.5 1	YES		0.0	6444.7
2021/01/19 14:02:07.5 2	YES		0.0	6432.8
2021/01/19 14:02:07.5 3	YES		0.0	6990.6
2021/01/19 14:02:07.5 4	YES		0.0	5875.8
2021/01/19 14:02:07.5 5	YES		0.0	5642.1
2021/01/19 14:02:07.5 6	YES		0.0	6369.2
2021/01/19 14:02:07.5 7	YES		0.0	6429.2
2021/01/19 14:02:07.5 8	YES		0.0	6251.4
2021/01/19 14:02:07.5 
2021/01/19 14:02:07.5 Spectrometer EOS_B0712: AVRG=20.
2021/01/19 14:02:07.5 Spectrometer EOS_B0734: AVRG=20.
2021/01/19 14:02:07.5 Spectrometer EOS_B0814: AVRG=20.
2021/01/19 14:02:07.5 Spectrometer EOS_B0735: AVRG=20.
2021/01/19 14:02:07.5 Spectrometer EOS_B0732: AVRG=20.
2021/01/19 14:02:07.5 Spectrometer EOS_B0808: AVRG=20.
2021/01/19 14:02:07.5 Spectrometer EOS_B0733: AVRG=20.
2021/01/19 14:02:07.5 Spectrometer EOS_B0817: AVRG=20.
2021/01/19 14:02:07.5 Spectrometer EOS_B0712: AVRG=20.
2021/01/19 14:02:07.5 Spectrometer EOS_B0734: AVRG=20.
2021/01/19 14:02:07.5 Spectrometer EOS_B0814: AVRG=20.
2021/01/19 14:02:07.5 Spectrometer EOS_B0735: AVRG=20.
2021/01/19 14:02:07.5 Spectrometer EOS_B0732: AVRG=20.
2021/01/19 14:02:07.5 Spectrometer EOS_B0808: AVRG=20.
2021/01/19 14:02:07.5 Spectrometer EOS_B0733: AVRG=20.
2021/01/19 14:02:07.5 Spectrometer EOS_B0817: AVRG=20.
2021/01/19 14:02:07.6 CH1: Correction Factor = 2.01204
2021/01/19 14:02:07.7 Spectrometer EOS_B0712: AVRG=20.
2021/01/19 14:02:07.7 CH2: Correction Factor = 2.01323
2021/01/19 14:02:07.7 Spectrometer EOS_B0734: AVRG=20.
2021/01/19 14:02:07.7 CH3: Correction Factor = 1.90479
2021/01/19 14:02:07.7 Spectrometer EOS_B0814: AVRG=20.
2021/01/19 14:02:07.7 CH4: Correction Factor = 2.15616
2021/01/19 14:02:07.7 Spectrometer EOS_B0735: AVRG=20.
2021/01/19 14:02:07.7 CH5: Correction Factor = 2.30547
2021/01/19 14:02:07.7 Spectrometer EOS_B0732: AVRG=20.
2021/01/19 14:02:07.7 CH6: Correction Factor = 2.07395
2021/01/19 14:02:07.7 Spectrometer EOS_B0808: AVRG=20.
2021/01/19 14:02:07.7 CH7: Correction Factor = 2.01727
2021/01/19 14:02:07.7 Spectrometer EOS_B0733: AVRG=20.
2021/01/19 14:02:07.7 CH8: Correction Factor = 2.10799
2021/01/19 14:02:07.7 Spectrometer EOS_B0817: AVRG=20.
2021/01/19 14:02:07.8 CH1: Correction Factor = 1.10848
2021/01/19 14:02:07.8 Spectrometer EOS_B0712: AVRG=20.
2021/01/19 14:02:07.8 CH2: Correction Factor = 1.11175
2021/01/19 14:02:07.8 Spectrometer EOS_B0734: AVRG=20.
2021/01/19 14:02:07.8 CH3: Correction Factor = 1.09882
2021/01/19 14:02:07.8 Spectrometer EOS_B0814: AVRG=20.
2021/01/19 14:02:07.8 CH4: Correction Factor = 1.11747
2021/01/19 14:02:07.8 Spectrometer EOS_B0735: AVRG=20.
2021/01/19 14:02:07.8 CH5: Correction Factor = 1.12404
2021/01/19 14:02:07.8 Spectrometer EOS_B0732: AVRG=20.
2021/01/19 14:02:07.8 CH6: Correction Factor = 1.11331
2021/01/19 14:02:07.8 Spectrometer EOS_B0808: AVRG=20.
2021/01/19 14:02:07.8 CH7: Correction Factor = 1.10668
2021/01/19 14:02:07.8 Spectrometer EOS_B0733: AVRG=20.
2021/01/19 14:02:07.8 CH8: Correction Factor = 1.11258
2021/01/19 14:02:07.8 Spectrometer EOS_B0817: AVRG=20.
2021/01/19 14:02:08.0 CH1: Correction Factor = 1.0193
2021/01/19 14:02:08.0 Spectrometer EOS_B0712: AVRG=20.
2021/01/19 14:02:08.0 CH2: Correction Factor = 1.01893
2021/01/19 14:02:08.0 Spectrometer EOS_B0734: AVRG=20.
2021/01/19 14:02:08.0 CH3: Correction Factor = 1.02244
2021/01/19 14:02:08.0 Spectrometer EOS_B0814: AVRG=20.
2021/01/19 14:02:08.0 CH4: Correction Factor = 1.02361
2021/01/19 14:02:08.0 Spectrometer EOS_B0735: AVRG=20.
2021/01/19 14:02:08.0 CH5: Correction Factor = 1.02487
2021/01/19 14:02:08.0 Spectrometer EOS_B0732: AVRG=20.
2021/01/19 14:02:08.0 CH6: Correction Factor = 1.01927
2021/01/19 14:02:08.0 Spectrometer EOS_B0808: AVRG=20.
2021/01/19 14:02:08.0 CH7: Correction Factor = 1.02248
2021/01/19 14:02:08.0 Spectrometer EOS_B0733: AVRG=20.
2021/01/19 14:02:08.0 CH8: Correction Factor = 1.01983
2021/01/19 14:02:08.0 Spectrometer EOS_B0817: AVRG=20.
2021/01/19 14:02:08.1 CH1: Correction Factor = 1.00512
2021/01/19 14:02:08.1 Spectrometer EOS_B0712: AVRG=20.
2021/01/19 14:02:08.1 CH2: Correction Factor = 1.00478
2021/01/19 14:02:08.1 Spectrometer EOS_B0734: AVRG=20.
2021/01/19 14:02:08.1 CH3: Correction Factor = 1.00345
2021/01/19 14:02:08.1 Spectrometer EOS_B0814: AVRG=20.
2021/01/19 14:02:08.1 CH4: Correction Factor = 1.00382
2021/01/19 14:02:08.1 Spectrometer EOS_B0735: AVRG=20.
2021/01/19 14:02:08.1 CH5: Correction Factor = 1.00344
2021/01/19 14:02:08.1 Spectrometer EOS_B0732: AVRG=20.
2021/01/19 14:02:08.1 CH6: Correction Factor = 1.00549
2021/01/19 14:02:08.1 Spectrometer EOS_B0808: AVRG=20.
2021/01/19 14:02:08.1 CH7: Correction Factor = 1.00487
2021/01/19 14:02:08.1 Spectrometer EOS_B0733: AVRG=20.
2021/01/19 14:02:08.1 CH8: Correction Factor = 1.00628
2021/01/19 14:02:08.1 Spectrometer EOS_B0817: AVRG=20.
2021/01/19 14:02:08.3 CH1: Correction Factor = 0.998028
2021/01/19 14:02:08.3 Spectrometer EOS_B0712: AVRG=20.
2021/01/19 14:02:08.3 CH2: Correction Factor = 1.00141
2021/01/19 14:02:08.3 Spectrometer EOS_B0734: AVRG=20.
2021/01/19 14:02:08.3 CH3: Correction Factor = 0.999922
2021/01/19 14:02:08.3 Spectrometer EOS_B0814: AVRG=20.
2021/01/19 14:02:08.3 CH4: Correction Factor = 1.00064
2021/01/19 14:02:08.3 Spectrometer EOS_B0735: AVRG=20.
2021/01/19 14:02:08.3 CH5: Correction Factor = 1.00053
2021/01/19 14:02:08.3 Spectrometer EOS_B0732: AVRG=20.
2021/01/19 14:02:08.3 CH6: Correction Factor = 0.99899
2021/01/19 14:02:08.3 Spectrometer EOS_B0808: AVRG=20.
2021/01/19 14:02:08.3 CH7: Correction Factor = 1.00139
2021/01/19 14:02:08.3 Spectrometer EOS_B0733: AVRG=20.
2021/01/19 14:02:08.3 CH8: Correction Factor = 0.999776
2021/01/19 14:02:08.3 Spectrometer EOS_B0817: AVRG=20.
2021/01/19 14:02:08.5 CH1: Correction Factor = 1.0015
2021/01/19 14:02:08.5 Spectrometer EOS_B0712: AVRG=20.
2021/01/19 14:02:08.5 CH2: Correction Factor = 0.997995
2021/01/19 14:02:08.5 Spectrometer EOS_B0734: AVRG=20.
2021/01/19 14:02:08.5 CH3: Correction Factor = 0.999839
2021/01/19 14:02:08.5 Spectrometer EOS_B0814: AVRG=20.
2021/01/19 14:02:08.5 CH4: Correction Factor = 1.00065
2021/01/19 14:02:08.5 Spectrometer EOS_B0735: AVRG=20.
2021/01/19 14:02:08.5 CH5: Correction Factor = 1.00047
2021/01/19 14:02:08.5 Spectrometer EOS_B0732: AVRG=20.
2021/01/19 14:02:08.5 CH6: Correction Factor = 1.00226
2021/01/19 14:02:08.5 Spectrometer EOS_B0808: AVRG=20.
2021/01/19 14:02:08.5 CH7: Correction Factor = 0.998037
2021/01/19 14:02:08.5 Spectrometer EOS_B0733: AVRG=20.
2021/01/19 14:02:08.5 CH8: Correction Factor = 0.99971
2021/01/19 14:02:08.5 Spectrometer EOS_B0817: AVRG=20.
2021/01/19 14:02:08.5 Sensor 1: Integration Time = 2.28 ms
2021/01/19 14:02:08.5 Sensor 2: Integration Time = 2.29 ms
2021/01/19 14:02:08.5 Sensor 3: Integration Time = 2.15 ms
2021/01/19 14:02:08.5 Sensor 4: Integration Time = 2.48 ms
2021/01/19 14:02:08.5 Sensor 5: Integration Time = 2.67 ms
2021/01/19 14:02:08.5 Sensor 6: Integration Time = 2.37 ms
2021/01/19 14:02:08.5 Sensor 7: Integration Time = 2.29 ms
2021/01/19 14:02:08.5 Sensor 8: Integration Time = 2.41 ms
2021/01/19 14:02:08.5 WaitControllerIdle(30000)
2021/01/19 14:02:08.6 GMC_HomeZ
2021/01/19 14:02:08.6 HomeDevice[0005]
2021/01/19 14:02:15.3 GMC_SetPosZ
2021/01/19 14:02:15.7 GMC_ServoHere
2021/01/19 14:02:16.0 Old State:9-ENGN_StateRunning_CheckingSensors New State:ENGN_StateRunning_PickingSensorsComplete
2021/01/19 14:02:16.0 StartShaker(1000)
2021/01/19 14:02:16.0 Integration Time: 2.28,2.29,2.15,2.48,2.67,2.37,2.29,2.41,
2021/01/19 14:02:16.1 Plate 1 temperature = 21 C
2021/01/19 14:02:16.1 Pickups: 321, Warn: 90, Error: 100
2021/01/19 14:02:16.1 Old State:10-ENGN_StateRunning_PickingSensorsComplete New State:ENGN_StateRunning_WaitGoingToSample
2021/01/19 14:02:16.1 Sample96_Q(0, 0, 0)
2021/01/19 14:02:16.1 
2021/01/19 14:02:16.1 START SCAN: 
2021/01/19 14:02:16.1    CycleTime (ms)    = 200
2021/01/19 14:02:16.1    CycleCount        = 300
2021/01/19 14:02:16.1    ScansToAverage    = 20
2021/01/19 14:02:16.1    FinishTime (s)    = 60
2021/01/19 14:02:16.1    Temperature       = 21.0
2021/01/19 14:02:16.1    Lamp Power        = 31
2021/01/19 14:02:16.1    Spec PCB temps    = 41.8, 41.4, 42.6, 42.1, 41.9, 41.6, 40.3, 40.6
2021/01/19 14:02:16.1 
2021/01/19 14:02:16.2 GMC_SetPosZ
2021/01/19 14:02:16.2 Old State:12-ENGN_StateRunning_WaitGoingToSample New State:ENGN_StateRunning_GoingToSample
2021/01/19 14:02:16.4 GMC_ServoHere
2021/01/19 14:02:16.5 GMC_SetPosXY
2021/01/19 14:02:18.6 GMC_SetPosZ
2021/01/19 14:02:21.8 Old State:13-ENGN_StateRunning_GoingToSample New State:ENGN_StateRunning_GoingToSampleComplete
2021/01/19 14:02:21.9 Old State:14-ENGN_StateRunning_GoingToSampleComplete New State:ENGN_StateRunning_WaitSampling
2021/01/19 14:02:21.9 Step started at: 14:02:16
2021/01/19 14:02:21.9 XYZ movement time  = 5596 ms + Scan start delay = 5601 ms
2021/01/19 14:02:22.0 Old State:15-ENGN_StateRunning_WaitSampling New State:ENGN_StateRunning_Sampling
2021/01/19 14:02:26.0 Shaker current speed = 996 rpm, set speed = 1000
2021/01/19 14:02:59.0 Temp: 2021/01/19 14:02:59, 21.0000, 23.8000
2021/01/19 14:03:21.7 
2021/01/19 14:03:21.7 STOP SCAN: 
2021/01/19 14:03:21.7    CycleTime (ms)    = 200
2021/01/19 14:03:21.7    CurrentCycle      = 300
2021/01/19 14:03:21.7    RunTime (ms)      = 59958
2021/01/19 14:03:21.7    AverageCycle (ms) = 199.9
2021/01/19 14:03:21.7    Temperature       = 21.0
2021/01/19 14:03:21.7    DATA OVERRUN %    = 0.00
2021/01/19 14:03:21.7    Spec PCB temps    = 41.9, 41.4, 42.6, 42.3, 42.0, 41.6, 40.4, 40.7
2021/01/19 14:03:21.7 
2021/01/19 14:03:21.7 Old State:16-cycle=300, Elapsed Time:   60.0,  Ideal Time:   59.8,  Delay time =   0.04  New State:ENGN_StateRunning_SamplingComplete
2021/01/19 14:03:21.9 Old State:18-ENGN_StateRunning_SamplingComplete New State:ENGN_StateRunning_WaitNewStep
2021/01/19 14:03:22.1 Old State:5-ENGN_StateRunning_WaitNewStep New State:ENGN_StateRunning_StartNewStep
2021/01/19 14:03:22.1 Spectrometer EOS_B0712: AVRG=20.
2021/01/19 14:03:22.1 Spectrometer EOS_B0734: AVRG=20.
2021/01/19 14:03:22.1 Spectrometer EOS_B0814: AVRG=20.
2021/01/19 14:03:22.1 Spectrometer EOS_B0735: AVRG=20.
2021/01/19 14:03:22.1 Spectrometer EOS_B0732: AVRG=20.
2021/01/19 14:03:22.1 Spectrometer EOS_B0808: AVRG=20.
2021/01/19 14:03:22.1 Spectrometer EOS_B0733: AVRG=20.
2021/01/19 14:03:22.1 Spectrometer EOS_B0817: AVRG=20.
2021/01/19 14:03:22.1 StartShaker(1000)
2021/01/19 14:03:22.3 Old State:6-ENGN_StateRunning_StartNewStep New State:ENGN_StateRunning_WaitGoingToSample
2021/01/19 14:03:22.3 Sample96_Q(0, 0, 1)
2021/01/19 14:03:22.3 
2021/01/19 14:03:22.3 START SCAN: 
2021/01/19 14:03:22.3    CycleTime (ms)    = 200
2021/01/19 14:03:22.3    CycleCount        = 1500
2021/01/19 14:03:22.3    ScansToAverage    = 20
2021/01/19 14:03:22.3    FinishTime (s)    = 300
2021/01/19 14:03:22.3    Temperature       = 21.0
2021/01/19 14:03:22.3    Lamp Power        = 31
2021/01/19 14:03:22.3    Spec PCB temps    = 41.9, 41.5, 42.6, 42.3, 42.0, 41.6, 40.4, 40.7
2021/01/19 14:03:22.3 
2021/01/19 14:03:22.4 Old State:12-ENGN_StateRunning_WaitGoingToSample New State:ENGN_StateRunning_GoingToSample
2021/01/19 14:03:22.4 Temp: 2021/01/19 14:03:22, 21.0000, 23.8000
2021/01/19 14:03:22.4 GMC_SetPosZ
2021/01/19 14:03:25.4 GMC_ServoHere
2021/01/19 14:03:25.6 GMC_SetPosXY
2021/01/19 14:03:25.9 GMC_SetPosZ
2021/01/19 14:03:29.4 Old State:13-cycle=1, Elapsed Time:    0.0,  Ideal Time:    0.0,  Delay time =   0.00  New State:ENGN_StateRunning_GoingToSampleComplete
2021/01/19 14:03:29.5 Old State:14-ENGN_StateRunning_GoingToSampleComplete New State:ENGN_StateRunning_WaitSampling
2021/01/19 14:03:29.5 Step started at: 14:03:22
2021/01/19 14:03:29.5 XYZ movement time  = 7064 ms + Scan start delay = 6999 ms
2021/01/19 14:03:29.6 Old State:15-ENGN_StateRunning_WaitSampling New State:ENGN_StateRunning_Sampling
2021/01/19 14:03:32.1 Shaker current speed = 997 rpm, set speed = 1000
2021/01/19 14:04:22.5 Temp: 2021/01/19 14:04:22, 21.0000, 23.8000
2021/01/19 14:05:22.7 Temp: 2021/01/19 14:05:22, 21.0000, 23.8000
2021/01/19 14:06:22.8 Temp: 2021/01/19 14:06:22, 21.0000, 23.8000
2021/01/19 14:07:23.0 Temp: 2021/01/19 14:07:23, 21.0000, 23.8000
2021/01/19 14:08:23.2 Temp: 2021/01/19 14:08:23, 21.0000, 23.8000
2021/01/19 14:08:29.2 
2021/01/19 14:08:29.2 STOP SCAN: 
2021/01/19 14:08:29.2    CycleTime (ms)    = 200
2021/01/19 14:08:29.2    CurrentCycle      = 1500
2021/01/19 14:08:29.2    RunTime (ms)      = 299957
2021/01/19 14:08:29.2    AverageCycle (ms) = 200.0
2021/01/19 14:08:29.2    Temperature       = 21.0
2021/01/19 14:08:29.2    DATA OVERRUN %    = 0.00
2021/01/19 14:08:29.2    Spec PCB temps    = 42.0, 41.5, 42.6, 42.3, 42.1, 41.7, 40.4, 40.8
2021/01/19 14:08:29.2 
2021/01/19 14:08:29.3 Old State:16-cycle=1500, Elapsed Time:  300.0,  Ideal Time:  299.8,  Delay time =   0.04  New State:ENGN_StateRunning_SamplingComplete
2021/01/19 14:08:29.5 Old State:18-ENGN_StateRunning_SamplingComplete New State:ENGN_StateRunning_WaitNewStep
2021/01/19 14:08:29.8 Old State:5-ENGN_StateRunning_WaitNewStep New State:ENGN_StateRunning_StartNewStep
2021/01/19 14:08:29.8 Spectrometer EOS_B0712: AVRG=20.
2021/01/19 14:08:29.8 Spectrometer EOS_B0734: AVRG=20.
2021/01/19 14:08:29.8 Spectrometer EOS_B0814: AVRG=20.
2021/01/19 14:08:29.8 Spectrometer EOS_B0735: AVRG=20.
2021/01/19 14:08:29.8 Spectrometer EOS_B0732: AVRG=20.
2021/01/19 14:08:29.8 Spectrometer EOS_B0808: AVRG=20.
2021/01/19 14:08:29.8 Spectrometer EOS_B0733: AVRG=20.
2021/01/19 14:08:29.8 Spectrometer EOS_B0817: AVRG=20.
2021/01/19 14:08:29.8 StartShaker(1000)
2021/01/19 14:08:29.9 Old State:6-ENGN_StateRunning_StartNewStep New State:ENGN_StateRunning_WaitGoingToSample
2021/01/19 14:08:29.9 Sample96_Q(0, 0, 2)
2021/01/19 14:08:29.9 
2021/01/19 14:08:29.9 START SCAN: 
2021/01/19 14:08:29.9    CycleTime (ms)    = 200
2021/01/19 14:08:29.9    CycleCount        = 150
2021/01/19 14:08:29.9    ScansToAverage    = 20
2021/01/19 14:08:29.9    FinishTime (s)    = 30
2021/01/19 14:08:29.9    Temperature       = 21.0
2021/01/19 14:08:29.9    Lamp Power        = 31
2021/01/19 14:08:29.9    Spec PCB temps    = 41.9, 41.5, 42.6, 42.3, 42.0, 41.7, 40.4, 40.8
2021/01/19 14:08:29.9 
2021/01/19 14:08:30.0 GMC_SetPosZ
2021/01/19 14:08:30.0 Temp: 2021/01/19 14:08:30, 21.0000, 23.8000
2021/01/19 14:08:30.0 Old State:12-ENGN_StateRunning_WaitGoingToSample New State:ENGN_StateRunning_GoingToSample
2021/01/19 14:08:33.4 GMC_ServoHere
2021/01/19 14:08:33.5 GMC_SetPosXY
2021/01/19 14:08:33.8 GMC_SetPosZ
2021/01/19 14:08:37.6 Old State:13-ENGN_StateRunning_GoingToSample New State:ENGN_StateRunning_GoingToSampleComplete
2021/01/19 14:08:37.7 Old State:14-cycle=1, Elapsed Time:    0.0,  Ideal Time:    0.0,  Delay time =   0.00  New State:ENGN_StateRunning_WaitSampling
2021/01/19 14:08:37.7 Step started at: 14:08:29
2021/01/19 14:08:37.7 XYZ movement time  = 7562 ms + Scan start delay = 7546 ms
2021/01/19 14:08:37.8 Old State:15-ENGN_StateRunning_WaitSampling New State:ENGN_StateRunning_Sampling
2021/01/19 14:08:39.8 Shaker current speed = 994 rpm, set speed = 1000
2021/01/19 14:09:07.4 
2021/01/19 14:09:07.4 STOP SCAN: 
2021/01/19 14:09:07.4    CycleTime (ms)    = 200
2021/01/19 14:09:07.4    CurrentCycle      = 150
2021/01/19 14:09:07.4    RunTime (ms)      = 29958
2021/01/19 14:09:07.4    AverageCycle (ms) = 199.7
2021/01/19 14:09:07.4    Temperature       = 21.0
2021/01/19 14:09:07.4    DATA OVERRUN %    = 0.00
2021/01/19 14:09:07.4    Spec PCB temps    = 41.9, 41.5, 42.6, 42.3, 42.1, 41.7, 40.4, 40.8
2021/01/19 14:09:07.4 
2021/01/19 14:09:07.5 Old State:16-cycle=150, Elapsed Time:   30.0,  Ideal Time:   29.8,  Delay time =   0.04  New State:ENGN_StateRunning_SamplingComplete
2021/01/19 14:09:07.6 Old State:18-ENGN_StateRunning_SamplingComplete New State:ENGN_StateRunning_WaitNewStep
2021/01/19 14:09:07.9 Old State:5-ENGN_StateRunning_WaitNewStep New State:ENGN_StateRunning_StartNewStep
2021/01/19 14:09:08.0 Spectrometer EOS_B0712: AVRG=20.
2021/01/19 14:09:08.0 Spectrometer EOS_B0734: AVRG=20.
2021/01/19 14:09:08.0 Spectrometer EOS_B0814: AVRG=20.
2021/01/19 14:09:08.0 Spectrometer EOS_B0735: AVRG=20.
2021/01/19 14:09:08.0 Spectrometer EOS_B0732: AVRG=20.
2021/01/19 14:09:08.0 Spectrometer EOS_B0808: AVRG=20.
2021/01/19 14:09:08.0 Spectrometer EOS_B0733: AVRG=20.
2021/01/19 14:09:08.0 Spectrometer EOS_B0817: AVRG=20.
2021/01/19 14:09:08.0 StartShaker(1000)
2021/01/19 14:09:08.1 Old State:6-ENGN_StateRunning_StartNewStep New State:ENGN_StateRunning_WaitGoingToSample
2021/01/19 14:09:08.1 Sample96_Q(0, 0, 3)
2021/01/19 14:09:08.1 
2021/01/19 14:09:08.1 START SCAN: 
2021/01/19 14:09:08.1    CycleTime (ms)    = 200
2021/01/19 14:09:08.1    CycleCount        = 600
2021/01/19 14:09:08.1    ScansToAverage    = 20
2021/01/19 14:09:08.1    FinishTime (s)    = 120
2021/01/19 14:09:08.1    Temperature       = 21.0
2021/01/19 14:09:08.1    Lamp Power        = 31
2021/01/19 14:09:08.1    Spec PCB temps    = 42.0, 41.5, 42.6, 42.3, 42.1, 41.7, 40.4, 40.8
2021/01/19 14:09:08.1 
2021/01/19 14:09:08.2 Temp: 2021/01/19 14:09:08, 21.0000, 23.8000
2021/01/19 14:09:08.2 Old State:12-ENGN_StateRunning_WaitGoingToSample New State:ENGN_StateRunning_GoingToSample
2021/01/19 14:09:08.2 GMC_SetPosZ
2021/01/19 14:09:11.7 GMC_ServoHere
2021/01/19 14:09:11.9 GMC_SetPosXY
2021/01/19 14:09:12.2 GMC_SetPosZ
2021/01/19 14:09:16.2 Old State:13-ENGN_StateRunning_GoingToSample New State:ENGN_StateRunning_GoingToSampleComplete
2021/01/19 14:09:16.3 Old State:14-cycle=1, Elapsed Time:    0.0,  Ideal Time:    0.0,  Delay time =   0.00  New State:ENGN_StateRunning_WaitSampling
2021/01/19 14:09:16.3 Step started at: 14:09:08
2021/01/19 14:09:16.3 XYZ movement time  = 8051 ms + Scan start delay = 8053 ms
2021/01/19 14:09:16.4 Old State:15-ENGN_StateRunning_WaitSampling New State:ENGN_StateRunning_Sampling
2021/01/19 14:09:18.1 Shaker current speed = 994 rpm, set speed = 1000
2021/01/19 14:10:08.3 Temp: 2021/01/19 14:10:08, 21.0000, 23.8000
2021/01/19 14:11:08.5 Temp: 2021/01/19 14:11:08, 21.0000, 23.8000
2021/01/19 14:11:16.1 
2021/01/19 14:11:16.1 STOP SCAN: 
2021/01/19 14:11:16.1    CycleTime (ms)    = 200
2021/01/19 14:11:16.1    CurrentCycle      = 600
2021/01/19 14:11:16.1    RunTime (ms)      = 119959
2021/01/19 14:11:16.1    AverageCycle (ms) = 199.9
2021/01/19 14:11:16.1    Temperature       = 21.0
2021/01/19 14:11:16.1    DATA OVERRUN %    = 0.00
2021/01/19 14:11:16.1    Spec PCB temps    = 42.0, 41.6, 42.6, 42.3, 42.1, 41.8, 40.5, 40.8
2021/01/19 14:11:16.1 
2021/01/19 14:11:16.1 Old State:16-cycle=600, Elapsed Time:  120.0,  Ideal Time:  119.8,  Delay time =   0.04  New State:ENGN_StateRunning_SamplingComplete
2021/01/19 14:11:16.3 Old State:18-ENGN_StateRunning_SamplingComplete New State:ENGN_StateRunning_WaitNewStep
2021/01/19 14:11:16.5 Old State:5-ENGN_StateRunning_WaitNewStep New State:ENGN_StateRunning_StartNewStep
2021/01/19 14:11:16.5 Spectrometer EOS_B0712: AVRG=20.
2021/01/19 14:11:16.5 Spectrometer EOS_B0734: AVRG=20.
2021/01/19 14:11:16.5 Spectrometer EOS_B0814: AVRG=20.
2021/01/19 14:11:16.5 Spectrometer EOS_B0735: AVRG=20.
2021/01/19 14:11:16.5 Spectrometer EOS_B0732: AVRG=20.
2021/01/19 14:11:16.5 Spectrometer EOS_B0808: AVRG=20.
2021/01/19 14:11:16.5 Spectrometer EOS_B0733: AVRG=20.
2021/01/19 14:11:16.5 Spectrometer EOS_B0817: AVRG=20.
2021/01/19 14:11:16.5 StartShaker(1000)
2021/01/19 14:11:16.7 Old State:6-ENGN_StateRunning_StartNewStep New State:ENGN_StateRunning_WaitGoingToSample
2021/01/19 14:11:16.7 Sample96_Q(0, 0, 4)
2021/01/19 14:11:16.7 
2021/01/19 14:11:16.7 START SCAN: 
2021/01/19 14:11:16.7    CycleTime (ms)    = 200
2021/01/19 14:11:16.7    CycleCount        = 250
2021/01/19 14:11:16.7    ScansToAverage    = 20
2021/01/19 14:11:16.7    FinishTime (s)    = 50
2021/01/19 14:11:16.7    Temperature       = 21.0
2021/01/19 14:11:16.7    Lamp Power        = 31
2021/01/19 14:11:16.7    Spec PCB temps    = 42.0, 41.6, 42.6, 42.3, 42.1, 41.8, 40.5, 40.8
2021/01/19 14:11:16.7 
2021/01/19 14:11:16.7 GMC_SetPosZ
2021/01/19 14:11:16.8 Temp: 2021/01/19 14:11:16, 21.0000, 23.8000
2021/01/19 14:11:16.8 Old State:12-ENGN_StateRunning_WaitGoingToSample New State:ENGN_StateRunning_GoingToSample
2021/01/19 14:11:20.5 GMC_ServoHere
2021/01/19 14:11:20.7 GMC_SetPosXY
2021/01/19 14:11:21.1 GMC_SetPosZ
2021/01/19 14:11:25.4 Old State:13-ENGN_StateRunning_GoingToSample New State:ENGN_StateRunning_GoingToSampleComplete
2021/01/19 14:11:25.5 Old State:14-cycle=1, Elapsed Time:    0.0,  Ideal Time:    0.0,  Delay time =   0.00  New State:ENGN_StateRunning_WaitSampling
2021/01/19 14:11:25.5 Step started at: 14:11:16
2021/01/19 14:11:25.5 XYZ movement time  = 8663 ms + Scan start delay = 8628 ms
2021/01/19 14:11:25.6 Old State:15-ENGN_StateRunning_WaitSampling New State:ENGN_StateRunning_Sampling
2021/01/19 14:11:26.5 Shaker current speed = 997 rpm, set speed = 1000
2021/01/19 14:12:15.3 
2021/01/19 14:12:15.3 STOP SCAN: 
2021/01/19 14:12:15.3    CycleTime (ms)    = 200
2021/01/19 14:12:15.3    CurrentCycle      = 250
2021/01/19 14:12:15.3    RunTime (ms)      = 49958
2021/01/19 14:12:15.3    AverageCycle (ms) = 199.8
2021/01/19 14:12:15.3    Temperature       = 21.0
2021/01/19 14:12:15.3    DATA OVERRUN %    = 0.00
2021/01/19 14:12:15.3    Spec PCB temps    = 42.0, 41.6, 42.6, 42.3, 42.0, 41.8, 40.5, 40.8
2021/01/19 14:12:15.3 
2021/01/19 14:12:15.3 Old State:16-cycle=250, Elapsed Time:   50.0,  Ideal Time:   49.8,  Delay time =   0.04  New State:ENGN_StateRunning_SamplingComplete
2021/01/19 14:12:15.4 Old State:18-ENGN_StateRunning_SamplingComplete New State:ENGN_StateRunning_WaitNewStep
2021/01/19 14:12:15.6 Old State:5-ENGN_StateRunning_WaitNewStep New State:ENGN_StateRunning_StartNewStep
2021/01/19 14:12:15.6 Spectrometer EOS_B0712: AVRG=20.
2021/01/19 14:12:15.6 Spectrometer EOS_B0734: AVRG=20.
2021/01/19 14:12:15.6 Spectrometer EOS_B0814: AVRG=20.
2021/01/19 14:12:15.6 Spectrometer EOS_B0735: AVRG=20.
2021/01/19 14:12:15.6 Spectrometer EOS_B0732: AVRG=20.
2021/01/19 14:12:15.6 Spectrometer EOS_B0808: AVRG=20.
2021/01/19 14:12:15.6 Spectrometer EOS_B0733: AVRG=20.
2021/01/19 14:12:15.6 Spectrometer EOS_B0817: AVRG=20.
2021/01/19 14:12:15.6 StartShaker(1000)
2021/01/19 14:12:15.7 Old State:6-ENGN_StateRunning_StartNewStep New State:ENGN_StateRunning_WaitGoingToSample
2021/01/19 14:12:15.7 Sample96_Q(0, 0, 3)
2021/01/19 14:12:15.7 
2021/01/19 14:12:15.7 START SCAN: 
2021/01/19 14:12:15.7    CycleTime (ms)    = 200
2021/01/19 14:12:15.7    CycleCount        = 600
2021/01/19 14:12:15.7    ScansToAverage    = 20
2021/01/19 14:12:15.7    FinishTime (s)    = 120
2021/01/19 14:12:15.7    Temperature       = 21.0
2021/01/19 14:12:15.7    Lamp Power        = 31
2021/01/19 14:12:15.7    Spec PCB temps    = 42.0, 41.6, 42.6, 42.3, 42.1, 41.8, 40.5, 40.8
2021/01/19 14:12:15.7 
2021/01/19 14:12:15.8 GMC_SetPosZ
2021/01/19 14:12:15.8 Old State:12-ENGN_StateRunning_WaitGoingToSample New State:ENGN_StateRunning_GoingToSample
2021/01/19 14:12:15.9 Temp: 2021/01/19 14:12:15, 21.0000, 23.8000
2021/01/19 14:12:19.9 GMC_ServoHere
2021/01/19 14:12:20.0 GMC_SetPosXY
2021/01/19 14:12:20.3 GMC_SetPosZ
2021/01/19 14:12:24.5 Old State:13-cycle=1, Elapsed Time:    0.0,  Ideal Time:    0.0,  Delay time =   0.00  New State:ENGN_StateRunning_GoingToSampleComplete
2021/01/19 14:12:24.6 Old State:14-ENGN_StateRunning_GoingToSampleComplete New State:ENGN_StateRunning_WaitSampling
2021/01/19 14:12:24.6 Step started at: 14:12:15
2021/01/19 14:12:24.6 XYZ movement time  = 8660 ms + Scan start delay = 8573 ms
2021/01/19 14:12:24.7 Old State:15-ENGN_StateRunning_WaitSampling New State:ENGN_StateRunning_Sampling
2021/01/19 14:12:25.7 Shaker current speed = 989 rpm, set speed = 1000
2021/01/19 14:13:16.0 Temp: 2021/01/19 14:13:16, 21.0000, 23.8000
2021/01/19 14:14:16.2 Temp: 2021/01/19 14:14:16, 21.0000, 23.8000
2021/01/19 14:14:24.3 
2021/01/19 14:14:24.3 STOP SCAN: 
2021/01/19 14:14:24.3    CycleTime (ms)    = 200
2021/01/19 14:14:24.3    CurrentCycle      = 600
2021/01/19 14:14:24.3    RunTime (ms)      = 119957
2021/01/19 14:14:24.3    AverageCycle (ms) = 199.9
2021/01/19 14:14:24.3    Temperature       = 21.0
2021/01/19 14:14:24.3    DATA OVERRUN %    = 0.00
2021/01/19 14:14:24.3    Spec PCB temps    = 42.1, 41.6, 42.5, 42.3, 42.0, 41.7, 40.5, 40.9
2021/01/19 14:14:24.3 
2021/01/19 14:14:24.3 Old State:16-cycle=600, Elapsed Time:  120.0,  Ideal Time:  119.8,  Delay time =   0.04  New State:ENGN_StateRunning_SamplingComplete
2021/01/19 14:14:24.4 Old State:18-ENGN_StateRunning_SamplingComplete New State:ENGN_StateRunning_WaitEjecting
2021/01/19 14:14:24.4 EJECT: To Tray, position=0,0
2021/01/19 14:14:24.4    EjectOffset = 10.0
2021/01/19 14:14:24.4 GMC_SetPosZ
2021/01/19 14:14:24.5 Old State:19-ENGN_StateRunning_WaitEjecting New State:ENGN_StateRunning_Ejecting
2021/01/19 14:14:28.2 GMC_ServoHere
2021/01/19 14:14:28.4 GMC_SetPosXY
2021/01/19 14:14:30.6 GMC_SetPosZ
2021/01/19 14:14:34.7 GMC_Eject
2021/01/19 14:14:34.7 EjectDone => EjectEject
2021/01/19 14:14:34.7 SetEjectorPosn(0)
2021/01/19 14:14:37.8 GMC_HomeZ
2021/01/19 14:14:37.8 HomeDevice[0005]
2021/01/19 14:14:43.3 GMC_SetPosZ
2021/01/19 14:14:43.6 GMC_ServoHere
2021/01/19 14:14:43.8 GMC_SetPosZ
2021/01/19 14:14:44.0 GMC_ServoHere
2021/01/19 14:14:44.2 GMC_SetPosXY
2021/01/19 14:14:46.4 GMC_SetPosZ
2021/01/19 14:14:46.7 Old State:20-ENGN_StateRunning_Ejecting New State:ENGN_StateRunning_EjectingComplete
2021/01/19 14:14:46.8 Old State:21-ENGN_StateRunning_EjectingComplete New State:ENGN_StateRunning_WaitNewStep
2021/01/19 14:14:46.9 Old State:5-ENGN_StateRunning_WaitNewStep New State:ENGN_StateRunning_StartNewStep
2021/01/19 14:14:46.9 Spectrometer EOS_B0712: AVRG=20.
2021/01/19 14:14:46.9 Spectrometer EOS_B0734: AVRG=20.
2021/01/19 14:14:46.9 Spectrometer EOS_B0814: AVRG=20.
2021/01/19 14:14:46.9 Spectrometer EOS_B0735: AVRG=20.
2021/01/19 14:14:46.9 Spectrometer EOS_B0732: AVRG=20.
2021/01/19 14:14:46.9 Spectrometer EOS_B0808: AVRG=20.
2021/01/19 14:14:46.9 Spectrometer EOS_B0733: AVRG=20.
2021/01/19 14:14:46.9 Spectrometer EOS_B0817: AVRG=20.
2021/01/19 14:14:47.0 Old State:6-ENGN_StateRunning_StartNewStep New State:ENGN_StateRunning_WaitPickingSensors
2021/01/19 14:14:47.0 Pick(0, 1)
2021/01/19 14:14:47.0 GMC_SetPosZ
2021/01/19 14:14:47.1 Old State:7-ENGN_StateRunning_WaitPickingSensors New State:ENGN_StateRunning_PickingSensors
2021/01/19 14:14:47.1 Temp: 2021/01/19 14:14:47, 21.0000, 23.8000
2021/01/19 14:14:47.2 GMC_ServoHere
2021/01/19 14:14:47.4 GMC_SetPosXY
2021/01/19 14:14:49.6 GMC_SetPosZ
2021/01/19 14:14:55.2 GMC_SetPosZ
2021/01/19 14:14:55.6 Old State:8-ENGN_StateRunning_PickingSensors New State:ENGN_StateRunning_CheckingSensors
2021/01/19 14:14:55.7 Spectrometer EOS_B0712: AVRG=20.
2021/01/19 14:14:55.7 Spectrometer EOS_B0734: AVRG=20.
2021/01/19 14:14:55.7 Spectrometer EOS_B0814: AVRG=20.
2021/01/19 14:14:55.7 Spectrometer EOS_B0735: AVRG=20.
2021/01/19 14:14:55.7 Spectrometer EOS_B0732: AVRG=20.
2021/01/19 14:14:55.7 Spectrometer EOS_B0808: AVRG=20.
2021/01/19 14:14:55.7 Spectrometer EOS_B0733: AVRG=20.
2021/01/19 14:14:55.7 Spectrometer EOS_B0817: AVRG=20.
2021/01/19 14:14:55.8 
2021/01/19 14:14:55.8 Detect Sensors
2021/01/19 14:14:55.8 ==============
2021/01/19 14:14:55.8 1	YES		0.0	6444.4
2021/01/19 14:14:55.8 2	YES		0.0	6821.8
2021/01/19 14:14:55.8 3	YES		0.0	6879.9
2021/01/19 14:14:55.8 4	YES		0.0	6133.0
2021/01/19 14:14:55.8 5	YES		0.0	5555.7
2021/01/19 14:14:55.8 6	YES		0.0	6502.4
2021/01/19 14:14:55.8 7	YES		0.0	6647.2
2021/01/19 14:14:55.8 8	YES		0.0	6037.1
2021/01/19 14:14:55.8 
2021/01/19 14:14:55.8 Spectrometer EOS_B0712: AVRG=20.
2021/01/19 14:14:55.8 Spectrometer EOS_B0734: AVRG=20.
2021/01/19 14:14:55.8 Spectrometer EOS_B0814: AVRG=20.
2021/01/19 14:14:55.8 Spectrometer EOS_B0735: AVRG=20.
2021/01/19 14:14:55.8 Spectrometer EOS_B0732: AVRG=20.
2021/01/19 14:14:55.8 Spectrometer EOS_B0808: AVRG=20.
2021/01/19 14:14:55.8 Spectrometer EOS_B0733: AVRG=20.
2021/01/19 14:14:55.8 Spectrometer EOS_B0817: AVRG=20.
2021/01/19 14:14:55.8 Spectrometer EOS_B0712: AVRG=20.
2021/01/19 14:14:55.8 Spectrometer EOS_B0734: AVRG=20.
2021/01/19 14:14:55.8 Spectrometer EOS_B0814: AVRG=20.
2021/01/19 14:14:55.8 Spectrometer EOS_B0735: AVRG=20.
2021/01/19 14:14:55.8 Spectrometer EOS_B0732: AVRG=20.
2021/01/19 14:14:55.8 Spectrometer EOS_B0808: AVRG=20.
2021/01/19 14:14:55.8 Spectrometer EOS_B0733: AVRG=20.
2021/01/19 14:14:55.8 Spectrometer EOS_B0817: AVRG=20.
2021/01/19 14:14:56.0 CH1: Correction Factor = 2.01388
2021/01/19 14:14:56.0 Spectrometer EOS_B0712: AVRG=20.
2021/01/19 14:14:56.0 CH2: Correction Factor = 1.91868
2021/01/19 14:14:56.0 Spectrometer EOS_B0734: AVRG=20.
2021/01/19 14:14:56.0 CH3: Correction Factor = 1.95487
2021/01/19 14:14:56.0 Spectrometer EOS_B0814: AVRG=20.
2021/01/19 14:14:56.0 CH4: Correction Factor = 2.09094
2021/01/19 14:14:56.0 Spectrometer EOS_B0735: AVRG=20.
2021/01/19 14:14:56.0 CH5: Correction Factor = 2.30773
2021/01/19 14:14:56.0 Spectrometer EOS_B0732: AVRG=20.
2021/01/19 14:14:56.0 CH6: Correction Factor = 2.04572
2021/01/19 14:14:56.0 Spectrometer EOS_B0808: AVRG=20.
2021/01/19 14:14:56.0 CH7: Correction Factor = 1.99471
2021/01/19 14:14:56.0 Spectrometer EOS_B0733: AVRG=20.
2021/01/19 14:14:56.0 CH8: Correction Factor = 2.13458
2021/01/19 14:14:56.0 Spectrometer EOS_B0817: AVRG=20.
2021/01/19 14:14:56.2 CH1: Correction Factor = 1.10939
2021/01/19 14:14:56.2 Spectrometer EOS_B0712: AVRG=20.
2021/01/19 14:14:56.2 CH2: Correction Factor = 1.10384
2021/01/19 14:14:56.2 Spectrometer EOS_B0734: AVRG=20.
2021/01/19 14:14:56.2 CH3: Correction Factor = 1.10139
2021/01/19 14:14:56.2 Spectrometer EOS_B0814: AVRG=20.
2021/01/19 14:14:56.2 CH4: Correction Factor = 1.11578
2021/01/19 14:14:56.2 Spectrometer EOS_B0735: AVRG=20.
2021/01/19 14:14:56.2 CH5: Correction Factor = 1.12486
2021/01/19 14:14:56.2 Spectrometer EOS_B0732: AVRG=20.
2021/01/19 14:14:56.2 CH6: Correction Factor = 1.10769
2021/01/19 14:14:56.2 Spectrometer EOS_B0808: AVRG=20.
2021/01/19 14:14:56.2 CH7: Correction Factor = 1.10388
2021/01/19 14:14:56.2 Spectrometer EOS_B0733: AVRG=20.
2021/01/19 14:14:56.2 CH8: Correction Factor = 1.11295
2021/01/19 14:14:56.2 Spectrometer EOS_B0817: AVRG=20.
2021/01/19 14:14:56.2 DFIO: Start writing C:\Users\Labuser\Desktop\Kristjan Sigmar\20210119 STAB-0005 STAB-0006 t3\210119_001.frd
2021/01/19 14:14:56.2 DFIO: Finished writing C:\Users\Labuser\Desktop\Kristjan Sigmar\20210119 STAB-0005 STAB-0006 t3\210119_001.frd
2021/01/19 14:14:56.2 DFIO: Start writing C:\Users\Labuser\Desktop\Kristjan Sigmar\20210119 STAB-0005 STAB-0006 t3\210119_002.frd
2021/01/19 14:14:56.2 DFIO: Finished writing C:\Users\Labuser\Desktop\Kristjan Sigmar\20210119 STAB-0005 STAB-0006 t3\210119_002.frd
2021/01/19 14:14:56.2 DFIO: Start writing C:\Users\Labuser\Desktop\Kristjan Sigmar\20210119 STAB-0005 STAB-0006 t3\210119_003.frd
2021/01/19 14:14:56.2 DFIO: Finished writing C:\Users\Labuser\Desktop\Kristjan Sigmar\20210119 STAB-0005 STAB-0006 t3\210119_003.frd
2021/01/19 14:14:56.2 DFIO: Start writing C:\Users\Labuser\Desktop\Kristjan Sigmar\20210119 STAB-0005 STAB-0006 t3\210119_004.frd
2021/01/19 14:14:56.2 DFIO: Finished writing C:\Users\Labuser\Desktop\Kristjan Sigmar\20210119 STAB-0005 STAB-0006 t3\210119_004.frd
2021/01/19 14:14:56.2 DFIO: Start writing C:\Users\Labuser\Desktop\Kristjan Sigmar\20210119 STAB-0005 STAB-0006 t3\210119_005.frd
2021/01/19 14:14:56.2 DFIO: Finished writing C:\Users\Labuser\Desktop\Kristjan Sigmar\20210119 STAB-0005 STAB-0006 t3\210119_005.frd
2021/01/19 14:14:56.2 DFIO: Start writing C:\Users\Labuser\Desktop\Kristjan Sigmar\20210119 STAB-0005 STAB-0006 t3\210119_006.frd
2021/01/19 14:14:56.2 DFIO: Finished writing C:\Users\Labuser\Desktop\Kristjan Sigmar\20210119 STAB-0005 STAB-0006 t3\210119_006.frd
2021/01/19 14:14:56.2 DFIO: Start writing C:\Users\Labuser\Desktop\Kristjan Sigmar\20210119 STAB-0005 STAB-0006 t3\210119_007.frd
2021/01/19 14:14:56.2 DFIO: Finished writing C:\Users\Labuser\Desktop\Kristjan Sigmar\20210119 STAB-0005 STAB-0006 t3\210119_007.frd
2021/01/19 14:14:56.2 DFIO: Start writing C:\Users\Labuser\Desktop\Kristjan Sigmar\20210119 STAB-0005 STAB-0006 t3\210119_008.frd
2021/01/19 14:14:56.2 DFIO: Finished writing C:\Users\Labuser\Desktop\Kristjan Sigmar\20210119 STAB-0005 STAB-0006 t3\210119_008.frd
2021/01/19 14:14:56.2 DFIO: Finished writing C:\Users\Labuser\Desktop\Kristjan Sigmar\20210119 STAB-0005 STAB-0006 t3\210119_Manifest.fmx
2021/01/19 14:14:56.3 CH1: Correction Factor = 1.02017
2021/01/19 14:14:56.3 Spectrometer EOS_B0712: AVRG=20.
2021/01/19 14:14:56.3 CH2: Correction Factor = 1.01943
2021/01/19 14:14:56.3 Spectrometer EOS_B0734: AVRG=20.
2021/01/19 14:14:56.3 CH3: Correction Factor = 1.02251
2021/01/19 14:14:56.3 Spectrometer EOS_B0814: AVRG=20.
2021/01/19 14:14:56.3 CH4: Correction Factor = 1.02231
2021/01/19 14:14:56.3 Spectrometer EOS_B0735: AVRG=20.
2021/01/19 14:14:56.3 CH5: Correction Factor = 1.02252
2021/01/19 14:14:56.3 Spectrometer EOS_B0732: AVRG=20.
2021/01/19 14:14:56.3 CH6: Correction Factor = 1.0206
2021/01/19 14:14:56.3 Spectrometer EOS_B0808: AVRG=20.
2021/01/19 14:14:56.3 CH7: Correction Factor = 1.02288
2021/01/19 14:14:56.3 Spectrometer EOS_B0733: AVRG=20.
2021/01/19 14:14:56.3 CH8: Correction Factor = 1.02115
2021/01/19 14:14:56.3 Spectrometer EOS_B0817: AVRG=20.
2021/01/19 14:14:56.5 CH1: Correction Factor = 1.00243
2021/01/19 14:14:56.5 Spectrometer EOS_B0712: AVRG=20.
2021/01/19 14:14:56.5 CH2: Correction Factor = 1.00442
2021/01/19 14:14:56.5 Spectrometer EOS_B0734: AVRG=20.
2021/01/19 14:14:56.5 CH3: Correction Factor = 1.00403
2021/01/19 14:14:56.5 Spectrometer EOS_B0814: AVRG=20.
2021/01/19 14:14:56.5 CH4: Correction Factor = 1.00215
2021/01/19 14:14:56.5 Spectrometer EOS_B0735: AVRG=20.
2021/01/19 14:14:56.5 CH5: Correction Factor = 1.00728
2021/01/19 14:14:56.5 Spectrometer EOS_B0732: AVRG=20.
2021/01/19 14:14:56.5 CH6: Correction Factor = 1.00671
2021/01/19 14:14:56.5 Spectrometer EOS_B0808: AVRG=20.
2021/01/19 14:14:56.5 CH7: Correction Factor = 1.00492
2021/01/19 14:14:56.5 Spectrometer EOS_B0733: AVRG=20.
2021/01/19 14:14:56.5 CH8: Correction Factor = 1.00456
2021/01/19 14:14:56.5 Spectrometer EOS_B0817: AVRG=20.
2021/01/19 14:14:56.7 CH1: Correction Factor = 0.998992
2021/01/19 14:14:56.7 Spectrometer EOS_B0712: AVRG=20.
2021/01/19 14:14:56.7 CH2: Correction Factor = 1.00094
2021/01/19 14:14:56.7 Spectrometer EOS_B0734: AVRG=20.
2021/01/19 14:14:56.7 CH3: Correction Factor = 1.00043
2021/01/19 14:14:56.7 Spectrometer EOS_B0814: AVRG=20.
2021/01/19 14:14:56.7 CH4: Correction Factor = 1.00222
2021/01/19 14:14:56.7 Spectrometer EOS_B0735: AVRG=20.
2021/01/19 14:14:56.7 CH5: Correction Factor = 1.00128
2021/01/19 14:14:56.7 Spectrometer EOS_B0732: AVRG=20.
2021/01/19 14:14:56.7 CH6: Correction Factor = 0.999755
2021/01/19 14:14:56.7 Spectrometer EOS_B0808: AVRG=20.
2021/01/19 14:14:56.7 CH7: Correction Factor = 1.00152
2021/01/19 14:14:56.7 Spectrometer EOS_B0733: AVRG=20.
2021/01/19 14:14:56.7 CH8: Correction Factor = 1.00135
2021/01/19 14:14:56.7 Spectrometer EOS_B0817: AVRG=20.
2021/01/19 14:14:56.8 CH1: Correction Factor = 1.00228
2021/01/19 14:14:56.8 Spectrometer EOS_B0712: AVRG=20.
2021/01/19 14:14:56.8 CH2: Correction Factor = 1.0008
2021/01/19 14:14:56.8 Spectrometer EOS_B0734: AVRG=20.
2021/01/19 14:14:56.8 CH3: Correction Factor = 1.00043
2021/01/19 14:14:56.8 Spectrometer EOS_B0814: AVRG=20.
2021/01/19 14:14:56.8 CH4: Correction Factor = 0.998949
2021/01/19 14:14:56.8 Spectrometer EOS_B0735: AVRG=20.
2021/01/19 14:14:56.8 CH5: Correction Factor = 0.99829
2021/01/19 14:14:56.8 Spectrometer EOS_B0732: AVRG=20.
2021/01/19 14:14:56.8 CH6: Correction Factor = 0.999829
2021/01/19 14:14:56.8 Spectrometer EOS_B0808: AVRG=20.
2021/01/19 14:14:56.8 CH7: Correction Factor = 0.998245
2021/01/19 14:14:56.8 Spectrometer EOS_B0733: AVRG=20.
2021/01/19 14:14:56.8 CH8: Correction Factor = 1.00151
2021/01/19 14:14:56.8 Spectrometer EOS_B0817: AVRG=20.
2021/01/19 14:14:56.8 Sensor 1: Integration Time = 2.29 ms
2021/01/19 14:14:56.8 Sensor 2: Integration Time = 2.17 ms
2021/01/19 14:14:56.8 Sensor 3: Integration Time = 2.21 ms
2021/01/19 14:14:56.8 Sensor 4: Integration Time = 2.39 ms
2021/01/19 14:14:56.8 Sensor 5: Integration Time = 2.67 ms
2021/01/19 14:14:56.8 Sensor 6: Integration Time = 2.33 ms
2021/01/19 14:14:56.8 Sensor 7: Integration Time = 2.26 ms
2021/01/19 14:14:56.8 Sensor 8: Integration Time = 2.44 ms
2021/01/19 14:14:56.8 WaitControllerIdle(30000)
2021/01/19 14:14:57.0 GMC_HomeZ
2021/01/19 14:14:57.0 HomeDevice[0005]
2021/01/19 14:15:03.7 GMC_SetPosZ
2021/01/19 14:15:04.1 GMC_ServoHere
2021/01/19 14:15:04.4 Old State:9-ENGN_StateRunning_CheckingSensors New State:ENGN_StateRunning_PickingSensorsComplete
2021/01/19 14:15:04.4 StartShaker(1000)
2021/01/19 14:15:04.4 Integration Time: 2.29,2.17,2.21,2.39,2.67,2.33,2.26,2.44,
2021/01/19 14:15:04.5 Old State:10-ENGN_StateRunning_PickingSensorsComplete New State:ENGN_StateRunning_WaitGoingToSample
2021/01/19 14:15:04.5 Sample96_Q(0, 0, 0)
2021/01/19 14:15:04.5 
2021/01/19 14:15:04.5 START SCAN: 
2021/01/19 14:15:04.5    CycleTime (ms)    = 200
2021/01/19 14:15:04.5    CycleCount        = 300
2021/01/19 14:15:04.5    ScansToAverage    = 20
2021/01/19 14:15:04.5    FinishTime (s)    = 60
2021/01/19 14:15:04.5    Temperature       = 21.0
2021/01/19 14:15:04.5    Lamp Power        = 31
2021/01/19 14:15:04.5    Spec PCB temps    = 42.1, 41.6, 42.6, 42.3, 42.1, 41.8, 40.5, 40.8
2021/01/19 14:15:04.5 
2021/01/19 14:15:04.6 GMC_SetPosZ
2021/01/19 14:15:04.6 Old State:12-ENGN_StateRunning_WaitGoingToSample New State:ENGN_StateRunning_GoingToSample
2021/01/19 14:15:04.7 Plate 1 temperature = 21 C
2021/01/19 14:15:04.7 Pickups: 322, Warn: 90, Error: 100
2021/01/19 14:15:04.7 GMC_ServoHere
2021/01/19 14:15:04.9 GMC_SetPosXY
2021/01/19 14:15:07.1 GMC_SetPosZ
2021/01/19 14:15:10.2 Old State:13-ENGN_StateRunning_GoingToSample New State:ENGN_StateRunning_GoingToSampleComplete
2021/01/19 14:15:10.3 Old State:14-cycle=1, Elapsed Time:    0.0,  Ideal Time:    0.0,  Delay time =   0.00  New State:ENGN_StateRunning_WaitSampling
2021/01/19 14:15:10.3 Step started at: 14:15:04
2021/01/19 14:15:10.3 XYZ movement time  = 5583 ms + Scan start delay = 5578 ms
2021/01/19 14:15:10.4 Old State:15-ENGN_StateRunning_WaitSampling New State:ENGN_StateRunning_Sampling
2021/01/19 14:15:14.4 Shaker current speed = 994 rpm, set speed = 1000
2021/01/19 14:15:47.3 Temp: 2021/01/19 14:15:47, 21.0000, 23.8000
2021/01/19 14:16:10.0 
2021/01/19 14:16:10.0 STOP SCAN: 
2021/01/19 14:16:10.0    CycleTime (ms)    = 200
2021/01/19 14:16:10.0    CurrentCycle      = 300
2021/01/19 14:16:10.0    RunTime (ms)      = 59958
2021/01/19 14:16:10.0    AverageCycle (ms) = 199.9
2021/01/19 14:16:10.0    Temperature       = 21.0
2021/01/19 14:16:10.0    DATA OVERRUN %    = 0.00
2021/01/19 14:16:10.0    Spec PCB temps    = 42.0, 41.6, 42.5, 42.3, 42.1, 41.8, 40.5, 40.8
2021/01/19 14:16:10.0 
2021/01/19 14:16:10.1 Old State:16-cycle=300, Elapsed Time:   60.0,  Ideal Time:   59.8,  Delay time =   0.04  New State:ENGN_StateRunning_SamplingComplete
2021/01/19 14:16:10.2 Old State:18-ENGN_StateRunning_SamplingComplete New State:ENGN_StateRunning_WaitNewStep
2021/01/19 14:16:10.4 Old State:5-ENGN_StateRunning_WaitNewStep New State:ENGN_StateRunning_StartNewStep
2021/01/19 14:16:10.4 Spectrometer EOS_B0712: AVRG=20.
2021/01/19 14:16:10.4 Spectrometer EOS_B0734: AVRG=20.
2021/01/19 14:16:10.4 Spectrometer EOS_B0814: AVRG=20.
2021/01/19 14:16:10.4 Spectrometer EOS_B0735: AVRG=20.
2021/01/19 14:16:10.4 Spectrometer EOS_B0732: AVRG=20.
2021/01/19 14:16:10.4 Spectrometer EOS_B0808: AVRG=20.
2021/01/19 14:16:10.4 Spectrometer EOS_B0733: AVRG=20.
2021/01/19 14:16:10.4 Spectrometer EOS_B0817: AVRG=20.
2021/01/19 14:16:10.4 StartShaker(1000)
2021/01/19 14:16:10.5 Old State:6-ENGN_StateRunning_StartNewStep New State:ENGN_StateRunning_WaitGoingToSample
2021/01/19 14:16:10.5 Sample96_Q(0, 0, 1)
2021/01/19 14:16:10.6 
2021/01/19 14:16:10.6 START SCAN: 
2021/01/19 14:16:10.6    CycleTime (ms)    = 200
2021/01/19 14:16:10.6    CycleCount        = 1500
2021/01/19 14:16:10.6    ScansToAverage    = 20
2021/01/19 14:16:10.6    FinishTime (s)    = 300
2021/01/19 14:16:10.6    Temperature       = 21.0
2021/01/19 14:16:10.6    Lamp Power        = 31
2021/01/19 14:16:10.6    Spec PCB temps    = 42.1, 41.6, 42.5, 42.3, 42.1, 41.8, 40.5, 40.8
2021/01/19 14:16:10.6 
2021/01/19 14:16:10.7 Old State:12-ENGN_StateRunning_WaitGoingToSample New State:ENGN_StateRunning_GoingToSample
2021/01/19 14:16:10.7 GMC_SetPosZ
2021/01/19 14:16:10.7 Temp: 2021/01/19 14:16:10, 21.0000, 23.8000
2021/01/19 14:16:13.7 GMC_ServoHere
2021/01/19 14:16:13.9 GMC_SetPosXY
2021/01/19 14:16:14.2 GMC_SetPosZ
2021/01/19 14:16:17.5 Old State:13-ENGN_StateRunning_GoingToSample New State:ENGN_StateRunning_GoingToSampleComplete
2021/01/19 14:16:17.6 Old State:14-cycle=1, Elapsed Time:    0.0,  Ideal Time:    0.0,  Delay time =   0.00  New State:ENGN_StateRunning_WaitSampling
2021/01/19 14:16:17.6 Step started at: 14:16:10
2021/01/19 14:16:17.6 XYZ movement time  = 6903 ms + Scan start delay = 6857 ms
2021/01/19 14:16:17.8 Old State:15-ENGN_StateRunning_WaitSampling New State:ENGN_StateRunning_Sampling
2021/01/19 14:16:20.5 Shaker current speed = 994 rpm, set speed = 1000
2021/01/19 14:17:10.8 Temp: 2021/01/19 14:17:10, 21.0000, 23.8000
2021/01/19 14:18:11.0 Temp: 2021/01/19 14:18:11, 21.0000, 23.8000
2021/01/19 14:19:11.1 Temp: 2021/01/19 14:19:11, 21.0000, 23.7000
2021/01/19 14:20:11.3 Temp: 2021/01/19 14:20:11, 21.0000, 23.8000
2021/01/19 14:21:11.4 Temp: 2021/01/19 14:21:11, 21.0000, 23.8000
2021/01/19 14:21:17.4 
2021/01/19 14:21:17.4 STOP SCAN: 
2021/01/19 14:21:17.4    CycleTime (ms)    = 200
2021/01/19 14:21:17.4    CurrentCycle      = 1500
2021/01/19 14:21:17.4    RunTime (ms)      = 299958
2021/01/19 14:21:17.4    AverageCycle (ms) = 200.0
2021/01/19 14:21:17.4    Temperature       = 21.0
2021/01/19 14:21:17.4    DATA OVERRUN %    = 0.00
2021/01/19 14:21:17.4    Spec PCB temps    = 42.1, 41.6, 42.6, 42.3, 42.1, 41.8, 40.6, 40.9
2021/01/19 14:21:17.4 
2021/01/19 14:21:17.4 Old State:16-cycle=1500, Elapsed Time:  300.0,  Ideal Time:  299.8,  Delay time =   0.04  New State:ENGN_StateRunning_SamplingComplete
2021/01/19 14:21:17.6 Old State:18-ENGN_StateRunning_SamplingComplete New State:ENGN_StateRunning_WaitNewStep
2021/01/19 14:21:17.7 Old State:5-ENGN_StateRunning_WaitNewStep New State:ENGN_StateRunning_StartNewStep
2021/01/19 14:21:17.7 Spectrometer EOS_B0712: AVRG=20.
2021/01/19 14:21:17.7 Spectrometer EOS_B0734: AVRG=20.
2021/01/19 14:21:17.7 Spectrometer EOS_B0814: AVRG=20.
2021/01/19 14:21:17.7 Spectrometer EOS_B0735: AVRG=20.
2021/01/19 14:21:17.7 Spectrometer EOS_B0732: AVRG=20.
2021/01/19 14:21:17.7 Spectrometer EOS_B0808: AVRG=20.
2021/01/19 14:21:17.7 Spectrometer EOS_B0733: AVRG=20.
2021/01/19 14:21:17.7 Spectrometer EOS_B0817: AVRG=20.
2021/01/19 14:21:17.7 StartShaker(1000)
2021/01/19 14:21:17.8 Old State:6-ENGN_StateRunning_StartNewStep New State:ENGN_StateRunning_WaitGoingToSample
2021/01/19 14:21:17.8 Sample96_Q(0, 0, 2)
2021/01/19 14:21:17.8 
2021/01/19 14:21:17.8 START SCAN: 
2021/01/19 14:21:17.8    CycleTime (ms)    = 200
2021/01/19 14:21:17.8    CycleCount        = 150
2021/01/19 14:21:17.8    ScansToAverage    = 20
2021/01/19 14:21:17.8    FinishTime (s)    = 30
2021/01/19 14:21:17.8    Temperature       = 21.0
2021/01/19 14:21:17.8    Lamp Power        = 31
2021/01/19 14:21:17.8    Spec PCB temps    = 42.1, 41.6, 42.6, 42.3, 42.1, 41.8, 40.6, 40.9
2021/01/19 14:21:17.8 
2021/01/19 14:21:17.9 Old State:12-ENGN_StateRunning_WaitGoingToSample New State:ENGN_StateRunning_GoingToSample
2021/01/19 14:21:18.0 GMC_SetPosZ
2021/01/19 14:21:18.0 Temp: 2021/01/19 14:21:18, 21.0000, 23.8000
2021/01/19 14:21:21.1 GMC_ServoHere
2021/01/19 14:21:21.3 GMC_SetPosXY
2021/01/19 14:21:21.6 GMC_SetPosZ
2021/01/19 14:21:25.4 Old State:13-ENGN_StateRunning_GoingToSample New State:ENGN_StateRunning_GoingToSampleComplete
2021/01/19 14:21:25.5 Old State:14-cycle=1, Elapsed Time:    0.0,  Ideal Time:    0.0,  Delay time =   0.00  New State:ENGN_StateRunning_WaitSampling
2021/01/19 14:21:25.5 Step started at: 14:21:17
2021/01/19 14:21:25.5 XYZ movement time  = 7447 ms + Scan start delay = 7434 ms
2021/01/19 14:21:25.6 Old State:15-ENGN_StateRunning_WaitSampling New State:ENGN_StateRunning_Sampling
2021/01/19 14:21:27.8 Shaker current speed = 990 rpm, set speed = 1000
2021/01/19 14:21:55.2 
2021/01/19 14:21:55.2 STOP SCAN: 
2021/01/19 14:21:55.2    CycleTime (ms)    = 200
2021/01/19 14:21:55.2    CurrentCycle      = 150
2021/01/19 14:21:55.2    RunTime (ms)      = 29959
2021/01/19 14:21:55.2    AverageCycle (ms) = 199.7
2021/01/19 14:21:55.2    Temperature       = 21.0
2021/01/19 14:21:55.2    DATA OVERRUN %    = 0.00
2021/01/19 14:21:55.2    Spec PCB temps    = 42.1, 41.6, 42.6, 42.3, 42.1, 41.8, 40.6, 40.9
2021/01/19 14:21:55.2 
2021/01/19 14:21:55.3 Old State:16-cycle=150, Elapsed Time:   30.0,  Ideal Time:   29.8,  Delay time =   0.04  New State:ENGN_StateRunning_SamplingComplete
2021/01/19 14:21:55.4 Old State:18-ENGN_StateRunning_SamplingComplete New State:ENGN_StateRunning_WaitNewStep
2021/01/19 14:21:55.5 Old State:5-ENGN_StateRunning_WaitNewStep New State:ENGN_StateRunning_StartNewStep
2021/01/19 14:21:55.5 Spectrometer EOS_B0712: AVRG=20.
2021/01/19 14:21:55.5 Spectrometer EOS_B0734: AVRG=20.
2021/01/19 14:21:55.5 Spectrometer EOS_B0814: AVRG=20.
2021/01/19 14:21:55.5 Spectrometer EOS_B0735: AVRG=20.
2021/01/19 14:21:55.5 Spectrometer EOS_B0732: AVRG=20.
2021/01/19 14:21:55.5 Spectrometer EOS_B0808: AVRG=20.
2021/01/19 14:21:55.5 Spectrometer EOS_B0733: AVRG=20.
2021/01/19 14:21:55.5 Spectrometer EOS_B0817: AVRG=20.
2021/01/19 14:21:55.5 StartShaker(1000)
2021/01/19 14:21:55.6 Old State:6-ENGN_StateRunning_StartNewStep New State:ENGN_StateRunning_WaitGoingToSample
2021/01/19 14:21:55.6 Sample96_Q(0, 0, 3)
2021/01/19 14:21:55.7 
2021/01/19 14:21:55.7 START SCAN: 
2021/01/19 14:21:55.7    CycleTime (ms)    = 200
2021/01/19 14:21:55.7    CycleCount        = 600
2021/01/19 14:21:55.7    ScansToAverage    = 20
2021/01/19 14:21:55.7    FinishTime (s)    = 120
2021/01/19 14:21:55.7    Temperature       = 21.0
2021/01/19 14:21:55.7    Lamp Power        = 31
2021/01/19 14:21:55.7    Spec PCB temps    = 42.1, 41.6, 42.6, 42.3, 42.1, 41.8, 40.6, 40.9
2021/01/19 14:21:55.7 
2021/01/19 14:21:55.7 GMC_SetPosZ
2021/01/19 14:21:55.7 Old State:12-ENGN_StateRunning_WaitGoingToSample New State:ENGN_StateRunning_GoingToSample
2021/01/19 14:21:55.8 Temp: 2021/01/19 14:21:55, 21.0000, 23.8000
2021/01/19 14:21:59.2 GMC_ServoHere
2021/01/19 14:21:59.3 GMC_SetPosXY
2021/01/19 14:21:59.7 GMC_SetPosZ
2021/01/19 14:22:03.6 Old State:13-ENGN_StateRunning_GoingToSample New State:ENGN_StateRunning_GoingToSampleComplete
2021/01/19 14:22:03.7 Old State:14-cycle=1, Elapsed Time:    0.0,  Ideal Time:    0.0,  Delay time =   0.00  New State:ENGN_StateRunning_WaitSampling
2021/01/19 14:22:03.7 Step started at: 14:21:55
2021/01/19 14:22:03.7 XYZ movement time  = 7865 ms + Scan start delay = 7815 ms
2021/01/19 14:22:03.8 Old State:15-ENGN_StateRunning_WaitSampling New State:ENGN_StateRunning_Sampling
2021/01/19 14:22:05.6 Shaker current speed = 994 rpm, set speed = 1000
2021/01/19 14:22:55.9 Temp: 2021/01/19 14:22:55, 21.0000, 23.8000
2021/01/19 14:23:56.1 Temp: 2021/01/19 14:23:56, 21.0000, 23.8000
2021/01/19 14:24:03.4 
2021/01/19 14:24:03.4 STOP SCAN: 
2021/01/19 14:24:03.4    CycleTime (ms)    = 200
2021/01/19 14:24:03.4    CurrentCycle      = 600
2021/01/19 14:24:03.4    RunTime (ms)      = 119957
2021/01/19 14:24:03.4    AverageCycle (ms) = 199.9
2021/01/19 14:24:03.4    Temperature       = 21.0
2021/01/19 14:24:03.4    DATA OVERRUN %    = 0.00
2021/01/19 14:24:03.4    Spec PCB temps    = 42.1, 41.6, 42.6, 42.3, 42.1, 41.8, 40.6, 40.9
2021/01/19 14:24:03.4 
2021/01/19 14:24:03.5 Old State:16-cycle=600, Elapsed Time:  120.0,  Ideal Time:  119.8,  Delay time =   0.04  New State:ENGN_StateRunning_SamplingComplete
2021/01/19 14:24:03.6 Old State:18-ENGN_StateRunning_SamplingComplete New State:ENGN_StateRunning_WaitNewStep
2021/01/19 14:24:03.7 Old State:5-ENGN_StateRunning_WaitNewStep New State:ENGN_StateRunning_StartNewStep
2021/01/19 14:24:03.7 Spectrometer EOS_B0712: AVRG=20.
2021/01/19 14:24:03.8 Spectrometer EOS_B0734: AVRG=20.
2021/01/19 14:24:03.8 Spectrometer EOS_B0814: AVRG=20.
2021/01/19 14:24:03.8 Spectrometer EOS_B0735: AVRG=20.
2021/01/19 14:24:03.8 Spectrometer EOS_B0732: AVRG=20.
2021/01/19 14:24:03.8 Spectrometer EOS_B0808: AVRG=20.
2021/01/19 14:24:03.8 Spectrometer EOS_B0733: AVRG=20.
2021/01/19 14:24:03.8 Spectrometer EOS_B0817: AVRG=20.
2021/01/19 14:24:03.8 StartShaker(1000)
2021/01/19 14:24:03.9 Old State:6-ENGN_StateRunning_StartNewStep New State:ENGN_StateRunning_WaitGoingToSample
2021/01/19 14:24:03.9 Sample96_Q(0, 0, 5)
2021/01/19 14:24:03.9 
2021/01/19 14:24:03.9 START SCAN: 
2021/01/19 14:24:03.9    CycleTime (ms)    = 200
2021/01/19 14:24:03.9    CycleCount        = 250
2021/01/19 14:24:03.9    ScansToAverage    = 20
2021/01/19 14:24:03.9    FinishTime (s)    = 50
2021/01/19 14:24:03.9    Temperature       = 21.0
2021/01/19 14:24:03.9    Lamp Power        = 31
2021/01/19 14:24:03.9    Spec PCB temps    = 42.1, 41.6, 42.6, 42.3, 42.1, 41.8, 40.6, 40.9
2021/01/19 14:24:03.9 
2021/01/19 14:24:03.9 GMC_SetPosZ
2021/01/19 14:24:04.0 Old State:12-ENGN_StateRunning_WaitGoingToSample New State:ENGN_StateRunning_GoingToSample
2021/01/19 14:24:04.0 Temp: 2021/01/19 14:24:04, 21.0000, 23.8000
2021/01/19 14:24:07.7 GMC_ServoHere
2021/01/19 14:24:07.9 GMC_SetPosXY
2021/01/19 14:24:08.4 GMC_SetPosZ
2021/01/19 14:24:12.9 Old State:13-ENGN_StateRunning_GoingToSample New State:ENGN_StateRunning_GoingToSampleComplete
2021/01/19 14:24:13.0 Old State:14-cycle=1, Elapsed Time:    0.0,  Ideal Time:    0.0,  Delay time =   0.00  New State:ENGN_StateRunning_WaitSampling
2021/01/19 14:24:13.0 Step started at: 14:24:03
2021/01/19 14:24:13.1 XYZ movement time  = 8988 ms + Scan start delay = 8939 ms
2021/01/19 14:24:13.2 Old State:15-ENGN_StateRunning_WaitSampling New State:ENGN_StateRunning_Sampling
2021/01/19 14:24:13.8 Shaker current speed = 990 rpm, set speed = 1000
2021/01/19 14:25:02.8 
2021/01/19 14:25:02.8 STOP SCAN: 
2021/01/19 14:25:02.8    CycleTime (ms)    = 200
2021/01/19 14:25:02.8    CurrentCycle      = 250
2021/01/19 14:25:02.8    RunTime (ms)      = 49959
2021/01/19 14:25:02.8    AverageCycle (ms) = 199.8
2021/01/19 14:25:02.8    Temperature       = 21.0
2021/01/19 14:25:02.8    DATA OVERRUN %    = 0.00
2021/01/19 14:25:02.8    Spec PCB temps    = 42.1, 41.6, 42.6, 42.3, 42.1, 41.8, 40.6, 40.8
2021/01/19 14:25:02.8 
2021/01/19 14:25:02.8 Old State:16-cycle=250, Elapsed Time:   50.0,  Ideal Time:   49.8,  Delay time =   0.04  New State:ENGN_StateRunning_SamplingComplete
2021/01/19 14:25:03.0 Old State:18-ENGN_StateRunning_SamplingComplete New State:ENGN_StateRunning_WaitNewStep
2021/01/19 14:25:03.2 Old State:5-ENGN_StateRunning_WaitNewStep New State:ENGN_StateRunning_StartNewStep
2021/01/19 14:25:03.2 Spectrometer EOS_B0712: AVRG=20.
2021/01/19 14:25:03.2 Spectrometer EOS_B0734: AVRG=20.
2021/01/19 14:25:03.2 Spectrometer EOS_B0814: AVRG=20.
2021/01/19 14:25:03.2 Spectrometer EOS_B0735: AVRG=20.
2021/01/19 14:25:03.2 Spectrometer EOS_B0732: AVRG=20.
2021/01/19 14:25:03.2 Spectrometer EOS_B0808: AVRG=20.
2021/01/19 14:25:03.2 Spectrometer EOS_B0733: AVRG=20.
2021/01/19 14:25:03.2 Spectrometer EOS_B0817: AVRG=20.
2021/01/19 14:25:03.2 StartShaker(1000)
2021/01/19 14:25:03.3 Old State:6-ENGN_StateRunning_StartNewStep New State:ENGN_StateRunning_WaitGoingToSample
2021/01/19 14:25:03.3 Sample96_Q(0, 0, 3)
2021/01/19 14:25:03.3 
2021/01/19 14:25:03.3 START SCAN: 
2021/01/19 14:25:03.3    CycleTime (ms)    = 200
2021/01/19 14:25:03.3    CycleCount        = 600
2021/01/19 14:25:03.3    ScansToAverage    = 20
2021/01/19 14:25:03.3    FinishTime (s)    = 120
2021/01/19 14:25:03.3    Temperature       = 21.0
2021/01/19 14:25:03.3    Lamp Power        = 31
2021/01/19 14:25:03.3    Spec PCB temps    = 42.1, 41.6, 42.6, 42.3, 42.1, 41.8, 40.6, 40.8
2021/01/19 14:25:03.3 
2021/01/19 14:25:03.4 GMC_SetPosZ
2021/01/19 14:25:03.5 Old State:12-ENGN_StateRunning_WaitGoingToSample New State:ENGN_StateRunning_GoingToSample
2021/01/19 14:25:03.5 Temp: 2021/01/19 14:25:03, 21.0000, 23.8000
2021/01/19 14:25:07.8 GMC_ServoHere
2021/01/19 14:25:08.0 GMC_SetPosXY
2021/01/19 14:25:08.5 GMC_SetPosZ
2021/01/19 14:25:12.5 Old State:13-ENGN_StateRunning_GoingToSample New State:ENGN_StateRunning_GoingToSampleComplete
2021/01/19 14:25:12.6 Old State:14-cycle=1, Elapsed Time:    0.0,  Ideal Time:    0.0,  Delay time =   0.00  New State:ENGN_StateRunning_WaitSampling
2021/01/19 14:25:12.6 Step started at: 14:25:03
2021/01/19 14:25:12.6 XYZ movement time  = 9069 ms + Scan start delay = 9071 ms
2021/01/19 14:25:12.7 Old State:15-ENGN_StateRunning_WaitSampling New State:ENGN_StateRunning_Sampling
2021/01/19 14:25:13.2 Shaker current speed = 999 rpm, set speed = 1000
2021/01/19 14:26:03.6 Temp: 2021/01/19 14:26:03, 21.0000, 23.8000
2021/01/19 14:27:03.8 Temp: 2021/01/19 14:27:03, 21.0000, 23.8000
2021/01/19 14:27:12.4 
2021/01/19 14:27:12.4 STOP SCAN: 
2021/01/19 14:27:12.4    CycleTime (ms)    = 200
2021/01/19 14:27:12.4    CurrentCycle      = 600
2021/01/19 14:27:12.4    RunTime (ms)      = 119958
2021/01/19 14:27:12.4    AverageCycle (ms) = 199.9
2021/01/19 14:27:12.4    Temperature       = 21.0
2021/01/19 14:27:12.4    DATA OVERRUN %    = 0.00
2021/01/19 14:27:12.4    Spec PCB temps    = 42.1, 41.6, 42.6, 42.3, 42.1, 41.8, 40.6, 40.9
2021/01/19 14:27:12.4 
2021/01/19 14:27:12.4 Old State:16-cycle=600, Elapsed Time:  120.0,  Ideal Time:  119.8,  Delay time =   0.04  New State:ENGN_StateRunning_SamplingComplete
2021/01/19 14:27:12.5 Old State:18-ENGN_StateRunning_SamplingComplete New State:ENGN_StateRunning_WaitEjecting
2021/01/19 14:27:12.5 EJECT: To Tray, position=1,0
2021/01/19 14:27:12.5    EjectOffset = 10.0
2021/01/19 14:27:12.6 GMC_SetPosZ
2021/01/19 14:27:12.7 Old State:19-ENGN_StateRunning_WaitEjecting New State:ENGN_StateRunning_Ejecting
2021/01/19 14:27:16.3 GMC_ServoHere
2021/01/19 14:27:16.5 GMC_SetPosXY
2021/01/19 14:27:18.7 GMC_SetPosZ
2021/01/19 14:27:22.8 GMC_Eject
2021/01/19 14:27:22.8 EjectDone => EjectEject
2021/01/19 14:27:22.8 SetEjectorPosn(0)
2021/01/19 14:27:26.0 GMC_HomeZ
2021/01/19 14:27:26.0 HomeDevice[0005]
2021/01/19 14:27:31.6 GMC_SetPosZ
2021/01/19 14:27:31.9 GMC_ServoHere
2021/01/19 14:27:32.1 GMC_SetPosZ
2021/01/19 14:27:32.3 GMC_ServoHere
2021/01/19 14:27:32.5 GMC_SetPosXY
2021/01/19 14:27:34.7 GMC_SetPosZ
2021/01/19 14:27:35.0 Old State:20-ENGN_StateRunning_Ejecting New State:ENGN_StateRunning_EjectingComplete
2021/01/19 14:27:35.1 Old State:21-ENGN_StateRunning_EjectingComplete New State:ENGN_StateRunning_Finished
2021/01/19 14:27:35.1 StopShaker
2021/01/19 14:27:35.3 Old State:22-ENGN_StateRunning_Finished New State:ENGN_State_SystemReady
2021/01/19 14:27:35.4 Temp: 2021/01/19 14:27:35, 21.0000, 23.8000
2021/01/19 14:27:35.4 Temperature monitor DISABLE
2021/01/19 14:27:35.4 
2021/01/19 14:27:35.4 ASSAY SUMMARY:
2021/01/19 14:27:35.4 ==============
2021/01/19 14:27:35.4   Step #   Duration   Cycles  Overrun %
2021/01/19 14:27:35.4        1      59     300    0.00
2021/01/19 14:27:35.4        2     299    1500    0.00
2021/01/19 14:27:35.4        3      29     150    0.00
2021/01/19 14:27:35.4        4     119     600    0.00
2021/01/19 14:27:35.4        5      49     250    0.00
2021/01/19 14:27:35.4        6     119     600    0.00
2021/01/19 14:27:35.4        7      59     300    0.00
2021/01/19 14:27:35.4        8     299    1500    0.00
2021/01/19 14:27:35.4        9      29     150    0.00
2021/01/19 14:27:35.4       10     119     600    0.00
2021/01/19 14:27:35.4       11      49     250    0.00
2021/01/19 14:27:35.4       12     119     600    0.00
2021/01/19 14:27:35.4 -------------------------------------------
2021/01/19 14:27:35.4    Total    1359    6800    0.00
2021/01/19 14:27:35.4 Temperature monitor DISABLE
2021/01/19 14:27:35.5 DFIO: Start writing C:\Users\Labuser\Desktop\Kristjan Sigmar\20210119 STAB-0005 STAB-0006 t3\210119_009.frd
2021/01/19 14:27:35.5 DFIO: Finished writing C:\Users\Labuser\Desktop\Kristjan Sigmar\20210119 STAB-0005 STAB-0006 t3\210119_009.frd
2021/01/19 14:27:35.5 DFIO: Start writing C:\Users\Labuser\Desktop\Kristjan Sigmar\20210119 STAB-0005 STAB-0006 t3\210119_010.frd
2021/01/19 14:27:35.5 DFIO: Finished writing C:\Users\Labuser\Desktop\Kristjan Sigmar\20210119 STAB-0005 STAB-0006 t3\210119_010.frd
2021/01/19 14:27:35.5 DFIO: Start writing C:\Users\Labuser\Desktop\Kristjan Sigmar\20210119 STAB-0005 STAB-0006 t3\210119_011.frd
2021/01/19 14:27:35.5 DFIO: Finished writing C:\Users\Labuser\Desktop\Kristjan Sigmar\20210119 STAB-0005 STAB-0006 t3\210119_011.frd
2021/01/19 14:27:35.5 DFIO: Start writing C:\Users\Labuser\Desktop\Kristjan Sigmar\20210119 STAB-0005 STAB-0006 t3\210119_012.frd
2021/01/19 14:27:35.5 DFIO: Finished writing C:\Users\Labuser\Desktop\Kristjan Sigmar\20210119 STAB-0005 STAB-0006 t3\210119_012.frd
2021/01/19 14:27:35.5 DFIO: Start writing C:\Users\Labuser\Desktop\Kristjan Sigmar\20210119 STAB-0005 STAB-0006 t3\210119_013.frd
2021/01/19 14:27:35.5 DFIO: Finished writing C:\Users\Labuser\Desktop\Kristjan Sigmar\20210119 STAB-0005 STAB-0006 t3\210119_013.frd
2021/01/19 14:27:35.5 DFIO: Start writing C:\Users\Labuser\Desktop\Kristjan Sigmar\20210119 STAB-0005 STAB-0006 t3\210119_014.frd
2021/01/19 14:27:35.6 DFIO: Finished writing C:\Users\Labuser\Desktop\Kristjan Sigmar\20210119 STAB-0005 STAB-0006 t3\210119_014.frd
2021/01/19 14:27:35.6 DFIO: Start writing C:\Users\Labuser\Desktop\Kristjan Sigmar\20210119 STAB-0005 STAB-0006 t3\210119_015.frd
2021/01/19 14:27:35.6 DFIO: Finished writing C:\Users\Labuser\Desktop\Kristjan Sigmar\20210119 STAB-0005 STAB-0006 t3\210119_015.frd
2021/01/19 14:27:35.6 DFIO: Start writing C:\Users\Labuser\Desktop\Kristjan Sigmar\20210119 STAB-0005 STAB-0006 t3\210119_016.frd
2021/01/19 14:27:35.6 DFIO: Finished writing C:\Users\Labuser\Desktop\Kristjan Sigmar\20210119 STAB-0005 STAB-0006 t3\210119_016.frd
2021/01/19 14:27:35.6 DFIO: Finished writing C:\Users\Labuser\Desktop\Kristjan Sigmar\20210119 STAB-0005 STAB-0006 t3\210119_Manifest.fmx
2021/01/19 16:49:06.1 Device Connected:  USB VID_058F&PID_6387 061D1979
2021/01/19 16:49:19.3 Closing application
2021/01/19 16:49:19.3 StopDevice[0006]
2021/01/19 16:49:19.3 StopShaker
2021/01/19 16:49:19.4 StopDevice[0006]
2021/01/19 16:49:19.7 Detected memory leaks!
2021/01/19 16:49:19.7 Dumping objects ->
2021/01/19 16:49:19.7 {8127450} 2021/01/19 16:49:19.7 normal block at 0x0E327630, 8 bytes long.
2021/01/19 16:49:19.7  Data: <    8   > F0 08 00 00 38 0E A2 08 
2021/01/19 16:49:19.7 {8127449} 2021/01/19 16:49:19.7 normal block at 0x08A20E38, 516 bytes long.
2021/01/19 16:49:19.7  Data: < fa  Gc t b   _ > C8 66 61 02 12 47 63 01 74 C7 62 01 B4 85 5F 01 
2021/01/19 16:49:19.7 Object dump complete.
2021/01/19 16:49:19.8 MultiWellPlate: DLL_PROCESS_DETACH.
2021/01/19 16:49:19.8 DataSecurity: DLL_PROCESS_DETACH.
2021/01/19 16:49:19.9 Detected memory leaks!
2021/01/19 16:49:19.9 Dumping objects ->
2021/01/19 16:49:19.9 {2741} 2021/01/19 16:49:19.9 normal block at 0x02998348, 64 bytes long.
2021/01/19 16:49:19.9  Data: <HA  (4  (9  (>  > 48 41 99 02 28 34 99 02 28 39 99 02 28 3E 99 02 
2021/01/19 16:49:19.9 {2740} 2021/01/19 16:49:19.9 normal block at 0x0297EE78, 32 bytes long.
2021/01/19 16:49:19.9  Data: <HA  (>  h:   =  > 48 41 99 02 28 3E 99 02 68 3A 99 02 88 3D 99 02 
2021/01/19 16:49:19.9 {2739} 2021/01/19 16:49:19.9 normal block at 0x0297E2E8, 32 bytes long.
2021/01/19 16:49:19.9  Data: <(4  (9  (>  H2  > 28 34 99 02 28 39 99 02 28 3E 99 02 48 32 99 02 
2021/01/19 16:49:19.9 {2738} 2021/01/19 16:49:19.9 normal block at 0x0297E298, 32 bytes long.
2021/01/19 16:49:19.9  Data: <(4  (9  (>  h:  > 28 34 99 02 28 39 99 02 28 3E 99 02 68 3A 99 02 
2021/01/19 16:49:19.9 {2736} 2021/01/19 16:49:19.9 normal block at 0x029D5A08, 12 bytes long.
2021/01/19 16:49:19.9  Data: <8<          > 38 3C 9C 02 00 00 00 00 01 00 00 00 
2021/01/19 16:49:19.9 {2734} 2021/01/19 16:49:19.9 normal block at 0x029C3C78, 20 bytes long.
2021/01/19 16:49:19.9  Data: <                > 07 00 00 00 98 E2 97 02 00 00 00 00 08 00 00 00 
2021/01/19 16:49:19.9 {2733} 2021/01/19 16:49:19.9 normal block at 0x029C3C38, 16 bytes long.
2021/01/19 16:49:19.9  Data: <    x<          > 01 00 00 00 78 3C 9C 02 00 00 00 00 00 00 00 00 
2021/01/19 16:49:19.9 {2732} 2021/01/19 16:49:19.9 normal block at 0x02997E78, 64 bytes long.
2021/01/19 16:49:19.9  Data: <            P   > B0 BF 94 02 C0 C1 94 02 90 C1 94 02 50 BF 94 02 
2021/01/19 16:49:19.9 {2731} 2021/01/19 16:49:19.9 normal block at 0x0294BFB0, 4 bytes long.
2021/01/19 16:49:19.9  Data: <0  h> 30 E7 0D 68 
2021/01/19 16:49:19.9 {2730} 2021/01/19 16:49:19.9 normal block at 0x02997BD8, 64 bytes long.
2021/01/19 16:49:19.9  Data: <     Z          > 00 00 00 00 08 5A 9D 02 00 00 00 00 00 00 00 00 
2021/01/19 16:49:19.9 {2729} 2021/01/19 16:49:19.9 normal block at 0x02941238, 96 bytes long.
2021/01/19 16:49:19.9  Data: < {     h   h    > D8 7B 99 02 90 07 0E 68 D0 07 0E 68 08 00 00 00 
2021/01/19 16:49:19.9 {2728} 2021/01/19 16:49:19.9 normal block at 0x029D5B58, 12 bytes long.
2021/01/19 16:49:19.9  Data: < ;          > B8 3B 9C 02 00 00 00 00 01 00 00 00 
2021/01/19 16:49:19.9 {2726} 2021/01/19 16:49:19.9 normal block at 0x029C3BF8, 20 bytes long.
2021/01/19 16:49:19.9  Data: <                > 06 00 00 00 E8 E2 97 02 00 00 00 00 08 00 00 00 
2021/01/19 16:49:19.9 {2725} 2021/01/19 16:49:19.9 normal block at 0x029C3BB8, 16 bytes long.
2021/01/19 16:49:19.9  Data: <     ;          > 01 00 00 00 F8 3B 9C 02 00 00 00 00 00 00 00 00 
2021/01/19 16:49:19.9 {2724} 2021/01/19 16:49:19.9 normal block at 0x0294C1C0, 4 bytes long.
2021/01/19 16:49:19.9  Data: <   h> F0 8E 0A 68 
2021/01/19 16:49:19.9 {2723} 2021/01/19 16:49:19.9 normal block at 0x02998188, 64 bytes long.
2021/01/19 16:49:19.9  Data: <    X[          > 00 00 00 00 58 5B 9D 02 00 00 00 00 00 00 00 00 
2021/01/19 16:49:19.9 {2722} 2021/01/19 16:49:19.9 normal block at 0x029413E8, 96 bytes long.
2021/01/19 16:49:19.9  Data: <       h   h    > 88 81 99 02 90 07 0E 68 D0 07 0E 68 08 00 00 00 
2021/01/19 16:49:19.9 {2721} 2021/01/19 16:49:19.9 normal block at 0x029D5C70, 12 bytes long.
2021/01/19 16:49:19.9  Data: < .          > B8 2E 9C 02 00 00 00 00 01 00 00 00 
2021/01/19 16:49:19.9 {2719} 2021/01/19 16:49:19.9 normal block at 0x029C3578, 20 bytes long.
2021/01/19 16:49:19.9  Data: <    x           > 04 00 00 00 78 EE 97 02 00 00 00 00 08 00 00 00 
2021/01/19 16:49:19.9 {2718} 2021/01/19 16:49:19.9 normal block at 0x029C2EB8, 16 bytes long.
2021/01/19 16:49:19.9  Data: <    x5          > 01 00 00 00 78 35 9C 02 00 00 00 00 01 00 00 00 
2021/01/19 16:49:19.9 {2717} 2021/01/19 16:49:19.9 normal block at 0x0294C190, 4 bytes long.
2021/01/19 16:49:19.9  Data: < ! h> E0 21 08 68 
2021/01/19 16:49:19.9 {2716} 2021/01/19 16:49:19.9 normal block at 0x02997F58, 64 bytes long.
2021/01/19 16:49:19.9  Data: <    p\          > 00 00 00 00 70 5C 9D 02 00 00 00 00 00 00 00 00 
2021/01/19 16:49:19.9 {2715} 2021/01/19 16:49:19.9 normal block at 0x02940F68, 96 bytes long.
2021/01/19 16:49:19.9  Data: <X      h   h    > 58 7F 99 02 90 07 0E 68 D0 07 0E 68 08 00 00 00 
2021/01/19 16:49:19.9 {2714} 2021/01/19 16:49:19.9 normal block at 0x029D5CA8, 12 bytes long.
2021/01/19 16:49:19.9  Data: <x4          > 78 34 9C 02 00 00 00 00 01 00 00 00 
2021/01/19 16:49:19.9 {2712} 2021/01/19 16:49:19.9 normal block at 0x029C34F8, 20 bytes long.
2021/01/19 16:49:19.9  Data: <    H           > 08 00 00 00 48 83 99 02 00 00 00 00 10 00 00 00 
2021/01/19 16:49:19.9 {2711} 2021/01/19 16:49:19.9 normal block at 0x029C3478, 16 bytes long.
2021/01/19 16:49:19.9  Data: <     4          > 01 00 00 00 F8 34 9C 02 00 00 00 00 00 00 00 00 
2021/01/19 16:49:19.9 {2710} 2021/01/19 16:49:19.9 normal block at 0x0294BF50, 4 bytes long.
2021/01/19 16:49:19.9  Data: <   h> B0 F1 0A 68 
2021/01/19 16:49:19.9 {2709} 2021/01/19 16:49:19.9 normal block at 0x02997D98, 64 bytes long.
2021/01/19 16:49:19.9  Data: <     \          > 00 00 00 00 A8 5C 9D 02 00 00 00 00 00 00 00 00 
2021/01/19 16:49:19.9 {2708} 2021/01/19 16:49:19.9 normal block at 0x029409C8, 96 bytes long.
2021/01/19 16:49:19.9  Data: < }     h   h    > 98 7D 99 02 90 07 0E 68 D0 07 0E 68 08 00 00 00 
2021/01/19 16:49:19.9 {2659} 2021/01/19 16:49:19.9 normal block at 0x02993F68, 112 bytes long.
2021/01/19 16:49:19.9  Data: <  ;h  ;h  Oh  Oh> 14 9F 3B 68 1C 9F 3B 68 94 85 4F 68 CC 85 4F 68 
2021/01/19 16:49:19.9 {2565} 2021/01/19 16:49:19.9 normal block at 0x029D3C10, 12 bytes long.
2021/01/19 16:49:19.9  Data: < /      /   > F8 2F 9C 02 00 00 00 00 2F 03 00 00 
2021/01/19 16:49:19.9 {2564} 2021/01/19 16:49:19.9 normal block at 0x029C3638, 16 bytes long.
2021/01/19 16:49:19.9  Data: < 2              > E8 32 99 02 00 00 00 00 00 00 00 00 00 00 00 00 
2021/01/19 16:49:19.9 {2563} 2021/01/19 16:49:19.9 normal block at 0x029C2E78, 20 bytes long.
2021/01/19 16:49:19.9  Data: <    86          > 01 00 00 00 38 36 9C 02 00 00 00 00 04 00 00 00 
2021/01/19 16:49:19.9 {2562} 2021/01/19 16:49:19.9 normal block at 0x029C2FF8, 16 bytes long.
2021/01/19 16:49:19.9  Data: </   x.          > 2F 03 00 00 78 2E 9C 02 00 00 00 00 00 00 00 00 
2021/01/19 16:49:19.9 {2561} 2021/01/19 16:49:19.9 normal block at 0x029D3BA0, 12 bytes long.
2021/01/19 16:49:19.9  Data: <x3      +   > 78 33 9C 02 00 00 00 00 2B 03 00 00 
2021/01/19 16:49:19.9 {2560} 2021/01/19 16:49:19.9 normal block at 0x029C33B8, 16 bytes long.
2021/01/19 16:49:19.9  Data: < 2              > E8 32 99 02 00 00 00 00 00 00 00 00 00 00 00 00 
2021/01/19 16:49:19.9 {2559} 2021/01/19 16:49:19.9 normal block at 0x029C2FB8, 20 bytes long.
2021/01/19 16:49:19.9  Data: <     3          > 01 00 00 00 B8 33 9C 02 00 00 00 00 04 00 00 00 
2021/01/19 16:49:19.9 {2558} 2021/01/19 16:49:19.9 normal block at 0x029C3378, 16 bytes long.
2021/01/19 16:49:19.9  Data: <+    /          > 2B 03 00 00 B8 2F 9C 02 00 00 00 00 00 00 00 00 
2021/01/19 16:49:19.9 {2557} 2021/01/19 16:49:19.9 normal block at 0x029D4070, 12 bytes long.
2021/01/19 16:49:19.9  Data: < 2      ,   > B8 32 9C 02 00 00 00 00 2C 03 00 00 
2021/01/19 16:49:19.9 {2556} 2021/01/19 16:49:19.9 normal block at 0x029C3438, 16 bytes long.
2021/01/19 16:49:19.9  Data: < 2              > E8 32 99 02 00 00 00 00 00 00 00 00 00 00 00 00 
2021/01/19 16:49:19.9 {2555} 2021/01/19 16:49:19.9 normal block at 0x029C30B8, 20 bytes long.
2021/01/19 16:49:19.9  Data: <    84          > 01 00 00 00 38 34 9C 02 00 00 00 00 04 00 00 00 
2021/01/19 16:49:19.9 {2554} 2021/01/19 16:49:19.9 normal block at 0x029C32B8, 16 bytes long.
2021/01/19 16:49:19.9  Data: <,    0          > 2C 03 00 00 B8 30 9C 02 00 00 00 00 00 00 00 00 
2021/01/19 16:49:19.9 {2552} 2021/01/19 16:49:19.9 normal block at 0x0294BEF0, 4 bytes long.
2021/01/19 16:49:19.9  Data: <   h> D0 8B 10 68 
2021/01/19 16:49:19.9 {2551} 2021/01/19 16:49:19.9 normal block at 0x02997EE8, 64 bytes long.
2021/01/19 16:49:19.9  Data: <             ;  > 00 00 00 00 00 00 00 00 00 00 00 00 A0 3B 9D 02 
2021/01/19 16:49:19.9 {2550} 2021/01/19 16:49:19.9 normal block at 0x029408A8, 96 bytes long.
2021/01/19 16:49:19.9  Data: < ~     h   h    > E8 7E 99 02 90 07 0E 68 D0 07 0E 68 08 00 00 00 
2021/01/19 16:49:19.9 {2549} 2021/01/19 16:49:19.9 normal block at 0x029D40A8, 12 bytes long.
2021/01/19 16:49:19.9  Data: <x0      /   > 78 30 9C 02 00 00 00 00 2F 03 00 00 
2021/01/19 16:49:19.9 {2548} 2021/01/19 16:49:19.9 normal block at 0x029C33F8, 16 bytes long.
2021/01/19 16:49:19.9  Data: < 2              > E8 32 99 02 00 00 00 00 00 00 00 00 00 00 00 00 
2021/01/19 16:49:19.9 {2547} 2021/01/19 16:49:19.9 normal block at 0x029C34B8, 20 bytes long.
2021/01/19 16:49:19.9  Data: <     3          > 01 00 00 00 F8 33 9C 02 00 00 00 00 04 00 00 00 
2021/01/19 16:49:19.9 {2546} 2021/01/19 16:49:19.9 normal block at 0x029C3078, 16 bytes long.
2021/01/19 16:49:19.9  Data: </    4          > 2F 03 00 00 B8 34 9C 02 00 00 00 00 00 00 00 00 
2021/01/19 16:49:19.9 {2545} 2021/01/19 16:49:19.9 normal block at 0x029D41F8, 12 bytes long.
2021/01/19 16:49:19.9  Data: <85      )   > 38 35 9C 02 00 00 00 00 29 03 00 00 
2021/01/19 16:49:19.9 {2544} 2021/01/19 16:49:19.9 normal block at 0x029C2F78, 16 bytes long.
2021/01/19 16:49:19.9  Data: < 2              > E8 32 99 02 00 00 00 00 00 00 00 00 00 00 00 00 
2021/01/19 16:49:19.9 {2543} 2021/01/19 16:49:19.9 normal block at 0x029C35B8, 20 bytes long.
2021/01/19 16:49:19.9  Data: <    x/          > 01 00 00 00 78 2F 9C 02 00 00 00 00 04 00 00 00 
2021/01/19 16:49:19.9 {2542} 2021/01/19 16:49:19.9 normal block at 0x029C3538, 16 bytes long.
2021/01/19 16:49:19.9  Data: <)    5          > 29 03 00 00 B8 35 9C 02 00 00 00 00 00 00 00 00 
2021/01/19 16:49:19.9 {2541} 2021/01/19 16:49:19.9 normal block at 0x0294C130, 4 bytes long.
2021/01/19 16:49:19.9  Data: <   h> 10 BA 09 68 
2021/01/19 16:49:19.9 {2540} 2021/01/19 16:49:19.9 normal block at 0x02997E08, 64 bytes long.
2021/01/19 16:49:19.9  Data: <     A          > 00 00 00 00 F8 41 9D 02 00 00 00 00 00 00 00 00 
2021/01/19 16:49:19.9 {2539} 2021/01/19 16:49:19.9 normal block at 0x02940C98, 96 bytes long.
2021/01/19 16:49:19.9  Data: < ~     h   h    > 08 7E 99 02 90 07 0E 68 D0 07 0E 68 08 00 00 00 
2021/01/19 16:49:19.9 {2538} 2021/01/19 16:49:19.9 normal block at 0x029D3E08, 12 bytes long.
2021/01/19 16:49:19.9  Data: <83      .   > 38 33 9C 02 00 00 00 00 2E 03 00 00 
2021/01/19 16:49:19.9 {2537} 2021/01/19 16:49:19.9 normal block at 0x029C2F38, 16 bytes long.
2021/01/19 16:49:19.9  Data: < 2              > E8 32 99 02 00 00 00 00 00 00 00 00 00 00 00 00 
2021/01/19 16:49:19.9 {2536} 2021/01/19 16:49:19.9 normal block at 0x029C31B8, 20 bytes long.
2021/01/19 16:49:19.9  Data: <    8/          > 01 00 00 00 38 2F 9C 02 00 00 00 00 04 00 00 00 
2021/01/19 16:49:19.9 {2535} 2021/01/19 16:49:19.9 normal block at 0x029C3338, 16 bytes long.
2021/01/19 16:49:19.9  Data: <.    1          > 2E 03 00 00 B8 31 9C 02 00 00 00 00 00 00 00 00 
2021/01/19 16:49:19.9 {2534} 2021/01/19 16:49:19.9 normal block at 0x029D3F58, 12 bytes long.
2021/01/19 16:49:19.9  Data: < y      -   > F0 79 95 02 00 00 00 00 2D 03 00 00 
2021/01/19 16:49:19.9 {2533} 2021/01/19 16:49:19.9 normal block at 0x029C2EF8, 16 bytes long.
2021/01/19 16:49:19.9  Data: < 2              > E8 32 99 02 00 00 00 00 00 00 00 00 00 00 00 00 
2021/01/19 16:49:19.9 {2532} 2021/01/19 16:49:19.9 normal block at 0x029C32F8, 20 bytes long.
2021/01/19 16:49:19.9  Data: <     .          > 01 00 00 00 F8 2E 9C 02 00 00 00 00 04 00 00 00 
2021/01/19 16:49:19.9 {2531} 2021/01/19 16:49:19.9 normal block at 0x029579F0, 16 bytes long.
2021/01/19 16:49:19.9  Data: <-    2          > 2D 03 00 00 F8 32 9C 02 00 00 00 00 00 00 00 00 
2021/01/19 16:49:19.9 {2530} 2021/01/19 16:49:19.9 normal block at 0x0294C100, 4 bytes long.
2021/01/19 16:49:19.9  Data: <   h> C0 07 0C 68 
2021/01/19 16:49:19.9 {2529} 2021/01/19 16:49:19.9 normal block at 0x02997D28, 64 bytes long.
2021/01/19 16:49:19.9  Data: <                > 00 00 00 00 00 00 00 00 00 00 00 00 00 00 00 00 
2021/01/19 16:49:19.9 {2528} 2021/01/19 16:49:19.9 normal block at 0x02941508, 96 bytes long.
2021/01/19 16:49:19.9  Data: <(}     h   h    > 28 7D 99 02 90 07 0E 68 D0 07 0E 68 08 00 00 00 
2021/01/19 16:49:19.9 {2527} 2021/01/19 16:49:19.9 normal block at 0x029A7128, 108 bytes long.
2021/01/19 16:49:19.9  Data: </       `  h   h> 2F 03 00 00 01 00 00 00 60 E6 14 68 A0 E2 14 68 
2021/01/19 16:49:19.9 {2526} 2021/01/19 16:49:19.9 normal block at 0x029A7420, 108 bytes long.
2021/01/19 16:49:19.9  Data: <+       @  h0  h> 2B 03 00 00 01 00 00 00 40 E1 14 68 30 DA 14 68 
2021/01/19 16:49:19.9 {2525} 2021/01/19 16:49:19.9 normal block at 0x029A6710, 108 bytes long.
2021/01/19 16:49:19.9  Data: <,       @  h0  h> 2C 03 00 00 01 00 00 00 40 E1 14 68 30 DA 14 68 
2021/01/19 16:49:19.9 {2524} 2021/01/19 16:49:19.9 normal block at 0x029D3CB8, 9 bytes long.
2021/01/19 16:49:19.9  Data: <GOST-MAC > 47 4F 53 54 2D 4D 41 43 00 
2021/01/19 16:49:19.9 {2523} 2021/01/19 16:49:19.9 normal block at 0x02957770, 18 bytes long.
2021/01/19 16:49:19.9  Data: <GOST 28147-89 MA> 47 4F 53 54 20 32 38 31 34 37 2D 38 39 20 4D 41 
2021/01/19 16:49:19.9 {2522} 2021/01/19 16:49:19.9 normal block at 0x029A6A08, 108 bytes long.
2021/01/19 16:49:19.9  Data: </   /        <  > 2F 03 00 00 2F 03 00 00 06 00 00 00 B8 3C 9D 02 
2021/01/19 16:49:19.9 {2521} 2021/01/19 16:49:19.9 normal block at 0x029D3DD0, 9 bytes long.
2021/01/19 16:49:19.9  Data: <GOST2001 > 47 4F 53 54 32 30 30 31 00 
2021/01/19 16:49:19.9 {2520} 2021/01/19 16:49:19.9 normal block at 0x02957730, 18 bytes long.
2021/01/19 16:49:19.9  Data: <GOST R 34.10-200> 47 4F 53 54 20 52 20 33 34 2E 31 30 2D 32 30 30 
2021/01/19 16:49:19.9 {2519} 2021/01/19 16:49:19.9 normal block at 0x029A74B8, 108 bytes long.
2021/01/19 16:49:19.9  Data: <+   +        =  > 2B 03 00 00 2B 03 00 00 06 00 00 00 D0 3D 9D 02 
2021/01/19 16:49:19.9 {2518} 2021/01/19 16:49:19.9 normal block at 0x029D3D60, 7 bytes long.
2021/01/19 16:49:19.9  Data: <GOST94 > 47 4F 53 54 39 34 00 
2021/01/19 16:49:19.9 {2517} 2021/01/19 16:49:19.9 normal block at 0x02957EF0, 16 bytes long.
2021/01/19 16:49:19.9  Data: <GOST R 34.10-94 > 47 4F 53 54 20 52 20 33 34 2E 31 30 2D 39 34 00 
2021/01/19 16:49:19.9 {2516} 2021/01/19 16:49:19.9 normal block at 0x029A6C68, 108 bytes long.
2021/01/19 16:49:19.9  Data: <,   ,       `=  > 2C 03 00 00 2C 03 00 00 06 00 00 00 60 3D 9D 02 
2021/01/19 16:49:19.9 {2515} 2021/01/19 16:49:19.9 normal block at 0x029932E8, 112 bytes long.
2021/01/19 16:49:19.9  Data: <  ;h  ;h        > 0C A9 3B 68 14 A9 3B 68 00 00 00 00 00 00 00 00 
2021/01/19 16:49:19.9 {2514} 2021/01/19 16:49:19.9 normal block at 0x02993EC8, 112 bytes long.
2021/01/19 16:49:19.9  Data: <L ;h0 Qh        > 4C 97 3B 68 30 99 51 68 00 00 00 00 00 00 00 00 
2021/01/19 16:49:19.9 {2491} 2021/01/19 16:49:19.9 normal block at 0x029939C8, 112 bytes long.
2021/01/19 16:49:19.9  Data: <  ;h  ;hd Oh  Oh> C4 95 3B 68 CC 95 3B 68 64 82 4F 68 9C 82 4F 68 
2021/01/19 16:49:19.9 {2467} 2021/01/19 16:49:19.9 normal block at 0x02993D88, 112 bytes long.
2021/01/19 16:49:19.9  Data: <  ;h, ;hL Oh  Oh> 20 91 3B 68 2C 91 3B 68 4C 80 4F 68 C0 80 4F 68 
2021/01/19 16:49:19.9 {2455} 2021/01/19 16:49:19.9 normal block at 0x02993248, 112 bytes long.
2021/01/19 16:49:19.9  Data: <  ;h  ;h ~Oh ~Oh> 0C 8F 3B 68 14 8F 3B 68 AC 7E 4F 68 E4 7E 4F 68 
2021/01/19 16:49:19.9 {2429} 2021/01/19 16:49:19.9 normal block at 0x02993A68, 112 bytes long.
2021/01/19 16:49:19.9  Data: <  ;h  ;h {Oh    > A4 89 3B 68 AC 89 3B 68 1C 7B 4F 68 00 00 00 00 
2021/01/19 16:49:19.9 {2409} 2021/01/19 16:49:19.9 normal block at 0x02993E28, 112 bytes long.
2021/01/19 16:49:19.9  Data: <  ;h  ;h,}Ohd}Oh> 18 8D 3B 68 20 8D 3B 68 2C 7D 4F 68 64 7D 4F 68 
2021/01/19 16:49:19.9 {2394} 2021/01/19 16:49:19.9 normal block at 0x02993928, 112 bytes long.
2021/01/19 16:49:19.9  Data: <  ;h  ;h\yOh yOh> 8C 85 3B 68 94 85 3B 68 5C 79 4F 68 94 79 4F 68 
2021/01/19 16:49:19.9 {2368} 2021/01/19 16:49:19.9 normal block at 0x02993428, 112 bytes long.
2021/01/19 16:49:19.9  Data: <t ;hx ;h wOh xOh> 74 83 3B 68 78 83 3B 68 CC 77 4F 68 04 78 4F 68 
2021/01/19 16:49:19.9 {2349} 2021/01/19 16:49:19.9 normal block at 0x02994148, 112 bytes long.
2021/01/19 16:49:19.9  Data: <  ;h  ;h\vOh    > A4 80 3B 68 B0 80 3B 68 5C 76 4F 68 00 00 00 00 
2021/01/19 16:49:19.9 {2344} 2021/01/19 16:49:19.9 normal block at 0x0294C0D0, 4 bytes long.
2021/01/19 16:49:19.9  Data: <   h> F0 05 0E 68 
2021/01/19 16:49:19.9 {2342} 2021/01/19 16:49:19.9 normal block at 0x02957BF0, 20 bytes long.
2021/01/19 16:49:19.9  Data: <    x~          > 08 00 00 00 78 7E 99 02 00 00 00 00 10 00 00 00 
2021/01/19 16:49:19.9 {2335} 2021/01/19 16:49:19.9 normal block at 0x02993888, 112 bytes long.
2021/01/19 16:49:19.9  Data: <  8h|{;h        > 98 88 38 68 7C 7B 3B 68 00 00 00 00 00 00 00 00 
2021/01/19 16:49:19.9 {2334} 2021/01/19 16:49:19.9 normal block at 0x029CF0F0, 12 bytes long.
2021/01/19 16:49:19.9  Data: <alg_section > 61 6C 67 5F 73 65 63 74 69 6F 6E 00 
2021/01/19 16:49:19.9 {2333} 2021/01/19 16:49:19.9 normal block at 0x029A7DF0, 24 bytes long.
2021/01/19 16:49:19.9  Data: <           h    > 00 00 00 00 F0 F0 9C 02 A0 09 0F 68 00 00 00 00 
2021/01/19 16:49:19.9 {2332} 2021/01/19 16:49:19.9 normal block at 0x029CF4A8, 8 bytes long.
2021/01/19 16:49:19.9  Data: <engines > 65 6E 67 69 6E 65 73 00 
2021/01/19 16:49:19.9 {2331} 2021/01/19 16:49:19.9 normal block at 0x029A7DA8, 24 bytes long.
2021/01/19 16:49:19.9  Data: <           h@  h> 00 00 00 00 A8 F4 9C 02 B0 10 0F 68 40 10 0F 68 
2021/01/19 16:49:19.9 {2330} 2021/01/19 16:49:19.9 normal block at 0x029CF198, 12 bytes long.
2021/01/19 16:49:19.9  Data: <oid_section > 6F 69 64 5F 73 65 63 74 69 6F 6E 00 
2021/01/19 16:49:19.9 {2329} 2021/01/19 16:49:19.9 normal block at 0x029A76A0, 24 bytes long.
2021/01/19 16:49:19.9  Data: <        p  h`  h> 00 00 00 00 98 F1 9C 02 70 08 0F 68 60 08 0F 68 
2021/01/19 16:49:19.9 {2328} 2021/01/19 16:49:19.9 normal block at 0x02957970, 16 bytes long.
2021/01/19 16:49:19.9  Data: < v   }   }      > A0 76 9A 02 A8 7D 9A 02 F0 7D 9A 02 00 00 00 00 
2021/01/19 16:49:19.9 {2327} 2021/01/19 16:49:19.9 normal block at 0x02957EB0, 20 bytes long.
2021/01/19 16:49:19.9  Data: <    py          > 03 00 00 00 70 79 95 02 00 00 00 00 04 00 00 00 
2021/01/19 16:49:19.9 Object dump complete.
2021/01/22 15:29:51.6 AcquisitionEngine: DLL_PROCESS_ATTACH - V11.1.0.1
2021/01/22 15:29:51.6 DataSecurity: DLL_PROCESS_ATTACH - V1.0.0.1
2021/01/22 15:29:51.6 MultiWellPlate: DLL_PROCESS_ATTACH - V1.0.0.6
2021/01/22 15:29:51.7 
2021/01/22 15:29:51.7 Octet Data Acquisition 11.1.2.24 (debug build)
2021/01/22 15:29:51.7 
2021/01/22 15:29:51.9 CreateToolbar()
2021/01/22 15:29:52.0 Instrument started...
2021/01/22 15:29:52.2 Opening instrument: Octet RED96e
2021/01/22 15:29:54.4 ReadInfo()
2021/01/22 15:29:54.9 Serial = FB-90326
2021/01/22 15:29:54.9 GALIL: XY steps/mm = 251.969, Z steps/mm = 1259.84
2021/01/22 15:29:55.7 GetSamplePos1_1 = (3.7179,-106.452,-22.2073)
2021/01/22 15:29:55.7 GetSamplePos1_9 = (3.7179,-36.4881,-39.3672)
2021/01/22 15:29:55.7 m_dPickRowSpacingMM = 9
2021/01/22 15:29:55.7 m_dPickColSpacingMM = -9
2021/01/22 15:29:55.7 m_ySample1RowSpacingMM = 8.74555
2021/01/22 15:29:55.7 m_zSample1RowSpacingMM = -2.14499
2021/01/22 15:29:55.7 m_ySample2RowSpacingMM = 9
2021/01/22 15:29:55.7 m_zSample2RowSpacingMM = 0
2021/01/22 15:29:55.7 
2021/01/22 15:29:55.7 gmcController polling at 6.7 Hz
2021/01/22 15:29:55.7 
2021/01/22 15:29:55.7 Galil Controller - Program Version: 4301
2021/01/22 15:29:55.7 
2021/01/22 15:29:55.7 Opened Octet RED96e with firmware 1.0.0.0, API 1.5
2021/01/22 15:29:55.7 
2021/01/22 15:29:55.7 Set Lamp[65535]=on
2021/01/22 15:29:55.7 Set Lamp[65535]=on
2021/01/22 15:29:55.7 SetHeaterTemperature 30.0
2021/01/22 15:29:55.7 Old State:24- New State:ENGN_State_Launched
2021/01/22 15:29:55.7 Temperature monitor ENABLE
2021/01/22 15:29:55.7 Set Heater[65535] @ 30.000000[C]
2021/01/22 15:29:55.9 Set Lamp[65535]=on
2021/01/22 15:29:56.0 
2021/01/22 15:29:56.0 
2021/01/22 15:29:56.0 ** Initializing Instrument **
2021/01/22 15:29:56.1 Old State:25-ENGN_State_Launched New State:ENGN_State_WaitInitialize
2021/01/22 15:29:56.1 StopDevice[0006]
2021/01/22 15:29:56.3 Opening instrument: Octet RED96e
2021/01/22 15:29:58.5 ReadInfo()
2021/01/22 15:29:59.0 Serial = FB-90326
2021/01/22 15:29:59.0 GALIL: XY steps/mm = 251.969, Z steps/mm = 1259.84
2021/01/22 15:29:59.8 GetSamplePos1_1 = (3.7179,-106.452,-22.2073)
2021/01/22 15:29:59.8 GetSamplePos1_9 = (3.7179,-36.4881,-39.3672)
2021/01/22 15:29:59.8 m_dPickRowSpacingMM = 9
2021/01/22 15:29:59.8 m_dPickColSpacingMM = -9
2021/01/22 15:29:59.8 m_ySample1RowSpacingMM = 8.74555
2021/01/22 15:29:59.8 m_zSample1RowSpacingMM = -2.14499
2021/01/22 15:29:59.8 m_ySample2RowSpacingMM = 9
2021/01/22 15:29:59.8 m_zSample2RowSpacingMM = 0
2021/01/22 15:29:59.8 
2021/01/22 15:29:59.8 gmcController polling at 6.7 Hz
2021/01/22 15:29:59.8 
2021/01/22 15:29:59.8 Galil Controller - Program Version: 4301
2021/01/22 15:29:59.9 
2021/01/22 15:29:59.9 Opened Octet RED96e with firmware 1.0.0.0, API 1.5
2021/01/22 15:29:59.9 
2021/01/22 15:29:59.9 Set Lamp[65535]=on
2021/01/22 15:29:59.9 Set Lamp[65535]=on
2021/01/22 15:29:59.9 Set Heater[65535] @ 30.000000[C]
2021/01/22 15:30:00.0 Looking for 8 devices with timeout of 60000 ms
2021/01/22 15:30:00.0 Found 8 devices in 0 ms
2021/01/22 15:30:00.0 Couldn't open spectrometer factory for type 'SPEC_TYPE_OCEANOPTICS'
2021/01/22 15:30:00.0 EOS EOS_B0735 API = 0x0100
2021/01/22 15:30:00.0 EOS EOS_B0814 API = 0x0100
2021/01/22 15:30:00.0 EOS EOS_B0808 API = 0x0100
2021/01/22 15:30:00.0 EOS EOS_B0732 API = 0x0100
2021/01/22 15:30:00.0 EOS EOS_B0734 API = 0x0100
2021/01/22 15:30:00.0 EOS EOS_B0712 API = 0x0100
2021/01/22 15:30:00.0 EOS EOS_B0817 API = 0x0100
2021/01/22 15:30:00.0 EOS EOS_B0733 API = 0x0100
2021/01/22 15:30:00.0 EOS EOS_B0712 API = 0x0100
2021/01/22 15:30:00.0 EOS EOS_B0734 API = 0x0100
2021/01/22 15:30:00.0 EOS EOS_B0814 API = 0x0100
2021/01/22 15:30:00.0 EOS EOS_B0735 API = 0x0100
2021/01/22 15:30:00.0 EOS EOS_B0732 API = 0x0100
2021/01/22 15:30:00.0 EOS EOS_B0808 API = 0x0100
2021/01/22 15:30:00.0 EOS EOS_B0733 API = 0x0100
2021/01/22 15:30:00.0 EOS EOS_B0817 API = 0x0100
2021/01/22 15:30:00.0 
2021/01/22 15:30:00.0 Calibration coefficients for EOS_B0712
2021/01/22 15:30:00.0 C0 = 5.724263e+02  C1 = 1.047210e-01  C2 = -4.525783e-06  C3 = -1.311609e-10  
2021/01/22 15:30:00.0 
2021/01/22 15:30:00.0 Linearity correction coefficients:
2021/01/22 15:30:00.0 L0 = 9.770590e-01 L1 = 1.598670e-06 L2 = 4.803710e-12 L3 = -5.086290e-15 
2021/01/22 15:30:00.0 L4 = 2.741280e-19 L5 = -6.802670e-24 L6 = 8.120710e-29 L7 = -3.786440e-34 
2021/01/22 15:30:00.0 
2021/01/22 15:30:00.0 Linearity correction order: 7
2021/01/22 15:30:00.0 PCA Serial number: 06 000 2113
2021/01/22 15:30:00.0 Stray light const: 0.000000e+00
2021/01/22 15:30:00.0 Configuration info: TS4 0
2021/01/22 15:30:00.0 Scale factor: 1.023984
2021/01/22 15:30:00.0 
2021/01/22 15:30:00.0 Calibration coefficients for EOS_B0734
2021/01/22 15:30:00.0 C0 = 5.724263e+02  C1 = 1.047210e-01  C2 = -4.525783e-06  C3 = -1.311609e-10  
2021/01/22 15:30:00.0 
2021/01/22 15:30:00.0 Linearity correction coefficients:
2021/01/22 15:30:00.0 L0 = 9.770590e-01 L1 = 1.598670e-06 L2 = 4.803710e-12 L3 = -5.086290e-15 
2021/01/22 15:30:00.0 L4 = 2.741280e-19 L5 = -6.802670e-24 L6 = 8.120710e-29 L7 = -3.786440e-34 
2021/01/22 15:30:00.0 
2021/01/22 15:30:00.0 Linearity correction order: 7
2021/01/22 15:30:00.0 PCA Serial number: 06 000 2172
2021/01/22 15:30:00.0 Stray light const: 0.000000e+00
2021/01/22 15:30:00.0 Configuration info: TS4 0
2021/01/22 15:30:00.0 Scale factor: 1.023984
2021/01/22 15:30:00.0 
2021/01/22 15:30:00.0 Calibration coefficients for EOS_B0814
2021/01/22 15:30:00.0 C0 = 5.724263e+02  C1 = 1.047210e-01  C2 = -4.525783e-06  C3 = -1.311609e-10  
2021/01/22 15:30:00.0 
2021/01/22 15:30:00.0 Linearity correction coefficients:
2021/01/22 15:30:00.0 L0 = 9.770590e-01 L1 = 1.598670e-06 L2 = 4.803710e-12 L3 = -5.086290e-15 
2021/01/22 15:30:00.0 L4 = 2.741280e-19 L5 = -6.802670e-24 L6 = 8.120710e-29 L7 = -3.786440e-34 
2021/01/22 15:30:00.0 
2021/01/22 15:30:00.0 Linearity correction order: 7
2021/01/22 15:30:00.0 PCA Serial number: 06 000 2287
2021/01/22 15:30:00.0 Stray light const: 0.000000e+00
2021/01/22 15:30:00.0 Configuration info: TS4 0
2021/01/22 15:30:00.0 Scale factor: 1.023984
2021/01/22 15:30:00.0 
2021/01/22 15:30:00.0 Calibration coefficients for EOS_B0735
2021/01/22 15:30:00.0 C0 = 5.724263e+02  C1 = 1.047210e-01  C2 = -4.525783e-06  C3 = -1.311609e-10  
2021/01/22 15:30:00.0 
2021/01/22 15:30:00.0 Linearity correction coefficients:
2021/01/22 15:30:00.0 L0 = 9.770590e-01 L1 = 1.598670e-06 L2 = 4.803710e-12 L3 = -5.086290e-15 
2021/01/22 15:30:00.0 L4 = 2.741280e-19 L5 = -6.802670e-24 L6 = 8.120710e-29 L7 = -3.786440e-34 
2021/01/22 15:30:00.0 
2021/01/22 15:30:00.0 Linearity correction order: 7
2021/01/22 15:30:00.0 PCA Serial number: 06 000 2105
2021/01/22 15:30:00.0 Stray light const: 0.000000e+00
2021/01/22 15:30:00.0 Configuration info: TS4 0
2021/01/22 15:30:00.0 Scale factor: 1.023984
2021/01/22 15:30:00.0 
2021/01/22 15:30:00.0 Calibration coefficients for EOS_B0732
2021/01/22 15:30:00.0 C0 = 5.724263e+02  C1 = 1.047210e-01  C2 = -4.525783e-06  C3 = -1.311609e-10  
2021/01/22 15:30:00.0 
2021/01/22 15:30:00.0 Linearity correction coefficients:
2021/01/22 15:30:00.0 L0 = 9.770590e-01 L1 = 1.598670e-06 L2 = 4.803710e-12 L3 = -5.086290e-15 
2021/01/22 15:30:00.0 L4 = 2.741280e-19 L5 = -6.802670e-24 L6 = 8.120710e-29 L7 = -3.786440e-34 
2021/01/22 15:30:00.0 
2021/01/22 15:30:00.0 Linearity correction order: 7
2021/01/22 15:30:00.0 PCA Serial number: 06 000 2214
2021/01/22 15:30:00.0 Stray light const: 0.000000e+00
2021/01/22 15:30:00.0 Configuration info: TS4 0
2021/01/22 15:30:00.0 Scale factor: 1.023984
2021/01/22 15:30:00.0 
2021/01/22 15:30:00.0 Calibration coefficients for EOS_B0808
2021/01/22 15:30:00.0 C0 = 5.724263e+02  C1 = 1.047210e-01  C2 = -4.525783e-06  C3 = -1.311609e-10  
2021/01/22 15:30:00.0 
2021/01/22 15:30:00.0 Linearity correction coefficients:
2021/01/22 15:30:00.0 L0 = 9.770590e-01 L1 = 1.598670e-06 L2 = 4.803710e-12 L3 = -5.086290e-15 
2021/01/22 15:30:00.0 L4 = 2.741280e-19 L5 = -6.802670e-24 L6 = 8.120710e-29 L7 = -3.786440e-34 
2021/01/22 15:30:00.0 
2021/01/22 15:30:00.0 Linearity correction order: 7
2021/01/22 15:30:00.0 PCA Serial number: 06 000 2239
2021/01/22 15:30:00.0 Stray light const: 0.000000e+00
2021/01/22 15:30:00.0 Configuration info: TS4 0
2021/01/22 15:30:00.0 Scale factor: 1.023984
2021/01/22 15:30:00.0 
2021/01/22 15:30:00.0 Calibration coefficients for EOS_B0733
2021/01/22 15:30:00.0 C0 = 5.724263e+02  C1 = 1.047210e-01  C2 = -4.525783e-06  C3 = -1.311609e-10  
2021/01/22 15:30:00.0 
2021/01/22 15:30:00.0 Linearity correction coefficients:
2021/01/22 15:30:00.0 L0 = 9.770590e-01 L1 = 1.598670e-06 L2 = 4.803710e-12 L3 = -5.086290e-15 
2021/01/22 15:30:00.0 L4 = 2.741280e-19 L5 = -6.802670e-24 L6 = 8.120710e-29 L7 = -3.786440e-34 
2021/01/22 15:30:00.0 
2021/01/22 15:30:00.0 Linearity correction order: 7
2021/01/22 15:30:00.0 PCA Serial number: 06 000 2106
2021/01/22 15:30:00.0 Stray light const: 0.000000e+00
2021/01/22 15:30:00.0 Configuration info: TS4 0
2021/01/22 15:30:00.0 Scale factor: 1.023984
2021/01/22 15:30:00.0 
2021/01/22 15:30:00.0 Calibration coefficients for EOS_B0817
2021/01/22 15:30:00.0 C0 = 5.724263e+02  C1 = 1.047210e-01  C2 = -4.525783e-06  C3 = -1.311609e-10  
2021/01/22 15:30:00.0 
2021/01/22 15:30:00.0 Linearity correction coefficients:
2021/01/22 15:30:00.0 L0 = 9.770590e-01 L1 = 1.598670e-06 L2 = 4.803710e-12 L3 = -5.086290e-15 
2021/01/22 15:30:00.0 L4 = 2.741280e-19 L5 = -6.802670e-24 L6 = 8.120710e-29 L7 = -3.786440e-34 
2021/01/22 15:30:00.0 
2021/01/22 15:30:00.0 Linearity correction order: 7
2021/01/22 15:30:00.0 PCA Serial number: 06 000 2286
2021/01/22 15:30:00.0 Stray light const: 0.000000e+00
2021/01/22 15:30:00.0 Configuration info: TS4 0
2021/01/22 15:30:00.0 Scale factor: 1.023984
2021/01/22 15:30:00.0 Opened: EOS_B0712, EOS_B0734, EOS_B0814, EOS_B0735, EOS_B0732, EOS_B0808, EOS_B0733, EOS_B0817, 
2021/01/22 15:30:00.1 
2021/01/22 15:30:00.1 DSP settings (Cross Correlation)
2021/01/22 15:30:00.1  - Transmission
2021/01/22 15:30:00.1 
2021/01/22 15:30:00.1 Ch	IgnoreLow	IgnoreHigh
2021/01/22 15:30:00.1 1	150		200
2021/01/22 15:30:00.1 2	150		200
2021/01/22 15:30:00.1 3	150		200
2021/01/22 15:30:00.1 4	150		200
2021/01/22 15:30:00.1 5	150		200
2021/01/22 15:30:00.1 6	150		200
2021/01/22 15:30:00.1 7	150		200
2021/01/22 15:30:00.1 8	150		200
2021/01/22 15:30:00.4 EJECT: To Chute, position=-2,-1
2021/01/22 15:30:00.5 GMC_Interlock
2021/01/22 15:30:00.6 Old State:1-ENGN_State_WaitInitialize New State:ENGN_State_Initializing
2021/01/22 15:30:00.8 GMC_HomeXYZ
2021/01/22 15:30:00.8 HomeDevice[0006]
2021/01/22 15:30:06.7 GMC_SetPosZ
2021/01/22 15:30:07.1 GMC_ServoHere
2021/01/22 15:30:07.3 GMC_MasterInit
2021/01/22 15:30:07.3 Init(0x0)
2021/01/22 15:30:11.6 GMC_SetPosZ
2021/01/22 15:30:11.8 GMC_ServoHere
2021/01/22 15:30:12.0 GMC_SetPosXY
2021/01/22 15:30:14.8 GMC_SetPosZ
2021/01/22 15:30:19.7 GMC_Eject
2021/01/22 15:30:19.7 EjectDone => EjectEject
2021/01/22 15:30:19.7 SetEjectorPosn(0)
2021/01/22 15:30:22.8 GMC_HomeZ
2021/01/22 15:30:22.8 HomeDevice[0005]
2021/01/22 15:30:29.1 GMC_SetPosZ
2021/01/22 15:30:29.4 GMC_ServoHere
2021/01/22 15:30:29.6 GMC_Interlock
2021/01/22 15:30:30.0 GMC_HomeXYZ
2021/01/22 15:30:30.0 HomeDevice[0006]
2021/01/22 15:30:37.6 GMC_SetPosZ
2021/01/22 15:30:37.9 GMC_ServoHere
2021/01/22 15:30:38.1 GMC_SetPosZ
2021/01/22 15:30:38.3 GMC_ServoHere
2021/01/22 15:30:38.5 GMC_SetPosXY
2021/01/22 15:30:41.5 GMC_SetPosZ
2021/01/22 15:30:41.9 Old State:2-ENGN_State_Initializing New State:ENGN_State_InitializeComplete
2021/01/22 15:30:41.9 Integration Time: 1.00,1.00,1.00,1.00,1.00,1.00,1.00,1.00,
2021/01/22 15:30:41.9 Set Lamp[65535]=on
2021/01/22 15:30:42.0 Old State:3-ENGN_State_InitializeComplete New State:ENGN_State_SystemReady
2021/01/22 15:30:42.0 Set Heater[65535] @ 30.000000[C]
2021/01/22 15:30:42.1 Pickups: 322, Warn: 90, Error: 100
2021/01/22 15:49:29.7 SetHeaterTemperature 25.0
2021/01/22 15:49:29.7 Set Heater[65535] @ 25.000000[C]
2021/01/22 15:49:29.8 Temperature monitor ENABLE
2021/01/22 15:49:29.8 Temp: 2021/01/22 15:49:29, 30.0000, 25.1000
2021/01/22 15:49:33.8 bShakeWhileWaiting=1, nShakerRPM=1000
2021/01/22 15:49:33.9 DELAYED START: Waiting for temperature 25 DegC to be reached
2021/01/22 15:49:33.9 GMC_SetPosZ
2021/01/22 15:49:34.0 Old State:4-ENGN_State_SystemReady New State:ENGN_State_ControllerBusy
2021/01/22 15:49:34.1 GMC_ServoHere
2021/01/22 15:49:34.3 GMC_SetPosXY
2021/01/22 15:49:34.4 GMC_SetPosZ
2021/01/22 15:49:34.7 Old State:29-ENGN_State_ControllerBusy New State:ENGN_State_ControllerDone
2021/01/22 15:49:34.9 Old State:30-ENGN_State_ControllerDone New State:ENGN_State_SystemReady
2021/01/22 15:49:34.9 StartShaker(1000)
2021/01/22 15:49:34.9 Temp: 2021/01/22 15:49:34, 28.5000, 25.2000
2021/01/22 15:49:35.9 Temp: 2021/01/22 15:49:35, 28.0000, 25.2000
2021/01/22 15:49:36.9 Temp: 2021/01/22 15:49:36, 27.5000, 25.1000
2021/01/22 15:49:37.9 Temp: 2021/01/22 15:49:37, 27.1000, 25.1000
2021/01/22 15:49:38.9 Temp: 2021/01/22 15:49:38, 26.7000, 25.2000
2021/01/22 15:49:39.9 DELAYED START: Reached temperature 25 DegC.
2021/01/22 15:49:39.9 Spectrometer EOS_B0712: AVRG=20.
2021/01/22 15:49:39.9 Spectrometer EOS_B0734: AVRG=20.
2021/01/22 15:49:39.9 Spectrometer EOS_B0814: AVRG=20.
2021/01/22 15:49:39.9 Spectrometer EOS_B0735: AVRG=20.
2021/01/22 15:49:39.9 Spectrometer EOS_B0732: AVRG=20.
2021/01/22 15:49:39.9 Spectrometer EOS_B0808: AVRG=20.
2021/01/22 15:49:39.9 Spectrometer EOS_B0733: AVRG=20.
2021/01/22 15:49:39.9 Spectrometer EOS_B0817: AVRG=20.
2021/01/22 15:49:40.0 Starting new assay
2021/01/22 15:49:40.2 Plate 1 temperature = 26 C
2021/01/22 15:49:40.2 Old State:4-ENGN_State_SystemReady New State:ENGN_StateRunning_StartNewStep
2021/01/22 15:49:40.3 Spectrometer EOS_B0712: AVRG=20.
2021/01/22 15:49:40.3 Spectrometer EOS_B0734: AVRG=20.
2021/01/22 15:49:40.3 Spectrometer EOS_B0814: AVRG=20.
2021/01/22 15:49:40.3 Spectrometer EOS_B0735: AVRG=20.
2021/01/22 15:49:40.3 Spectrometer EOS_B0732: AVRG=20.
2021/01/22 15:49:40.3 Spectrometer EOS_B0808: AVRG=20.
2021/01/22 15:49:40.3 Spectrometer EOS_B0733: AVRG=20.
2021/01/22 15:49:40.3 Spectrometer EOS_B0817: AVRG=20.
2021/01/22 15:49:40.4 Old State:6-ENGN_StateRunning_StartNewStep New State:ENGN_StateRunning_WaitPickingSensors
2021/01/22 15:49:40.4 Pick(0, 0)
2021/01/22 15:49:40.4 GMC_SetPosZ
2021/01/22 15:49:40.5 Old State:7-ENGN_StateRunning_WaitPickingSensors New State:ENGN_StateRunning_PickingSensors
2021/01/22 15:49:40.5 Temp: 2021/01/22 15:49:40, 26.2000, 25.2000
2021/01/22 15:49:40.6 GMC_ServoHere
2021/01/22 15:49:40.8 GMC_SetPosXY
2021/01/22 15:49:43.0 GMC_SetPosZ
2021/01/22 15:49:48.5 GMC_SetPosZ
2021/01/22 15:49:49.0 Old State:8-ENGN_StateRunning_PickingSensors New State:ENGN_StateRunning_CheckingSensors
2021/01/22 15:49:49.1 Spectrometer EOS_B0712: AVRG=20.
2021/01/22 15:49:49.1 Spectrometer EOS_B0734: AVRG=20.
2021/01/22 15:49:49.1 Spectrometer EOS_B0814: AVRG=20.
2021/01/22 15:49:49.1 Spectrometer EOS_B0735: AVRG=20.
2021/01/22 15:49:49.1 Spectrometer EOS_B0732: AVRG=20.
2021/01/22 15:49:49.1 Spectrometer EOS_B0808: AVRG=20.
2021/01/22 15:49:49.1 Spectrometer EOS_B0733: AVRG=20.
2021/01/22 15:49:49.1 Spectrometer EOS_B0817: AVRG=20.
2021/01/22 15:49:49.2 
2021/01/22 15:49:49.2 Detect Sensors
2021/01/22 15:49:49.2 ==============
2021/01/22 15:49:49.2 1	YES		0.0	6457.4
2021/01/22 15:49:49.2 2	YES		0.0	6696.3
2021/01/22 15:49:49.2 3	YES		0.0	6822.6
2021/01/22 15:49:49.2 4	YES		0.0	5972.9
2021/01/22 15:49:49.2 5	YES		0.0	5761.1
2021/01/22 15:49:49.2 6	YES		0.0	6480.1
2021/01/22 15:49:49.2 7	YES		0.0	6574.0
2021/01/22 15:49:49.2 8	YES		0.0	5978.0
2021/01/22 15:49:49.2 
2021/01/22 15:49:49.2 Spectrometer EOS_B0712: AVRG=20.
2021/01/22 15:49:49.2 Spectrometer EOS_B0734: AVRG=20.
2021/01/22 15:49:49.3 Spectrometer EOS_B0814: AVRG=20.
2021/01/22 15:49:49.3 Spectrometer EOS_B0735: AVRG=20.
2021/01/22 15:49:49.3 Spectrometer EOS_B0732: AVRG=20.
2021/01/22 15:49:49.3 Spectrometer EOS_B0808: AVRG=20.
2021/01/22 15:49:49.3 Spectrometer EOS_B0733: AVRG=20.
2021/01/22 15:49:49.3 Spectrometer EOS_B0817: AVRG=20.
2021/01/22 15:49:49.3 Spectrometer EOS_B0712: AVRG=20.
2021/01/22 15:49:49.3 Spectrometer EOS_B0734: AVRG=20.
2021/01/22 15:49:49.3 Spectrometer EOS_B0814: AVRG=20.
2021/01/22 15:49:49.3 Spectrometer EOS_B0735: AVRG=20.
2021/01/22 15:49:49.3 Spectrometer EOS_B0732: AVRG=20.
2021/01/22 15:49:49.3 Spectrometer EOS_B0808: AVRG=20.
2021/01/22 15:49:49.3 Spectrometer EOS_B0733: AVRG=20.
2021/01/22 15:49:49.3 Spectrometer EOS_B0817: AVRG=20.
2021/01/22 15:49:49.4 CH1: Correction Factor = 1.97292
2021/01/22 15:49:49.4 Spectrometer EOS_B0712: AVRG=20.
2021/01/22 15:49:49.4 CH2: Correction Factor = 1.95398
2021/01/22 15:49:49.4 Spectrometer EOS_B0734: AVRG=20.
2021/01/22 15:49:49.4 CH3: Correction Factor = 1.96401
2021/01/22 15:49:49.4 Spectrometer EOS_B0814: AVRG=20.
2021/01/22 15:49:49.4 CH4: Correction Factor = 2.09007
2021/01/22 15:49:49.4 Spectrometer EOS_B0735: AVRG=20.
2021/01/22 15:49:49.4 CH5: Correction Factor = 2.23706
2021/01/22 15:49:49.4 Spectrometer EOS_B0732: AVRG=20.
2021/01/22 15:49:49.4 CH6: Correction Factor = 2.05023
2021/01/22 15:49:49.4 Spectrometer EOS_B0808: AVRG=20.
2021/01/22 15:49:49.4 CH7: Correction Factor = 2.00731
2021/01/22 15:49:49.4 Spectrometer EOS_B0733: AVRG=20.
2021/01/22 15:49:49.4 CH8: Correction Factor = 2.15586
2021/01/22 15:49:49.4 Spectrometer EOS_B0817: AVRG=20.
2021/01/22 15:49:49.6 CH1: Correction Factor = 1.1066
2021/01/22 15:49:49.6 Spectrometer EOS_B0712: AVRG=20.
2021/01/22 15:49:49.6 CH2: Correction Factor = 1.10882
2021/01/22 15:49:49.6 Spectrometer EOS_B0734: AVRG=20.
2021/01/22 15:49:49.6 CH3: Correction Factor = 1.10647
2021/01/22 15:49:49.6 Spectrometer EOS_B0814: AVRG=20.
2021/01/22 15:49:49.6 CH4: Correction Factor = 1.1157
2021/01/22 15:49:49.6 Spectrometer EOS_B0735: AVRG=20.
2021/01/22 15:49:49.6 CH5: Correction Factor = 1.12097
2021/01/22 15:49:49.6 Spectrometer EOS_B0732: AVRG=20.
2021/01/22 15:49:49.6 CH6: Correction Factor = 1.11024
2021/01/22 15:49:49.6 Spectrometer EOS_B0808: AVRG=20.
2021/01/22 15:49:49.6 CH7: Correction Factor = 1.10618
2021/01/22 15:49:49.6 Spectrometer EOS_B0733: AVRG=20.
2021/01/22 15:49:49.6 CH8: Correction Factor = 1.11459
2021/01/22 15:49:49.6 Spectrometer EOS_B0817: AVRG=20.
2021/01/22 15:49:49.7 CH1: Correction Factor = 1.0199
2021/01/22 15:49:49.7 Spectrometer EOS_B0712: AVRG=20.
2021/01/22 15:49:49.7 CH2: Correction Factor = 1.01755
2021/01/22 15:49:49.7 Spectrometer EOS_B0734: AVRG=20.
2021/01/22 15:49:49.7 CH3: Correction Factor = 1.01962
2021/01/22 15:49:49.8 Spectrometer EOS_B0814: AVRG=20.
2021/01/22 15:49:49.8 CH4: Correction Factor = 1.02246
2021/01/22 15:49:49.8 Spectrometer EOS_B0735: AVRG=20.
2021/01/22 15:49:49.8 CH5: Correction Factor = 1.02269
2021/01/22 15:49:49.8 Spectrometer EOS_B0732: AVRG=20.
2021/01/22 15:49:49.8 CH6: Correction Factor = 1.01932
2021/01/22 15:49:49.8 Spectrometer EOS_B0808: AVRG=20.
2021/01/22 15:49:49.8 CH7: Correction Factor = 1.02134
2021/01/22 15:49:49.8 Spectrometer EOS_B0733: AVRG=20.
2021/01/22 15:49:49.8 CH8: Correction Factor = 1.02378
2021/01/22 15:49:49.8 Spectrometer EOS_B0817: AVRG=20.
2021/01/22 15:49:49.9 CH1: Correction Factor = 1.00189
2021/01/22 15:49:49.9 Spectrometer EOS_B0712: AVRG=20.
2021/01/22 15:49:49.9 CH2: Correction Factor = 1.00305
2021/01/22 15:49:49.9 Spectrometer EOS_B0734: AVRG=20.
2021/01/22 15:49:49.9 CH3: Correction Factor = 1.00137
2021/01/22 15:49:49.9 Spectrometer EOS_B0814: AVRG=20.
2021/01/22 15:49:49.9 CH4: Correction Factor = 1.00549
2021/01/22 15:49:49.9 Spectrometer EOS_B0735: AVRG=20.
2021/01/22 15:49:49.9 CH5: Correction Factor = 1.0038
2021/01/22 15:49:49.9 Spectrometer EOS_B0732: AVRG=20.
2021/01/22 15:49:49.9 CH6: Correction Factor = 1.00537
2021/01/22 15:49:49.9 Spectrometer EOS_B0808: AVRG=20.
2021/01/22 15:49:49.9 CH7: Correction Factor = 1.00357
2021/01/22 15:49:49.9 Spectrometer EOS_B0733: AVRG=20.
2021/01/22 15:49:49.9 CH8: Correction Factor = 1.00394
2021/01/22 15:49:49.9 Spectrometer EOS_B0817: AVRG=20.
2021/01/22 15:49:50.1 CH1: Correction Factor = 1.00187
2021/01/22 15:49:50.1 Spectrometer EOS_B0712: AVRG=20.
2021/01/22 15:49:50.1 CH2: Correction Factor = 1.00302
2021/01/22 15:49:50.1 Spectrometer EOS_B0734: AVRG=20.
2021/01/22 15:49:50.1 CH3: Correction Factor = 1.00147
2021/01/22 15:49:50.1 Spectrometer EOS_B0814: AVRG=20.
2021/01/22 15:49:50.1 CH4: Correction Factor = 0.998873
2021/01/22 15:49:50.1 Spectrometer EOS_B0735: AVRG=20.
2021/01/22 15:49:50.1 CH5: Correction Factor = 1.00384
2021/01/22 15:49:50.1 Spectrometer EOS_B0732: AVRG=20.
2021/01/22 15:49:50.1 CH6: Correction Factor = 1.00217
2021/01/22 15:49:50.1 Spectrometer EOS_B0808: AVRG=20.
2021/01/22 15:49:50.1 CH7: Correction Factor = 1.00022
2021/01/22 15:49:50.1 Spectrometer EOS_B0733: AVRG=20.
2021/01/22 15:49:50.1 CH8: Correction Factor = 1.00069
2021/01/22 15:49:50.1 Spectrometer EOS_B0817: AVRG=20.
2021/01/22 15:49:50.2 CH1: Correction Factor = 0.998245
2021/01/22 15:49:50.2 Spectrometer EOS_B0712: AVRG=20.
2021/01/22 15:49:50.2 CH2: Correction Factor = 0.999586
2021/01/22 15:49:50.2 Spectrometer EOS_B0734: AVRG=20.
2021/01/22 15:49:50.2 CH3: Correction Factor = 1.00141
2021/01/22 15:49:50.2 Spectrometer EOS_B0814: AVRG=20.
2021/01/22 15:49:50.2 CH4: Correction Factor = 0.998824
2021/01/22 15:49:50.2 Spectrometer EOS_B0735: AVRG=20.
2021/01/22 15:49:50.2 CH5: Correction Factor = 1.00071
2021/01/22 15:49:50.2 Spectrometer EOS_B0732: AVRG=20.
2021/01/22 15:49:50.2 CH6: Correction Factor = 0.998621
2021/01/22 15:49:50.2 Spectrometer EOS_B0808: AVRG=20.
2021/01/22 15:49:50.2 CH7: Correction Factor = 1.00024
2021/01/22 15:49:50.2 Spectrometer EOS_B0733: AVRG=20.
2021/01/22 15:49:50.2 CH8: Correction Factor = 1.00059
2021/01/22 15:49:50.2 Spectrometer EOS_B0817: AVRG=20.
2021/01/22 15:49:50.3 Sensor 1: Integration Time = 2.23 ms
2021/01/22 15:49:50.3 Sensor 2: Integration Time = 2.22 ms
2021/01/22 15:49:50.3 Sensor 3: Integration Time = 2.23 ms
2021/01/22 15:49:50.3 Sensor 4: Integration Time = 2.39 ms
2021/01/22 15:49:50.3 Sensor 5: Integration Time = 2.59 ms
2021/01/22 15:49:50.3 Sensor 6: Integration Time = 2.33 ms
2021/01/22 15:49:50.3 Sensor 7: Integration Time = 2.28 ms
2021/01/22 15:49:50.3 Sensor 8: Integration Time = 2.47 ms
2021/01/22 15:49:50.3 WaitControllerIdle(30000)
2021/01/22 15:49:50.3 GMC_HomeZ
2021/01/22 15:49:50.3 HomeDevice[0005]
2021/01/22 15:49:57.1 GMC_SetPosZ
2021/01/22 15:49:57.4 GMC_ServoHere
2021/01/22 15:49:57.7 Old State:9-ENGN_StateRunning_CheckingSensors New State:ENGN_StateRunning_PickingSensorsComplete
2021/01/22 15:49:57.7 StartShaker(1000)
2021/01/22 15:49:57.7 Integration Time: 2.23,2.22,2.23,2.39,2.59,2.33,2.28,2.47,
2021/01/22 15:49:57.8 Old State:10-ENGN_StateRunning_PickingSensorsComplete New State:ENGN_StateRunning_WaitGoingToSample
2021/01/22 15:49:57.8 Sample96_Q(0, 0, 0)
2021/01/22 15:49:57.9 
2021/01/22 15:49:57.9 START SCAN: 
2021/01/22 15:49:57.9    CycleTime (ms)    = 200
2021/01/22 15:49:57.9    CycleCount        = 300
2021/01/22 15:49:57.9    ScansToAverage    = 20
2021/01/22 15:49:57.9    FinishTime (s)    = 60
2021/01/22 15:49:57.9    Temperature       = 24.8
2021/01/22 15:49:57.9    Lamp Power        = 31
2021/01/22 15:49:57.9    Spec PCB temps    = 42.1, 41.8, 42.7, 42.4, 42.1, 41.9, 40.7, 41.1
2021/01/22 15:49:57.9 
2021/01/22 15:49:57.9 GMC_SetPosZ
2021/01/22 15:49:58.0 Old State:12-ENGN_StateRunning_WaitGoingToSample New State:ENGN_StateRunning_GoingToSample
2021/01/22 15:49:58.0 Plate 1 temperature = 25 C
2021/01/22 15:49:58.0 Pickups: 323, Warn: 90, Error: 100
2021/01/22 15:49:58.1 GMC_ServoHere
2021/01/22 15:49:58.3 GMC_SetPosXY
2021/01/22 15:50:00.4 GMC_SetPosZ
2021/01/22 15:50:03.6 Old State:13-ENGN_StateRunning_GoingToSample New State:ENGN_StateRunning_GoingToSampleComplete
2021/01/22 15:50:03.7 Old State:14-ENGN_StateRunning_GoingToSampleComplete New State:ENGN_StateRunning_WaitSampling
2021/01/22 15:50:03.7 Step started at: 15:49:57
2021/01/22 15:50:03.7 XYZ movement time  = 5678 ms + Scan start delay = 5579 ms
2021/01/22 15:50:03.8 Old State:15-ENGN_StateRunning_WaitSampling New State:ENGN_StateRunning_Sampling
2021/01/22 15:50:07.7 Shaker current speed = 992 rpm, set speed = 1000
2021/01/22 15:50:40.7 Temp: 2021/01/22 15:50:40, 24.6000, 25.2000
2021/01/22 15:51:03.4 
2021/01/22 15:51:03.4 STOP SCAN: 
2021/01/22 15:51:03.4    CycleTime (ms)    = 200
2021/01/22 15:51:03.4    CurrentCycle      = 300
2021/01/22 15:51:03.4    RunTime (ms)      = 59959
2021/01/22 15:51:03.4    AverageCycle (ms) = 199.9
2021/01/22 15:51:03.4    Temperature       = 24.7
2021/01/22 15:51:03.4    DATA OVERRUN %    = 0.00
2021/01/22 15:51:03.4    Spec PCB temps    = 42.1, 41.8, 42.6, 42.4, 42.1, 41.9, 40.7, 41.1
2021/01/22 15:51:03.4 
2021/01/22 15:51:03.5 Old State:16-cycle=300, Elapsed Time:   60.0,  Ideal Time:   59.8,  Delay time =   0.04  New State:ENGN_StateRunning_SamplingComplete
2021/01/22 15:51:03.6 Old State:18-ENGN_StateRunning_SamplingComplete New State:ENGN_StateRunning_WaitNewStep
2021/01/22 15:51:03.7 Old State:5-ENGN_StateRunning_WaitNewStep New State:ENGN_StateRunning_StartNewStep
2021/01/22 15:51:03.7 Spectrometer EOS_B0712: AVRG=20.
2021/01/22 15:51:03.7 Spectrometer EOS_B0734: AVRG=20.
2021/01/22 15:51:03.7 Spectrometer EOS_B0814: AVRG=20.
2021/01/22 15:51:03.7 Spectrometer EOS_B0735: AVRG=20.
2021/01/22 15:51:03.7 Spectrometer EOS_B0732: AVRG=20.
2021/01/22 15:51:03.7 Spectrometer EOS_B0808: AVRG=20.
2021/01/22 15:51:03.7 Spectrometer EOS_B0733: AVRG=20.
2021/01/22 15:51:03.7 Spectrometer EOS_B0817: AVRG=20.
2021/01/22 15:51:03.7 StartShaker(500)
2021/01/22 15:51:03.8 Old State:6-ENGN_StateRunning_StartNewStep New State:ENGN_StateRunning_WaitGoingToSample
2021/01/22 15:51:03.8 Sample96_Q(0, 0, 1)
2021/01/22 15:51:03.9 
2021/01/22 15:51:03.9 START SCAN: 
2021/01/22 15:51:03.9    CycleTime (ms)    = 200
2021/01/22 15:51:03.9    CycleCount        = 1000
2021/01/22 15:51:03.9    ScansToAverage    = 20
2021/01/22 15:51:03.9    FinishTime (s)    = 200
2021/01/22 15:51:03.9    Temperature       = 24.7
2021/01/22 15:51:03.9    Lamp Power        = 31
2021/01/22 15:51:03.9    Spec PCB temps    = 42.1, 41.8, 42.6, 42.4, 42.1, 41.9, 40.7, 41.1
2021/01/22 15:51:03.9 
2021/01/22 15:51:03.9 Old State:12-ENGN_StateRunning_WaitGoingToSample New State:ENGN_StateRunning_GoingToSample
2021/01/22 15:51:04.0 GMC_SetPosZ
2021/01/22 15:51:04.0 Temp: 2021/01/22 15:51:04, 24.7000, 25.2000
2021/01/22 15:51:07.1 GMC_ServoHere
2021/01/22 15:51:07.2 GMC_SetPosXY
2021/01/22 15:51:07.5 GMC_SetPosZ
2021/01/22 15:51:11.0 Old State:13-ENGN_StateRunning_GoingToSample New State:ENGN_StateRunning_GoingToSampleComplete
2021/01/22 15:51:11.1 Old State:14-cycle=1, Elapsed Time:    0.0,  Ideal Time:    0.0,  Delay time =   0.00  New State:ENGN_StateRunning_WaitSampling
2021/01/22 15:51:11.1 Step started at: 15:51:03
2021/01/22 15:51:11.1 XYZ movement time  = 7050 ms + Scan start delay = 7044 ms
2021/01/22 15:51:11.2 Old State:15-ENGN_StateRunning_WaitSampling New State:ENGN_StateRunning_Sampling
2021/01/22 15:51:13.8 Shaker current speed = 494 rpm, set speed = 500
2021/01/22 15:52:04.2 Temp: 2021/01/22 15:52:04, 24.8000, 25.2000
2021/01/22 15:53:04.3 Temp: 2021/01/22 15:53:04, 25.0000, 25.2000
2021/01/22 15:54:04.5 Temp: 2021/01/22 15:54:04, 25.0000, 25.2000
2021/01/22 15:54:08.5 Step 2 extended by 60 seconds
2021/01/22 15:55:04.7 Temp: 2021/01/22 15:55:04, 25.0000, 25.2000
2021/01/22 15:55:30.8 
2021/01/22 15:55:30.8 STOP SCAN: 
2021/01/22 15:55:30.8    CycleTime (ms)    = 200
2021/01/22 15:55:30.8    CurrentCycle      = 1300
2021/01/22 15:55:30.8    RunTime (ms)      = 259956
2021/01/22 15:55:30.8    AverageCycle (ms) = 200.0
2021/01/22 15:55:30.8    Temperature       = 25.0
2021/01/22 15:55:30.8    DATA OVERRUN %    = 0.00
2021/01/22 15:55:30.8    Spec PCB temps    = 42.1, 41.8, 42.7, 42.4, 42.2, 41.9, 40.8, 41.1
2021/01/22 15:55:30.8 
2021/01/22 15:55:30.9 Old State:16-cycle=1300, Elapsed Time:  260.0,  Ideal Time:  259.8,  Delay time =   0.04  New State:ENGN_StateRunning_SamplingComplete
2021/01/22 15:55:31.1 Old State:18-ENGN_StateRunning_SamplingComplete New State:ENGN_StateRunning_WaitNewStep
2021/01/22 15:55:31.2 Old State:5-ENGN_StateRunning_WaitNewStep New State:ENGN_StateRunning_StartNewStep
2021/01/22 15:55:31.2 Spectrometer EOS_B0712: AVRG=20.
2021/01/22 15:55:31.2 Spectrometer EOS_B0734: AVRG=20.
2021/01/22 15:55:31.2 Spectrometer EOS_B0814: AVRG=20.
2021/01/22 15:55:31.2 Spectrometer EOS_B0735: AVRG=20.
2021/01/22 15:55:31.2 Spectrometer EOS_B0732: AVRG=20.
2021/01/22 15:55:31.2 Spectrometer EOS_B0808: AVRG=20.
2021/01/22 15:55:31.2 Spectrometer EOS_B0733: AVRG=20.
2021/01/22 15:55:31.2 Spectrometer EOS_B0817: AVRG=20.
2021/01/22 15:55:31.2 StartShaker(500)
2021/01/22 15:55:31.3 Old State:6-ENGN_StateRunning_StartNewStep New State:ENGN_StateRunning_WaitGoingToSample
2021/01/22 15:55:31.3 Sample96_Q(0, 0, 2)
2021/01/22 15:55:31.3 
2021/01/22 15:55:31.3 START SCAN: 
2021/01/22 15:55:31.3    CycleTime (ms)    = 200
2021/01/22 15:55:31.3    CycleCount        = 150
2021/01/22 15:55:31.3    ScansToAverage    = 20
2021/01/22 15:55:31.3    FinishTime (s)    = 30
2021/01/22 15:55:31.3    Temperature       = 25.0
2021/01/22 15:55:31.3    Lamp Power        = 31
2021/01/22 15:55:31.3    Spec PCB temps    = 42.1, 41.8, 42.7, 42.4, 42.2, 41.9, 40.8, 41.1
2021/01/22 15:55:31.3 
2021/01/22 15:55:31.4 GMC_SetPosZ
2021/01/22 15:55:31.4 Old State:12-ENGN_StateRunning_WaitGoingToSample New State:ENGN_StateRunning_GoingToSample
2021/01/22 15:55:31.5 Temp: 2021/01/22 15:55:31, 25.0000, 25.2000
2021/01/22 15:55:34.7 GMC_ServoHere
2021/01/22 15:55:34.9 GMC_SetPosXY
2021/01/22 15:55:35.2 GMC_SetPosZ
2021/01/22 15:55:38.9 Old State:13-ENGN_StateRunning_GoingToSample New State:ENGN_StateRunning_GoingToSampleComplete
2021/01/22 15:55:39.0 Old State:14-cycle=1, Elapsed Time:    0.0,  Ideal Time:    0.0,  Delay time =   0.00  New State:ENGN_StateRunning_WaitSampling
2021/01/22 15:55:39.0 Step started at: 15:55:31
2021/01/22 15:55:39.0 XYZ movement time  = 7541 ms + Scan start delay = 7545 ms
2021/01/22 15:55:39.1 Old State:15-ENGN_StateRunning_WaitSampling New State:ENGN_StateRunning_Sampling
2021/01/22 15:55:41.3 Shaker current speed = 494 rpm, set speed = 500
2021/01/22 15:56:08.8 
2021/01/22 15:56:08.8 STOP SCAN: 
2021/01/22 15:56:08.8    CycleTime (ms)    = 200
2021/01/22 15:56:08.8    CurrentCycle      = 150
2021/01/22 15:56:08.8    RunTime (ms)      = 29956
2021/01/22 15:56:08.8    AverageCycle (ms) = 199.7
2021/01/22 15:56:08.8    Temperature       = 25.0
2021/01/22 15:56:08.8    DATA OVERRUN %    = 0.00
2021/01/22 15:56:08.8    Spec PCB temps    = 42.1, 41.8, 42.6, 42.4, 42.2, 41.9, 40.8, 41.1
2021/01/22 15:56:08.8 
2021/01/22 15:56:08.9 Old State:16-cycle=150, Elapsed Time:   30.0,  Ideal Time:   29.8,  Delay time =   0.04  New State:ENGN_StateRunning_SamplingComplete
2021/01/22 15:56:09.1 Old State:18-ENGN_StateRunning_SamplingComplete New State:ENGN_StateRunning_WaitNewStep
2021/01/22 15:56:09.4 Old State:5-ENGN_StateRunning_WaitNewStep New State:ENGN_StateRunning_StartNewStep
2021/01/22 15:56:09.4 Spectrometer EOS_B0712: AVRG=20.
2021/01/22 15:56:09.4 Spectrometer EOS_B0734: AVRG=20.
2021/01/22 15:56:09.4 Spectrometer EOS_B0814: AVRG=20.
2021/01/22 15:56:09.4 Spectrometer EOS_B0735: AVRG=20.
2021/01/22 15:56:09.4 Spectrometer EOS_B0732: AVRG=20.
2021/01/22 15:56:09.4 Spectrometer EOS_B0808: AVRG=20.
2021/01/22 15:56:09.4 Spectrometer EOS_B0733: AVRG=20.
2021/01/22 15:56:09.4 Spectrometer EOS_B0817: AVRG=20.
2021/01/22 15:56:09.4 StartShaker(500)
2021/01/22 15:56:09.5 Old State:6-ENGN_StateRunning_StartNewStep New State:ENGN_StateRunning_WaitGoingToSample
2021/01/22 15:56:09.5 Sample96_Q(0, 0, 3)
2021/01/22 15:56:09.6 
2021/01/22 15:56:09.6 START SCAN: 
2021/01/22 15:56:09.6    CycleTime (ms)    = 200
2021/01/22 15:56:09.6    CycleCount        = 1500
2021/01/22 15:56:09.6    ScansToAverage    = 20
2021/01/22 15:56:09.6    FinishTime (s)    = 300
2021/01/22 15:56:09.6    Temperature       = 25.0
2021/01/22 15:56:09.6    Lamp Power        = 31
2021/01/22 15:56:09.6    Spec PCB temps    = 42.2, 41.8, 42.6, 42.4, 42.2, 41.9, 40.8, 41.1
2021/01/22 15:56:09.6 
2021/01/22 15:56:09.6 GMC_SetPosZ
2021/01/22 15:56:09.6 Temp: 2021/01/22 15:56:09, 25.0000, 25.3000
2021/01/22 15:56:09.7 Old State:12-ENGN_StateRunning_WaitGoingToSample New State:ENGN_StateRunning_GoingToSample
2021/01/22 15:56:13.1 GMC_ServoHere
2021/01/22 15:56:13.2 GMC_SetPosXY
2021/01/22 15:56:13.6 GMC_SetPosZ
2021/01/22 15:56:17.6 Old State:13-ENGN_StateRunning_GoingToSample New State:ENGN_StateRunning_GoingToSampleComplete
2021/01/22 15:56:17.7 Old State:14-cycle=1, Elapsed Time:    0.0,  Ideal Time:    0.0,  Delay time =   0.00  New State:ENGN_StateRunning_WaitSampling
2021/01/22 15:56:17.7 Step started at: 15:56:09
2021/01/22 15:56:17.7 XYZ movement time  = 7950 ms + Scan start delay = 7953 ms
2021/01/22 15:56:17.8 Old State:15-ENGN_StateRunning_WaitSampling New State:ENGN_StateRunning_Sampling
2021/01/22 15:56:19.5 Shaker current speed = 494 rpm, set speed = 500
2021/01/22 15:57:09.8 Temp: 2021/01/22 15:57:09, 25.0000, 25.2000
2021/01/22 15:58:09.9 Temp: 2021/01/22 15:58:09, 25.0000, 25.3000
2021/01/22 15:59:10.1 Temp: 2021/01/22 15:59:10, 25.0000, 25.4000
2021/01/22 16:00:10.3 Temp: 2021/01/22 16:00:10, 25.0000, 25.3000
2021/01/22 16:01:10.4 Temp: 2021/01/22 16:01:10, 25.0000, 25.4000
2021/01/22 16:01:17.5 
2021/01/22 16:01:17.5 STOP SCAN: 
2021/01/22 16:01:17.5    CycleTime (ms)    = 200
2021/01/22 16:01:17.5    CurrentCycle      = 1500
2021/01/22 16:01:17.5    RunTime (ms)      = 299959
2021/01/22 16:01:17.5    AverageCycle (ms) = 200.0
2021/01/22 16:01:17.5    Temperature       = 25.0
2021/01/22 16:01:17.5    DATA OVERRUN %    = 0.00
2021/01/22 16:01:17.5    Spec PCB temps    = 42.2, 41.9, 42.7, 42.4, 42.1, 41.9, 40.8, 41.1
2021/01/22 16:01:17.5 
2021/01/22 16:01:17.6 Old State:16-cycle=1500, Elapsed Time:  300.0,  Ideal Time:  299.8,  Delay time =   0.04  New State:ENGN_StateRunning_SamplingComplete
2021/01/22 16:01:17.7 Old State:18-ENGN_StateRunning_SamplingComplete New State:ENGN_StateRunning_WaitNewStep
2021/01/22 16:01:18.0 Old State:5-ENGN_StateRunning_WaitNewStep New State:ENGN_StateRunning_StartNewStep
2021/01/22 16:01:18.1 Spectrometer EOS_B0712: AVRG=20.
2021/01/22 16:01:18.1 Spectrometer EOS_B0734: AVRG=20.
2021/01/22 16:01:18.1 Spectrometer EOS_B0814: AVRG=20.
2021/01/22 16:01:18.1 Spectrometer EOS_B0735: AVRG=20.
2021/01/22 16:01:18.1 Spectrometer EOS_B0732: AVRG=20.
2021/01/22 16:01:18.1 Spectrometer EOS_B0808: AVRG=20.
2021/01/22 16:01:18.1 Spectrometer EOS_B0733: AVRG=20.
2021/01/22 16:01:18.1 Spectrometer EOS_B0817: AVRG=20.
2021/01/22 16:01:18.1 StartShaker(500)
2021/01/22 16:01:18.2 Old State:6-ENGN_StateRunning_StartNewStep New State:ENGN_StateRunning_WaitGoingToSample
2021/01/22 16:01:18.2 Sample96_Q(0, 0, 11)
2021/01/22 16:01:18.2 
2021/01/22 16:01:18.2 START SCAN: 
2021/01/22 16:01:18.2    CycleTime (ms)    = 200
2021/01/22 16:01:18.2    CycleCount        = 25
2021/01/22 16:01:18.2    ScansToAverage    = 20
2021/01/22 16:01:18.2    FinishTime (s)    = 5
2021/01/22 16:01:18.2    Temperature       = 25.0
2021/01/22 16:01:18.2    Lamp Power        = 31
2021/01/22 16:01:18.2    Spec PCB temps    = 42.2, 41.9, 42.7, 42.4, 42.2, 41.9, 40.8, 41.1
2021/01/22 16:01:18.2 
2021/01/22 16:01:18.3 Old State:12-ENGN_StateRunning_WaitGoingToSample New State:ENGN_StateRunning_GoingToSample
2021/01/22 16:01:18.3 Temp: 2021/01/22 16:01:18, 25.0000, 25.4000
2021/01/22 16:01:18.3 GMC_SetPosZ
2021/01/22 16:01:22.1 GMC_ServoHere
2021/01/22 16:01:22.3 GMC_SetPosXY
2021/01/22 16:01:23.7 GMC_SetPosZ
2021/01/22 16:01:29.8 Shaker current speed = 494 rpm, set speed = 500
2021/01/22 16:01:29.9 Old State:13-ENGN_StateRunning_GoingToSample New State:ENGN_StateRunning_GoingToSampleComplete
2021/01/22 16:01:30.0 Old State:14-cycle=1, Elapsed Time:    0.0,  Ideal Time:    0.0,  Delay time =   0.00  New State:ENGN_StateRunning_WaitSampling
2021/01/22 16:01:30.0 Step started at: 16:01:18
2021/01/22 16:01:30.0 XYZ movement time  = 11575 ms + Scan start delay = 11522 ms
2021/01/22 16:01:30.1 Old State:15-ENGN_StateRunning_WaitSampling New State:ENGN_StateRunning_Sampling
2021/01/22 16:01:34.7 
2021/01/22 16:01:34.7 STOP SCAN: 
2021/01/22 16:01:34.7    CycleTime (ms)    = 200
2021/01/22 16:01:34.7    CurrentCycle      = 25
2021/01/22 16:01:34.7    RunTime (ms)      = 4957
2021/01/22 16:01:34.7    AverageCycle (ms) = 198.3
2021/01/22 16:01:34.7    Temperature       = 25.0
2021/01/22 16:01:34.7    DATA OVERRUN %    = 0.00
2021/01/22 16:01:34.7    Spec PCB temps    = 42.2, 41.8, 42.6, 42.4, 42.1, 41.9, 40.8, 41.1
2021/01/22 16:01:34.7 
2021/01/22 16:01:34.7 Old State:16-cycle=25, Elapsed Time:    5.0,  Ideal Time:    4.8,  Delay time =   0.04  New State:ENGN_StateRunning_SamplingComplete
2021/01/22 16:01:34.8 Old State:18-ENGN_StateRunning_SamplingComplete New State:ENGN_StateRunning_WaitNewStep
2021/01/22 16:01:35.2 Old State:5-ENGN_StateRunning_WaitNewStep New State:ENGN_StateRunning_StartNewStep
2021/01/22 16:01:35.2 Spectrometer EOS_B0712: AVRG=20.
2021/01/22 16:01:35.2 Spectrometer EOS_B0734: AVRG=20.
2021/01/22 16:01:35.2 Spectrometer EOS_B0814: AVRG=20.
2021/01/22 16:01:35.2 Spectrometer EOS_B0735: AVRG=20.
2021/01/22 16:01:35.2 Spectrometer EOS_B0732: AVRG=20.
2021/01/22 16:01:35.2 Spectrometer EOS_B0808: AVRG=20.
2021/01/22 16:01:35.2 Spectrometer EOS_B0733: AVRG=20.
2021/01/22 16:01:35.2 Spectrometer EOS_B0817: AVRG=20.
2021/01/22 16:01:35.2 StartShaker(500)
2021/01/22 16:01:35.3 Old State:6-ENGN_StateRunning_StartNewStep New State:ENGN_StateRunning_WaitGoingToSample
2021/01/22 16:01:35.3 Sample96_Q(0, 0, 10)
2021/01/22 16:01:35.3 GMC_SetPosZ
2021/01/22 16:01:35.3 
2021/01/22 16:01:35.3 START SCAN: 
2021/01/22 16:01:35.3    CycleTime (ms)    = 200
2021/01/22 16:01:35.3    CycleCount        = 25
2021/01/22 16:01:35.3    ScansToAverage    = 20
2021/01/22 16:01:35.3    FinishTime (s)    = 5
2021/01/22 16:01:35.3    Temperature       = 25.0
2021/01/22 16:01:35.3    Lamp Power        = 31
2021/01/22 16:01:35.3    Spec PCB temps    = 42.2, 41.8, 42.6, 42.4, 42.1, 41.9, 40.8, 41.1
2021/01/22 16:01:35.3 
2021/01/22 16:01:35.4 Old State:12-ENGN_StateRunning_WaitGoingToSample New State:ENGN_StateRunning_GoingToSample
2021/01/22 16:01:35.5 Temp: 2021/01/22 16:01:35, 25.0000, 25.4000
2021/01/22 16:01:41.4 GMC_ServoHere
2021/01/22 16:01:41.5 GMC_SetPosXY
2021/01/22 16:01:41.9 GMC_SetPosZ
2021/01/22 16:01:47.8 Shaker current speed = 494 rpm, set speed = 500
2021/01/22 16:01:47.9 Old State:13-ENGN_StateRunning_GoingToSample New State:ENGN_StateRunning_GoingToSampleComplete
2021/01/22 16:01:48.0 Old State:14-cycle=1, Elapsed Time:    0.0,  Ideal Time:    0.0,  Delay time =   0.00  New State:ENGN_StateRunning_WaitSampling
2021/01/22 16:01:48.0 Step started at: 16:01:35
2021/01/22 16:01:48.0 XYZ movement time  = 12499 ms + Scan start delay = 12467 ms
2021/01/22 16:01:48.1 Old State:15-ENGN_StateRunning_WaitSampling New State:ENGN_StateRunning_Sampling
2021/01/22 16:01:52.7 
2021/01/22 16:01:52.7 STOP SCAN: 
2021/01/22 16:01:52.7    CycleTime (ms)    = 200
2021/01/22 16:01:52.7    CurrentCycle      = 25
2021/01/22 16:01:52.7    RunTime (ms)      = 4959
2021/01/22 16:01:52.7    AverageCycle (ms) = 198.4
2021/01/22 16:01:52.7    Temperature       = 25.0
2021/01/22 16:01:52.7    DATA OVERRUN %    = 0.00
2021/01/22 16:01:52.7    Spec PCB temps    = 42.2, 41.8, 42.7, 42.4, 42.1, 41.9, 40.8, 41.1
2021/01/22 16:01:52.7 
2021/01/22 16:01:52.9 Old State:16-cycle=25, Elapsed Time:    5.0,  Ideal Time:    4.8,  Delay time =   0.04  New State:ENGN_StateRunning_SamplingComplete
2021/01/22 16:01:53.0 Old State:18-ENGN_StateRunning_SamplingComplete New State:ENGN_StateRunning_WaitNewStep
2021/01/22 16:01:53.1 Old State:5-ENGN_StateRunning_WaitNewStep New State:ENGN_StateRunning_StartNewStep
2021/01/22 16:01:53.1 Spectrometer EOS_B0712: AVRG=20.
2021/01/22 16:01:53.1 Spectrometer EOS_B0734: AVRG=20.
2021/01/22 16:01:53.1 Spectrometer EOS_B0814: AVRG=20.
2021/01/22 16:01:53.1 Spectrometer EOS_B0735: AVRG=20.
2021/01/22 16:01:53.1 Spectrometer EOS_B0732: AVRG=20.
2021/01/22 16:01:53.1 Spectrometer EOS_B0808: AVRG=20.
2021/01/22 16:01:53.1 Spectrometer EOS_B0733: AVRG=20.
2021/01/22 16:01:53.1 Spectrometer EOS_B0817: AVRG=20.
2021/01/22 16:01:53.1 StartShaker(500)
2021/01/22 16:01:53.3 Old State:6-ENGN_StateRunning_StartNewStep New State:ENGN_StateRunning_WaitGoingToSample
2021/01/22 16:01:53.3 Sample96_Q(0, 0, 11)
2021/01/22 16:01:53.3 
2021/01/22 16:01:53.3 START SCAN: 
2021/01/22 16:01:53.3    CycleTime (ms)    = 200
2021/01/22 16:01:53.3    CycleCount        = 25
2021/01/22 16:01:53.3    ScansToAverage    = 20
2021/01/22 16:01:53.3    FinishTime (s)    = 5
2021/01/22 16:01:53.3    Temperature       = 25.0
2021/01/22 16:01:53.3    Lamp Power        = 31
2021/01/22 16:01:53.3    Spec PCB temps    = 42.2, 41.8, 42.7, 42.4, 42.1, 41.9, 40.8, 41.1
2021/01/22 16:01:53.3 
2021/01/22 16:01:53.4 Old State:12-ENGN_StateRunning_WaitGoingToSample New State:ENGN_StateRunning_GoingToSample
2021/01/22 16:01:53.4 GMC_SetPosZ
2021/01/22 16:01:53.4 Temp: 2021/01/22 16:01:53, 25.0000, 25.4000
2021/01/22 16:01:59.1 GMC_ServoHere
2021/01/22 16:01:59.3 GMC_SetPosXY
2021/01/22 16:01:59.6 GMC_SetPosZ
2021/01/22 16:02:05.9 Shaker current speed = 494 rpm, set speed = 500
2021/01/22 16:02:06.0 Old State:13-ENGN_StateRunning_GoingToSample New State:ENGN_StateRunning_GoingToSampleComplete
2021/01/22 16:02:06.1 Old State:14-cycle=1, Elapsed Time:    0.0,  Ideal Time:    0.0,  Delay time =   0.00  New State:ENGN_StateRunning_WaitSampling
2021/01/22 16:02:06.1 Step started at: 16:01:53
2021/01/22 16:02:06.1 XYZ movement time  = 12594 ms + Scan start delay = 12563 ms
2021/01/22 16:02:06.2 Old State:15-ENGN_StateRunning_WaitSampling New State:ENGN_StateRunning_Sampling
2021/01/22 16:02:10.8 
2021/01/22 16:02:10.8 STOP SCAN: 
2021/01/22 16:02:10.8    CycleTime (ms)    = 200
2021/01/22 16:02:10.8    CurrentCycle      = 25
2021/01/22 16:02:10.8    RunTime (ms)      = 4956
2021/01/22 16:02:10.8    AverageCycle (ms) = 198.2
2021/01/22 16:02:10.8    Temperature       = 25.0
2021/01/22 16:02:10.8    DATA OVERRUN %    = 0.00
2021/01/22 16:02:10.8    Spec PCB temps    = 42.2, 41.8, 42.7, 42.4, 42.2, 41.9, 40.8, 41.1
2021/01/22 16:02:10.8 
2021/01/22 16:02:10.9 Old State:16-cycle=25, Elapsed Time:    5.0,  Ideal Time:    4.8,  Delay time =   0.04  New State:ENGN_StateRunning_SamplingComplete
2021/01/22 16:02:11.0 Old State:18-ENGN_StateRunning_SamplingComplete New State:ENGN_StateRunning_WaitNewStep
2021/01/22 16:02:11.3 Old State:5-ENGN_StateRunning_WaitNewStep New State:ENGN_StateRunning_StartNewStep
2021/01/22 16:02:11.3 Spectrometer EOS_B0712: AVRG=20.
2021/01/22 16:02:11.3 Spectrometer EOS_B0734: AVRG=20.
2021/01/22 16:02:11.3 Spectrometer EOS_B0814: AVRG=20.
2021/01/22 16:02:11.3 Spectrometer EOS_B0735: AVRG=20.
2021/01/22 16:02:11.3 Spectrometer EOS_B0732: AVRG=20.
2021/01/22 16:02:11.3 Spectrometer EOS_B0808: AVRG=20.
2021/01/22 16:02:11.3 Spectrometer EOS_B0733: AVRG=20.
2021/01/22 16:02:11.3 Spectrometer EOS_B0817: AVRG=20.
2021/01/22 16:02:11.3 StartShaker(500)
2021/01/22 16:02:11.4 Old State:6-ENGN_StateRunning_StartNewStep New State:ENGN_StateRunning_WaitGoingToSample
2021/01/22 16:02:11.4 Sample96_Q(0, 0, 10)
2021/01/22 16:02:11.4 GMC_SetPosZ
2021/01/22 16:02:11.4 
2021/01/22 16:02:11.4 START SCAN: 
2021/01/22 16:02:11.4    CycleTime (ms)    = 200
2021/01/22 16:02:11.4    CycleCount        = 25
2021/01/22 16:02:11.4    ScansToAverage    = 20
2021/01/22 16:02:11.4    FinishTime (s)    = 5
2021/01/22 16:02:11.4    Temperature       = 25.0
2021/01/22 16:02:11.4    Lamp Power        = 31
2021/01/22 16:02:11.4    Spec PCB temps    = 42.2, 41.8, 42.7, 42.4, 42.2, 41.9, 40.8, 41.1
2021/01/22 16:02:11.4 
2021/01/22 16:02:11.5 Old State:12-ENGN_StateRunning_WaitGoingToSample New State:ENGN_StateRunning_GoingToSample
2021/01/22 16:02:11.6 Temp: 2021/01/22 16:02:11, 25.0000, 25.4000
2021/01/22 16:02:17.5 GMC_ServoHere
2021/01/22 16:02:17.6 GMC_SetPosXY
2021/01/22 16:02:18.0 GMC_SetPosZ
2021/01/22 16:02:23.8 Shaker current speed = 494 rpm, set speed = 500
2021/01/22 16:02:23.9 Old State:13-cycle=1, Elapsed Time:    0.0,  Ideal Time:    0.0,  Delay time =   0.00  New State:ENGN_StateRunning_GoingToSampleComplete
2021/01/22 16:02:24.0 Old State:14-ENGN_StateRunning_GoingToSampleComplete New State:ENGN_StateRunning_WaitSampling
2021/01/22 16:02:24.0 Step started at: 16:02:11
2021/01/22 16:02:24.0 XYZ movement time  = 12380 ms + Scan start delay = 12310 ms
2021/01/22 16:02:24.1 Old State:15-ENGN_StateRunning_WaitSampling New State:ENGN_StateRunning_Sampling
2021/01/22 16:02:28.7 
2021/01/22 16:02:28.7 STOP SCAN: 
2021/01/22 16:02:28.7    CycleTime (ms)    = 200
2021/01/22 16:02:28.7    CurrentCycle      = 25
2021/01/22 16:02:28.7    RunTime (ms)      = 4958
2021/01/22 16:02:28.7    AverageCycle (ms) = 198.3
2021/01/22 16:02:28.7    Temperature       = 25.0
2021/01/22 16:02:28.7    DATA OVERRUN %    = 0.00
2021/01/22 16:02:28.7    Spec PCB temps    = 42.2, 41.8, 42.6, 42.4, 42.1, 41.9, 40.8, 41.1
2021/01/22 16:02:28.7 
2021/01/22 16:02:28.7 Old State:16-cycle=25, Elapsed Time:    5.0,  Ideal Time:    4.8,  Delay time =   0.04  New State:ENGN_StateRunning_SamplingComplete
2021/01/22 16:02:28.9 Old State:18-ENGN_StateRunning_SamplingComplete New State:ENGN_StateRunning_WaitNewStep
2021/01/22 16:02:29.2 Old State:5-ENGN_StateRunning_WaitNewStep New State:ENGN_StateRunning_StartNewStep
2021/01/22 16:02:29.2 Spectrometer EOS_B0712: AVRG=20.
2021/01/22 16:02:29.2 Spectrometer EOS_B0734: AVRG=20.
2021/01/22 16:02:29.2 Spectrometer EOS_B0814: AVRG=20.
2021/01/22 16:02:29.2 Spectrometer EOS_B0735: AVRG=20.
2021/01/22 16:02:29.2 Spectrometer EOS_B0732: AVRG=20.
2021/01/22 16:02:29.2 Spectrometer EOS_B0808: AVRG=20.
2021/01/22 16:02:29.2 Spectrometer EOS_B0733: AVRG=20.
2021/01/22 16:02:29.2 Spectrometer EOS_B0817: AVRG=20.
2021/01/22 16:02:29.2 StartShaker(500)
2021/01/22 16:02:29.3 Old State:6-ENGN_StateRunning_StartNewStep New State:ENGN_StateRunning_WaitGoingToSample
2021/01/22 16:02:29.3 Sample96_Q(0, 0, 11)
2021/01/22 16:02:29.3 
2021/01/22 16:02:29.3 START SCAN: 
2021/01/22 16:02:29.3    CycleTime (ms)    = 200
2021/01/22 16:02:29.3    CycleCount        = 25
2021/01/22 16:02:29.3    ScansToAverage    = 20
2021/01/22 16:02:29.3    FinishTime (s)    = 5
2021/01/22 16:02:29.3    Temperature       = 25.0
2021/01/22 16:02:29.3    Lamp Power        = 31
2021/01/22 16:02:29.3    Spec PCB temps    = 42.2, 41.8, 42.6, 42.4, 42.1, 41.9, 40.8, 41.1
2021/01/22 16:02:29.3 
2021/01/22 16:02:29.4 Old State:12-ENGN_StateRunning_WaitGoingToSample New State:ENGN_StateRunning_GoingToSample
2021/01/22 16:02:29.5 Temp: 2021/01/22 16:02:29, 25.0000, 25.4000
2021/01/22 16:02:29.5 GMC_SetPosZ
2021/01/22 16:02:35.2 GMC_ServoHere
2021/01/22 16:02:35.4 GMC_SetPosXY
2021/01/22 16:02:35.7 GMC_SetPosZ
2021/01/22 16:02:41.8 Shaker current speed = 494 rpm, set speed = 500
2021/01/22 16:02:41.9 Old State:13-ENGN_StateRunning_GoingToSample New State:ENGN_StateRunning_GoingToSampleComplete
2021/01/22 16:02:42.0 Old State:14-cycle=1, Elapsed Time:    0.0,  Ideal Time:    0.0,  Delay time =   0.00  New State:ENGN_StateRunning_WaitSampling
2021/01/22 16:02:42.0 Step started at: 16:02:29
2021/01/22 16:02:42.0 XYZ movement time  = 12483 ms + Scan start delay = 12478 ms
2021/01/22 16:02:42.1 Old State:15-ENGN_StateRunning_WaitSampling New State:ENGN_StateRunning_Sampling
2021/01/22 16:02:46.8 
2021/01/22 16:02:46.8 STOP SCAN: 
2021/01/22 16:02:46.8    CycleTime (ms)    = 200
2021/01/22 16:02:46.8    CurrentCycle      = 25
2021/01/22 16:02:46.8    RunTime (ms)      = 4959
2021/01/22 16:02:46.8    AverageCycle (ms) = 198.4
2021/01/22 16:02:46.8    Temperature       = 25.0
2021/01/22 16:02:46.8    DATA OVERRUN %    = 0.00
2021/01/22 16:02:46.8    Spec PCB temps    = 42.2, 41.8, 42.7, 42.4, 42.1, 41.9, 40.8, 41.1
2021/01/22 16:02:46.8 
2021/01/22 16:02:46.8 Old State:16-cycle=25, Elapsed Time:    5.0,  Ideal Time:    4.8,  Delay time =   0.04  New State:ENGN_StateRunning_SamplingComplete
2021/01/22 16:02:47.0 Old State:18-ENGN_StateRunning_SamplingComplete New State:ENGN_StateRunning_WaitNewStep
2021/01/22 16:02:47.1 Old State:5-ENGN_StateRunning_WaitNewStep New State:ENGN_StateRunning_StartNewStep
2021/01/22 16:02:47.1 Spectrometer EOS_B0712: AVRG=20.
2021/01/22 16:02:47.1 Spectrometer EOS_B0734: AVRG=20.
2021/01/22 16:02:47.1 Spectrometer EOS_B0814: AVRG=20.
2021/01/22 16:02:47.1 Spectrometer EOS_B0735: AVRG=20.
2021/01/22 16:02:47.1 Spectrometer EOS_B0732: AVRG=20.
2021/01/22 16:02:47.1 Spectrometer EOS_B0808: AVRG=20.
2021/01/22 16:02:47.1 Spectrometer EOS_B0733: AVRG=20.
2021/01/22 16:02:47.1 Spectrometer EOS_B0817: AVRG=20.
2021/01/22 16:02:47.1 StartShaker(500)
2021/01/22 16:02:47.2 Old State:6-ENGN_StateRunning_StartNewStep New State:ENGN_StateRunning_WaitGoingToSample
2021/01/22 16:02:47.2 Sample96_Q(0, 0, 10)
2021/01/22 16:02:47.2 GMC_SetPosZ
2021/01/22 16:02:47.3 
2021/01/22 16:02:47.3 START SCAN: 
2021/01/22 16:02:47.3    CycleTime (ms)    = 200
2021/01/22 16:02:47.3    CycleCount        = 25
2021/01/22 16:02:47.3    ScansToAverage    = 20
2021/01/22 16:02:47.3    FinishTime (s)    = 5
2021/01/22 16:02:47.3    Temperature       = 25.0
2021/01/22 16:02:47.3    Lamp Power        = 31
2021/01/22 16:02:47.3    Spec PCB temps    = 42.2, 41.8, 42.7, 42.4, 42.1, 41.9, 40.8, 41.1
2021/01/22 16:02:47.3 
2021/01/22 16:02:47.3 Old State:12-ENGN_StateRunning_WaitGoingToSample New State:ENGN_StateRunning_GoingToSample
2021/01/22 16:02:47.3 Temp: 2021/01/22 16:02:47, 25.0000, 25.4000
2021/01/22 16:02:53.3 GMC_ServoHere
2021/01/22 16:02:53.5 GMC_SetPosXY
2021/01/22 16:02:53.8 GMC_SetPosZ
2021/01/22 16:02:59.7 Shaker current speed = 494 rpm, set speed = 500
2021/01/22 16:02:59.8 Old State:13-cycle=1, Elapsed Time:    0.0,  Ideal Time:    0.0,  Delay time =   0.00  New State:ENGN_StateRunning_GoingToSampleComplete
2021/01/22 16:02:59.9 Old State:14-ENGN_StateRunning_GoingToSampleComplete New State:ENGN_StateRunning_WaitSampling
2021/01/22 16:02:59.9 Step started at: 16:02:47
2021/01/22 16:02:59.9 XYZ movement time  = 12426 ms + Scan start delay = 12329 ms
2021/01/22 16:03:00.0 Old State:15-ENGN_StateRunning_WaitSampling New State:ENGN_StateRunning_Sampling
2021/01/22 16:03:04.5 
2021/01/22 16:03:04.5 STOP SCAN: 
2021/01/22 16:03:04.5    CycleTime (ms)    = 200
2021/01/22 16:03:04.5    CurrentCycle      = 25
2021/01/22 16:03:04.5    RunTime (ms)      = 4956
2021/01/22 16:03:04.5    AverageCycle (ms) = 198.2
2021/01/22 16:03:04.5    Temperature       = 25.0
2021/01/22 16:03:04.5    DATA OVERRUN %    = 0.00
2021/01/22 16:03:04.5    Spec PCB temps    = 42.2, 41.9, 42.7, 42.4, 42.2, 41.9, 40.8, 41.1
2021/01/22 16:03:04.5 
2021/01/22 16:03:04.6 Old State:16-cycle=25, Elapsed Time:    5.0,  Ideal Time:    4.8,  Delay time =   0.04  New State:ENGN_StateRunning_SamplingComplete
2021/01/22 16:03:04.7 Old State:18-ENGN_StateRunning_SamplingComplete New State:ENGN_StateRunning_WaitNewStep
2021/01/22 16:03:04.9 Old State:5-ENGN_StateRunning_WaitNewStep New State:ENGN_StateRunning_StartNewStep
2021/01/22 16:03:05.0 Spectrometer EOS_B0712: AVRG=20.
2021/01/22 16:03:05.0 Spectrometer EOS_B0734: AVRG=20.
2021/01/22 16:03:05.0 Spectrometer EOS_B0814: AVRG=20.
2021/01/22 16:03:05.0 Spectrometer EOS_B0735: AVRG=20.
2021/01/22 16:03:05.0 Spectrometer EOS_B0732: AVRG=20.
2021/01/22 16:03:05.0 Spectrometer EOS_B0808: AVRG=20.
2021/01/22 16:03:05.0 Spectrometer EOS_B0733: AVRG=20.
2021/01/22 16:03:05.0 Spectrometer EOS_B0817: AVRG=20.
2021/01/22 16:03:05.0 StartShaker(500)
2021/01/22 16:03:05.1 Old State:6-ENGN_StateRunning_StartNewStep New State:ENGN_StateRunning_WaitGoingToSample
2021/01/22 16:03:05.1 Sample96_Q(0, 0, 3)
2021/01/22 16:03:05.1 
2021/01/22 16:03:05.1 START SCAN: 
2021/01/22 16:03:05.1    CycleTime (ms)    = 200
2021/01/22 16:03:05.1    CycleCount        = 1500
2021/01/22 16:03:05.1    ScansToAverage    = 20
2021/01/22 16:03:05.1    FinishTime (s)    = 300
2021/01/22 16:03:05.1    Temperature       = 25.0
2021/01/22 16:03:05.1    Lamp Power        = 31
2021/01/22 16:03:05.1    Spec PCB temps    = 42.2, 41.9, 42.7, 42.4, 42.2, 41.9, 40.8, 41.1
2021/01/22 16:03:05.1 
2021/01/22 16:03:05.2 GMC_SetPosZ
2021/01/22 16:03:05.2 Old State:12-ENGN_StateRunning_WaitGoingToSample New State:ENGN_StateRunning_GoingToSample
2021/01/22 16:03:05.3 Temp: 2021/01/22 16:03:05, 25.0000, 25.4000
2021/01/22 16:03:10.9 GMC_ServoHere
2021/01/22 16:03:11.1 GMC_SetPosXY
2021/01/22 16:03:12.3 GMC_SetPosZ
2021/01/22 16:03:16.3 Shaker current speed = 494 rpm, set speed = 500
2021/01/22 16:03:16.4 Old State:13-ENGN_StateRunning_GoingToSample New State:ENGN_StateRunning_GoingToSampleComplete
2021/01/22 16:03:16.5 Old State:14-cycle=1, Elapsed Time:    0.0,  Ideal Time:    0.0,  Delay time =   0.00  New State:ENGN_StateRunning_WaitSampling
2021/01/22 16:03:16.5 Step started at: 16:03:05
2021/01/22 16:03:16.5 XYZ movement time  = 11189 ms + Scan start delay = 11162 ms
2021/01/22 16:03:16.6 Old State:15-ENGN_StateRunning_WaitSampling New State:ENGN_StateRunning_Sampling
2021/01/22 16:04:05.4 Temp: 2021/01/22 16:04:05, 25.0000, 25.4000
2021/01/22 16:05:05.6 Temp: 2021/01/22 16:05:05, 25.0000, 25.4000
2021/01/22 16:06:05.7 Temp: 2021/01/22 16:06:05, 25.0000, 25.4000
2021/01/22 16:07:05.9 Temp: 2021/01/22 16:07:05, 25.0000, 25.4000
2021/01/22 16:08:06.0 Temp: 2021/01/22 16:08:06, 25.0000, 25.4000
2021/01/22 16:08:16.2 
2021/01/22 16:08:16.2 STOP SCAN: 
2021/01/22 16:08:16.2    CycleTime (ms)    = 200
2021/01/22 16:08:16.2    CurrentCycle      = 1500
2021/01/22 16:08:16.2    RunTime (ms)      = 299959
2021/01/22 16:08:16.2    AverageCycle (ms) = 200.0
2021/01/22 16:08:16.2    Temperature       = 25.0
2021/01/22 16:08:16.2    DATA OVERRUN %    = 0.00
2021/01/22 16:08:16.2    Spec PCB temps    = 42.3, 41.9, 42.7, 42.5, 42.2, 41.9, 40.8, 41.2
2021/01/22 16:08:16.2 
2021/01/22 16:08:16.3 Old State:16-cycle=1500, Elapsed Time:  300.0,  Ideal Time:  299.8,  Delay time =   0.04  New State:ENGN_StateRunning_SamplingComplete
2021/01/22 16:08:16.5 Old State:18-ENGN_StateRunning_SamplingComplete New State:ENGN_StateRunning_WaitNewStep
2021/01/22 16:08:16.7 Old State:5-ENGN_StateRunning_WaitNewStep New State:ENGN_StateRunning_StartNewStep
2021/01/22 16:08:16.7 Spectrometer EOS_B0712: AVRG=20.
2021/01/22 16:08:16.7 Spectrometer EOS_B0734: AVRG=20.
2021/01/22 16:08:16.7 Spectrometer EOS_B0814: AVRG=20.
2021/01/22 16:08:16.7 Spectrometer EOS_B0735: AVRG=20.
2021/01/22 16:08:16.7 Spectrometer EOS_B0732: AVRG=20.
2021/01/22 16:08:16.7 Spectrometer EOS_B0808: AVRG=20.
2021/01/22 16:08:16.7 Spectrometer EOS_B0733: AVRG=20.
2021/01/22 16:08:16.7 Spectrometer EOS_B0817: AVRG=20.
2021/01/22 16:08:16.7 StartShaker(500)
2021/01/22 16:08:16.8 Old State:6-ENGN_StateRunning_StartNewStep New State:ENGN_StateRunning_WaitGoingToSample
2021/01/22 16:08:16.8 Sample96_Q(0, 0, 4)
2021/01/22 16:08:16.8 
2021/01/22 16:08:16.8 START SCAN: 
2021/01/22 16:08:16.8    CycleTime (ms)    = 200
2021/01/22 16:08:16.8    CycleCount        = 250
2021/01/22 16:08:16.8    ScansToAverage    = 20
2021/01/22 16:08:16.8    FinishTime (s)    = 50
2021/01/22 16:08:16.8    Temperature       = 25.0
2021/01/22 16:08:16.8    Lamp Power        = 31
2021/01/22 16:08:16.8    Spec PCB temps    = 42.2, 41.9, 42.7, 42.5, 42.2, 41.9, 40.8, 41.2
2021/01/22 16:08:16.8 
2021/01/22 16:08:16.9 Old State:12-ENGN_StateRunning_WaitGoingToSample New State:ENGN_StateRunning_GoingToSample
2021/01/22 16:08:16.9 GMC_SetPosZ
2021/01/22 16:08:17.0 Temp: 2021/01/22 16:08:17, 25.0000, 25.4000
2021/01/22 16:08:20.7 GMC_ServoHere
2021/01/22 16:08:20.9 GMC_SetPosXY
2021/01/22 16:08:21.2 GMC_SetPosZ
2021/01/22 16:08:25.5 Old State:13-ENGN_StateRunning_GoingToSample New State:ENGN_StateRunning_GoingToSampleComplete
2021/01/22 16:08:25.6 Old State:14-cycle=1, Elapsed Time:    0.0,  Ideal Time:    0.0,  Delay time =   0.00  New State:ENGN_StateRunning_WaitSampling
2021/01/22 16:08:25.6 Step started at: 16:08:16
2021/01/22 16:08:25.6 XYZ movement time  = 8557 ms + Scan start delay = 8526 ms
2021/01/22 16:08:25.7 Old State:15-ENGN_StateRunning_WaitSampling New State:ENGN_StateRunning_Sampling
2021/01/22 16:08:26.8 Shaker current speed = 494 rpm, set speed = 500
2021/01/22 16:09:15.3 
2021/01/22 16:09:15.3 STOP SCAN: 
2021/01/22 16:09:15.3    CycleTime (ms)    = 200
2021/01/22 16:09:15.3    CurrentCycle      = 250
2021/01/22 16:09:15.3    RunTime (ms)      = 49957
2021/01/22 16:09:15.3    AverageCycle (ms) = 199.8
2021/01/22 16:09:15.3    Temperature       = 25.0
2021/01/22 16:09:15.3    DATA OVERRUN %    = 0.00
2021/01/22 16:09:15.3    Spec PCB temps    = 42.3, 41.9, 42.7, 42.4, 42.1, 41.9, 40.8, 41.1
2021/01/22 16:09:15.3 
2021/01/22 16:09:15.4 Old State:16-cycle=250, Elapsed Time:   50.0,  Ideal Time:   49.8,  Delay time =   0.04  New State:ENGN_StateRunning_SamplingComplete
2021/01/22 16:09:15.5 Old State:18-ENGN_StateRunning_SamplingComplete New State:ENGN_StateRunning_WaitNewStep
2021/01/22 16:09:15.8 Old State:5-ENGN_StateRunning_WaitNewStep New State:ENGN_StateRunning_StartNewStep
2021/01/22 16:09:15.8 Spectrometer EOS_B0712: AVRG=20.
2021/01/22 16:09:15.8 Spectrometer EOS_B0734: AVRG=20.
2021/01/22 16:09:15.9 Spectrometer EOS_B0814: AVRG=20.
2021/01/22 16:09:15.9 Spectrometer EOS_B0735: AVRG=20.
2021/01/22 16:09:15.9 Spectrometer EOS_B0732: AVRG=20.
2021/01/22 16:09:15.9 Spectrometer EOS_B0808: AVRG=20.
2021/01/22 16:09:15.9 Spectrometer EOS_B0733: AVRG=20.
2021/01/22 16:09:15.9 Spectrometer EOS_B0817: AVRG=20.
2021/01/22 16:09:15.9 StartShaker(500)
2021/01/22 16:09:16.0 Old State:6-ENGN_StateRunning_StartNewStep New State:ENGN_StateRunning_WaitGoingToSample
2021/01/22 16:09:16.0 Sample96_Q(0, 0, 3)
2021/01/22 16:09:16.0 GMC_SetPosZ
2021/01/22 16:09:16.0 
2021/01/22 16:09:16.0 START SCAN: 
2021/01/22 16:09:16.0    CycleTime (ms)    = 200
2021/01/22 16:09:16.0    CycleCount        = 1500
2021/01/22 16:09:16.0    ScansToAverage    = 20
2021/01/22 16:09:16.0    FinishTime (s)    = 300
2021/01/22 16:09:16.0    Temperature       = 25.0
2021/01/22 16:09:16.0    Lamp Power        = 31
2021/01/22 16:09:16.0    Spec PCB temps    = 42.3, 41.9, 42.7, 42.4, 42.1, 41.9, 40.8, 41.1
2021/01/22 16:09:16.0 
2021/01/22 16:09:16.1 Old State:12-ENGN_StateRunning_WaitGoingToSample New State:ENGN_StateRunning_GoingToSample
2021/01/22 16:09:16.1 Temp: 2021/01/22 16:09:16, 25.0000, 25.4000
2021/01/22 16:09:20.1 GMC_ServoHere
2021/01/22 16:09:20.2 GMC_SetPosXY
2021/01/22 16:09:20.5 GMC_SetPosZ
2021/01/22 16:09:24.5 Old State:13-ENGN_StateRunning_GoingToSample New State:ENGN_StateRunning_GoingToSampleComplete
2021/01/22 16:09:24.6 Old State:14-cycle=1, Elapsed Time:    0.0,  Ideal Time:    0.0,  Delay time =   0.00  New State:ENGN_StateRunning_WaitSampling
2021/01/22 16:09:24.6 Step started at: 16:09:16
2021/01/22 16:09:24.6 XYZ movement time  = 8408 ms + Scan start delay = 8379 ms
2021/01/22 16:09:24.7 Old State:15-ENGN_StateRunning_WaitSampling New State:ENGN_StateRunning_Sampling
2021/01/22 16:09:25.9 Shaker current speed = 494 rpm, set speed = 500
2021/01/22 16:10:16.3 Temp: 2021/01/22 16:10:16, 25.0000, 25.4000
2021/01/22 16:11:16.5 Temp: 2021/01/22 16:11:16, 25.0000, 25.4000
2021/01/22 16:12:16.6 Temp: 2021/01/22 16:12:16, 25.0000, 25.4000
2021/01/22 16:13:16.8 Temp: 2021/01/22 16:13:16, 25.0000, 25.4000
2021/01/22 16:14:16.9 Temp: 2021/01/22 16:14:16, 25.0000, 25.4000
2021/01/22 16:14:24.3 
2021/01/22 16:14:24.3 STOP SCAN: 
2021/01/22 16:14:24.3    CycleTime (ms)    = 200
2021/01/22 16:14:24.3    CurrentCycle      = 1500
2021/01/22 16:14:24.3    RunTime (ms)      = 299958
2021/01/22 16:14:24.3    AverageCycle (ms) = 200.0
2021/01/22 16:14:24.3    Temperature       = 25.0
2021/01/22 16:14:24.3    DATA OVERRUN %    = 0.00
2021/01/22 16:14:24.3    Spec PCB temps    = 42.3, 41.9, 42.8, 42.4, 42.2, 41.9, 40.8, 41.2
2021/01/22 16:14:24.3 
2021/01/22 16:14:24.4 Old State:16-cycle=1500, Elapsed Time:  300.0,  Ideal Time:  299.8,  Delay time =   0.04  New State:ENGN_StateRunning_SamplingComplete
2021/01/22 16:14:24.5 Old State:18-ENGN_StateRunning_SamplingComplete New State:ENGN_StateRunning_WaitNewStep
2021/01/22 16:14:24.9 Old State:5-ENGN_StateRunning_WaitNewStep New State:ENGN_StateRunning_StartNewStep
2021/01/22 16:14:25.0 Spectrometer EOS_B0712: AVRG=20.
2021/01/22 16:14:25.0 Spectrometer EOS_B0734: AVRG=20.
2021/01/22 16:14:25.0 Spectrometer EOS_B0814: AVRG=20.
2021/01/22 16:14:25.0 Spectrometer EOS_B0735: AVRG=20.
2021/01/22 16:14:25.0 Spectrometer EOS_B0732: AVRG=20.
2021/01/22 16:14:25.0 Spectrometer EOS_B0808: AVRG=20.
2021/01/22 16:14:25.0 Spectrometer EOS_B0733: AVRG=20.
2021/01/22 16:14:25.0 Spectrometer EOS_B0817: AVRG=20.
2021/01/22 16:14:25.0 StartShaker(500)
2021/01/22 16:14:25.1 Old State:6-ENGN_StateRunning_StartNewStep New State:ENGN_StateRunning_WaitGoingToSample
2021/01/22 16:14:25.1 Sample96_Q(0, 0, 11)
2021/01/22 16:14:25.1 
2021/01/22 16:14:25.1 START SCAN: 
2021/01/22 16:14:25.1    CycleTime (ms)    = 200
2021/01/22 16:14:25.1    CycleCount        = 25
2021/01/22 16:14:25.1    ScansToAverage    = 20
2021/01/22 16:14:25.1    FinishTime (s)    = 5
2021/01/22 16:14:25.1    Temperature       = 25.0
2021/01/22 16:14:25.1    Lamp Power        = 31
2021/01/22 16:14:25.1    Spec PCB temps    = 42.3, 41.9, 42.8, 42.4, 42.2, 41.9, 40.9, 41.2
2021/01/22 16:14:25.1 
2021/01/22 16:14:25.2 Temp: 2021/01/22 16:14:25, 25.0000, 25.4000
2021/01/22 16:14:25.2 Old State:12-ENGN_StateRunning_WaitGoingToSample New State:ENGN_StateRunning_GoingToSample
2021/01/22 16:14:25.2 GMC_SetPosZ
2021/01/22 16:14:29.0 GMC_ServoHere
2021/01/22 16:14:29.2 GMC_SetPosXY
2021/01/22 16:14:30.6 GMC_SetPosZ
2021/01/22 16:14:36.9 Old State:13-ENGN_StateRunning_GoingToSample New State:ENGN_StateRunning_GoingToSampleComplete
2021/01/22 16:14:36.9 Shaker current speed = 494 rpm, set speed = 500
2021/01/22 16:14:37.0 Old State:14-cycle=1, Elapsed Time:    0.0,  Ideal Time:    0.0,  Delay time =   0.00  New State:ENGN_StateRunning_WaitSampling
2021/01/22 16:14:37.0 Step started at: 16:14:25
2021/01/22 16:14:37.0 XYZ movement time  = 11694 ms + Scan start delay = 11709 ms
2021/01/22 16:14:37.1 Old State:15-ENGN_StateRunning_WaitSampling New State:ENGN_StateRunning_Sampling
2021/01/22 16:14:41.8 
2021/01/22 16:14:41.8 STOP SCAN: 
2021/01/22 16:14:41.8    CycleTime (ms)    = 200
2021/01/22 16:14:41.8    CurrentCycle      = 25
2021/01/22 16:14:41.8    RunTime (ms)      = 4959
2021/01/22 16:14:41.8    AverageCycle (ms) = 198.4
2021/01/22 16:14:41.8    Temperature       = 25.0
2021/01/22 16:14:41.8    DATA OVERRUN %    = 0.00
2021/01/22 16:14:41.8    Spec PCB temps    = 42.2, 41.9, 42.7, 42.4, 42.2, 42.0, 40.9, 41.2
2021/01/22 16:14:41.8 
2021/01/22 16:14:41.8 Old State:16-cycle=25, Elapsed Time:    5.0,  Ideal Time:    4.8,  Delay time =   0.05  New State:ENGN_StateRunning_SamplingComplete
2021/01/22 16:14:41.9 Old State:18-ENGN_StateRunning_SamplingComplete New State:ENGN_StateRunning_WaitNewStep
2021/01/22 16:14:42.1 Old State:5-ENGN_StateRunning_WaitNewStep New State:ENGN_StateRunning_StartNewStep
2021/01/22 16:14:42.1 Spectrometer EOS_B0712: AVRG=20.
2021/01/22 16:14:42.1 Spectrometer EOS_B0734: AVRG=20.
2021/01/22 16:14:42.1 Spectrometer EOS_B0814: AVRG=20.
2021/01/22 16:14:42.1 Spectrometer EOS_B0735: AVRG=20.
2021/01/22 16:14:42.1 Spectrometer EOS_B0732: AVRG=20.
2021/01/22 16:14:42.1 Spectrometer EOS_B0808: AVRG=20.
2021/01/22 16:14:42.1 Spectrometer EOS_B0733: AVRG=20.
2021/01/22 16:14:42.1 Spectrometer EOS_B0817: AVRG=20.
2021/01/22 16:14:42.1 StartShaker(500)
2021/01/22 16:14:42.2 Old State:6-ENGN_StateRunning_StartNewStep New State:ENGN_StateRunning_WaitGoingToSample
2021/01/22 16:14:42.2 Sample96_Q(0, 0, 10)
2021/01/22 16:14:42.2 GMC_SetPosZ
2021/01/22 16:14:42.2 
2021/01/22 16:14:42.2 START SCAN: 
2021/01/22 16:14:42.2    CycleTime (ms)    = 200
2021/01/22 16:14:42.2    CycleCount        = 25
2021/01/22 16:14:42.2    ScansToAverage    = 20
2021/01/22 16:14:42.2    FinishTime (s)    = 5
2021/01/22 16:14:42.2    Temperature       = 25.0
2021/01/22 16:14:42.2    Lamp Power        = 31
2021/01/22 16:14:42.3    Spec PCB temps    = 42.2, 41.9, 42.7, 42.4, 42.2, 42.0, 40.9, 41.2
2021/01/22 16:14:42.3 
2021/01/22 16:14:42.3 Old State:12-ENGN_StateRunning_WaitGoingToSample New State:ENGN_StateRunning_GoingToSample
2021/01/22 16:14:42.4 Temp: 2021/01/22 16:14:42, 25.0000, 25.5000
2021/01/22 16:14:48.3 GMC_ServoHere
2021/01/22 16:14:48.4 GMC_SetPosXY
2021/01/22 16:14:48.8 GMC_SetPosZ
2021/01/22 16:14:54.6 Shaker current speed = 494 rpm, set speed = 500
2021/01/22 16:14:54.7 Old State:13-cycle=1, Elapsed Time:    0.0,  Ideal Time:    0.0,  Delay time =   0.00  New State:ENGN_StateRunning_GoingToSampleComplete
2021/01/22 16:14:54.8 Old State:14-ENGN_StateRunning_GoingToSampleComplete New State:ENGN_StateRunning_WaitSampling
2021/01/22 16:14:54.8 Step started at: 16:14:42
2021/01/22 16:14:54.8 XYZ movement time  = 12427 ms + Scan start delay = 12330 ms
2021/01/22 16:14:54.9 Old State:15-ENGN_StateRunning_WaitSampling New State:ENGN_StateRunning_Sampling
2021/01/22 16:14:59.5 
2021/01/22 16:14:59.5 STOP SCAN: 
2021/01/22 16:14:59.5    CycleTime (ms)    = 200
2021/01/22 16:14:59.5    CurrentCycle      = 25
2021/01/22 16:14:59.5    RunTime (ms)      = 4959
2021/01/22 16:14:59.5    AverageCycle (ms) = 198.4
2021/01/22 16:14:59.5    Temperature       = 25.0
2021/01/22 16:14:59.5    DATA OVERRUN %    = 0.00
2021/01/22 16:14:59.5    Spec PCB temps    = 42.3, 41.9, 42.7, 42.4, 42.2, 41.9, 40.9, 41.1
2021/01/22 16:14:59.5 
2021/01/22 16:14:59.6 Old State:16-cycle=25, Elapsed Time:    5.0,  Ideal Time:    4.8,  Delay time =   0.04  New State:ENGN_StateRunning_SamplingComplete
2021/01/22 16:14:59.7 Old State:18-ENGN_StateRunning_SamplingComplete New State:ENGN_StateRunning_WaitNewStep
2021/01/22 16:14:59.9 Old State:5-ENGN_StateRunning_WaitNewStep New State:ENGN_StateRunning_StartNewStep
2021/01/22 16:14:59.9 Spectrometer EOS_B0712: AVRG=20.
2021/01/22 16:14:59.9 Spectrometer EOS_B0734: AVRG=20.
2021/01/22 16:14:59.9 Spectrometer EOS_B0814: AVRG=20.
2021/01/22 16:14:59.9 Spectrometer EOS_B0735: AVRG=20.
2021/01/22 16:14:59.9 Spectrometer EOS_B0732: AVRG=20.
2021/01/22 16:14:59.9 Spectrometer EOS_B0808: AVRG=20.
2021/01/22 16:14:59.9 Spectrometer EOS_B0733: AVRG=20.
2021/01/22 16:14:59.9 Spectrometer EOS_B0817: AVRG=20.
2021/01/22 16:14:59.9 StartShaker(500)
2021/01/22 16:15:00.1 Old State:6-ENGN_StateRunning_StartNewStep New State:ENGN_StateRunning_WaitGoingToSample
2021/01/22 16:15:00.1 Sample96_Q(0, 0, 11)
2021/01/22 16:15:00.1 
2021/01/22 16:15:00.1 START SCAN: 
2021/01/22 16:15:00.1    CycleTime (ms)    = 200
2021/01/22 16:15:00.1    CycleCount        = 25
2021/01/22 16:15:00.1    ScansToAverage    = 20
2021/01/22 16:15:00.1    FinishTime (s)    = 5
2021/01/22 16:15:00.1    Temperature       = 25.0
2021/01/22 16:15:00.1    Lamp Power        = 31
2021/01/22 16:15:00.1    Spec PCB temps    = 42.3, 41.9, 42.7, 42.4, 42.2, 42.0, 40.9, 41.1
2021/01/22 16:15:00.1 
2021/01/22 16:15:00.1 GMC_SetPosZ
2021/01/22 16:15:00.2 Old State:12-ENGN_StateRunning_WaitGoingToSample New State:ENGN_StateRunning_GoingToSample
2021/01/22 16:15:00.2 Temp: 2021/01/22 16:15:00, 25.0000, 25.5000
2021/01/22 16:15:05.9 GMC_ServoHere
2021/01/22 16:15:06.0 GMC_SetPosXY
2021/01/22 16:15:06.4 GMC_SetPosZ
2021/01/22 16:15:12.5 Shaker current speed = 494 rpm, set speed = 500
2021/01/22 16:15:12.6 Old State:13-ENGN_StateRunning_GoingToSample New State:ENGN_StateRunning_GoingToSampleComplete
2021/01/22 16:15:12.7 Old State:14-cycle=1, Elapsed Time:    0.0,  Ideal Time:    0.0,  Delay time =   0.00  New State:ENGN_StateRunning_WaitSampling
2021/01/22 16:15:12.7 Step started at: 16:15:00
2021/01/22 16:15:12.7 XYZ movement time  = 12379 ms + Scan start delay = 12360 ms
2021/01/22 16:15:12.8 Old State:15-ENGN_StateRunning_WaitSampling New State:ENGN_StateRunning_Sampling
2021/01/22 16:15:17.4 
2021/01/22 16:15:17.4 STOP SCAN: 
2021/01/22 16:15:17.4    CycleTime (ms)    = 200
2021/01/22 16:15:17.4    CurrentCycle      = 25
2021/01/22 16:15:17.4    RunTime (ms)      = 4959
2021/01/22 16:15:17.4    AverageCycle (ms) = 198.4
2021/01/22 16:15:17.4    Temperature       = 25.0
2021/01/22 16:15:17.4    DATA OVERRUN %    = 0.00
2021/01/22 16:15:17.4    Spec PCB temps    = 42.3, 41.9, 42.7, 42.4, 42.2, 42.0, 40.9, 41.1
2021/01/22 16:15:17.4 
2021/01/22 16:15:17.5 Old State:16-cycle=25, Elapsed Time:    5.0,  Ideal Time:    4.8,  Delay time =   0.04  New State:ENGN_StateRunning_SamplingComplete
2021/01/22 16:15:17.6 Old State:18-ENGN_StateRunning_SamplingComplete New State:ENGN_StateRunning_WaitNewStep
2021/01/22 16:15:17.9 Old State:5-ENGN_StateRunning_WaitNewStep New State:ENGN_StateRunning_StartNewStep
2021/01/22 16:15:17.9 Spectrometer EOS_B0712: AVRG=20.
2021/01/22 16:15:17.9 Spectrometer EOS_B0734: AVRG=20.
2021/01/22 16:15:17.9 Spectrometer EOS_B0814: AVRG=20.
2021/01/22 16:15:17.9 Spectrometer EOS_B0735: AVRG=20.
2021/01/22 16:15:17.9 Spectrometer EOS_B0732: AVRG=20.
2021/01/22 16:15:17.9 Spectrometer EOS_B0808: AVRG=20.
2021/01/22 16:15:17.9 Spectrometer EOS_B0733: AVRG=20.
2021/01/22 16:15:17.9 Spectrometer EOS_B0817: AVRG=20.
2021/01/22 16:15:17.9 StartShaker(500)
2021/01/22 16:15:18.0 Old State:6-ENGN_StateRunning_StartNewStep New State:ENGN_StateRunning_WaitGoingToSample
2021/01/22 16:15:18.0 Sample96_Q(0, 0, 10)
2021/01/22 16:15:18.0 GMC_SetPosZ
2021/01/22 16:15:18.1 
2021/01/22 16:15:18.1 START SCAN: 
2021/01/22 16:15:18.1    CycleTime (ms)    = 200
2021/01/22 16:15:18.1    CycleCount        = 25
2021/01/22 16:15:18.1    ScansToAverage    = 20
2021/01/22 16:15:18.1    FinishTime (s)    = 5
2021/01/22 16:15:18.1    Temperature       = 25.0
2021/01/22 16:15:18.1    Lamp Power        = 31
2021/01/22 16:15:18.1    Spec PCB temps    = 42.3, 41.9, 42.7, 42.4, 42.2, 41.9, 40.9, 41.1
2021/01/22 16:15:18.1 
2021/01/22 16:15:18.1 Temp: 2021/01/22 16:15:18, 25.0000, 25.5000
2021/01/22 16:15:18.2 Old State:12-ENGN_StateRunning_WaitGoingToSample New State:ENGN_StateRunning_GoingToSample
2021/01/22 16:15:24.1 GMC_ServoHere
2021/01/22 16:15:24.2 GMC_SetPosXY
2021/01/22 16:15:24.6 GMC_SetPosZ
2021/01/22 16:15:30.4 Shaker current speed = 494 rpm, set speed = 500
2021/01/22 16:15:30.5 Old State:13-cycle=1, Elapsed Time:    0.0,  Ideal Time:    0.0,  Delay time =   0.00  New State:ENGN_StateRunning_GoingToSampleComplete
2021/01/22 16:15:30.6 Old State:14-ENGN_StateRunning_GoingToSampleComplete New State:ENGN_StateRunning_WaitSampling
2021/01/22 16:15:30.6 Step started at: 16:15:18
2021/01/22 16:15:30.6 XYZ movement time  = 12424 ms + Scan start delay = 12326 ms
2021/01/22 16:15:30.7 Old State:15-ENGN_StateRunning_WaitSampling New State:ENGN_StateRunning_Sampling
2021/01/22 16:15:35.3 
2021/01/22 16:15:35.3 STOP SCAN: 
2021/01/22 16:15:35.3    CycleTime (ms)    = 200
2021/01/22 16:15:35.3    CurrentCycle      = 25
2021/01/22 16:15:35.3    RunTime (ms)      = 4959
2021/01/22 16:15:35.3    AverageCycle (ms) = 198.4
2021/01/22 16:15:35.3    Temperature       = 25.0
2021/01/22 16:15:35.3    DATA OVERRUN %    = 0.00
2021/01/22 16:15:35.3    Spec PCB temps    = 42.3, 41.9, 42.7, 42.4, 42.2, 42.0, 40.8, 41.2
2021/01/22 16:15:35.3 
2021/01/22 16:15:35.4 Old State:16-cycle=25, Elapsed Time:    5.0,  Ideal Time:    4.8,  Delay time =   0.04  New State:ENGN_StateRunning_SamplingComplete
2021/01/22 16:15:35.5 Old State:18-ENGN_StateRunning_SamplingComplete New State:ENGN_StateRunning_WaitNewStep
2021/01/22 16:15:35.7 Old State:5-ENGN_StateRunning_WaitNewStep New State:ENGN_StateRunning_StartNewStep
2021/01/22 16:15:35.7 Spectrometer EOS_B0712: AVRG=20.
2021/01/22 16:15:35.7 Spectrometer EOS_B0734: AVRG=20.
2021/01/22 16:15:35.7 Spectrometer EOS_B0814: AVRG=20.
2021/01/22 16:15:35.7 Spectrometer EOS_B0735: AVRG=20.
2021/01/22 16:15:35.7 Spectrometer EOS_B0732: AVRG=20.
2021/01/22 16:15:35.7 Spectrometer EOS_B0808: AVRG=20.
2021/01/22 16:15:35.7 Spectrometer EOS_B0733: AVRG=20.
2021/01/22 16:15:35.7 Spectrometer EOS_B0817: AVRG=20.
2021/01/22 16:15:35.7 StartShaker(500)
2021/01/22 16:15:35.8 Old State:6-ENGN_StateRunning_StartNewStep New State:ENGN_StateRunning_WaitGoingToSample
2021/01/22 16:15:35.8 Sample96_Q(0, 0, 11)
2021/01/22 16:15:35.9 
2021/01/22 16:15:35.9 START SCAN: 
2021/01/22 16:15:35.9    CycleTime (ms)    = 200
2021/01/22 16:15:35.9    CycleCount        = 25
2021/01/22 16:15:35.9    ScansToAverage    = 20
2021/01/22 16:15:35.9    FinishTime (s)    = 5
2021/01/22 16:15:35.9    Temperature       = 25.0
2021/01/22 16:15:35.9    Lamp Power        = 31
2021/01/22 16:15:35.9    Spec PCB temps    = 42.3, 41.9, 42.7, 42.4, 42.2, 41.9, 40.8, 41.2
2021/01/22 16:15:35.9 
2021/01/22 16:15:35.9 GMC_SetPosZ
2021/01/22 16:15:36.0 Old State:12-ENGN_StateRunning_WaitGoingToSample New State:ENGN_StateRunning_GoingToSample
2021/01/22 16:15:36.0 Temp: 2021/01/22 16:15:36, 25.0000, 25.4000
2021/01/22 16:15:41.7 GMC_ServoHere
2021/01/22 16:15:41.8 GMC_SetPosXY
2021/01/22 16:15:42.2 GMC_SetPosZ
2021/01/22 16:15:48.4 Shaker current speed = 494 rpm, set speed = 500
2021/01/22 16:15:48.5 Old State:13-ENGN_StateRunning_GoingToSample New State:ENGN_StateRunning_GoingToSampleComplete
2021/01/22 16:15:48.6 Old State:14-cycle=1, Elapsed Time:    0.0,  Ideal Time:    0.0,  Delay time =   0.00  New State:ENGN_StateRunning_WaitSampling
2021/01/22 16:15:48.6 Step started at: 16:15:35
2021/01/22 16:15:48.6 XYZ movement time  = 12587 ms + Scan start delay = 12545 ms
2021/01/22 16:15:48.7 Old State:15-ENGN_StateRunning_WaitSampling New State:ENGN_StateRunning_Sampling
2021/01/22 16:15:53.4 
2021/01/22 16:15:53.4 STOP SCAN: 
2021/01/22 16:15:53.4    CycleTime (ms)    = 200
2021/01/22 16:15:53.4    CurrentCycle      = 25
2021/01/22 16:15:53.4    RunTime (ms)      = 4958
2021/01/22 16:15:53.4    AverageCycle (ms) = 198.3
2021/01/22 16:15:53.4    Temperature       = 25.0
2021/01/22 16:15:53.4    DATA OVERRUN %    = 0.00
2021/01/22 16:15:53.4    Spec PCB temps    = 42.3, 41.9, 42.7, 42.4, 42.2, 41.9, 40.9, 41.1
2021/01/22 16:15:53.4 
2021/01/22 16:15:53.4 Old State:16-cycle=25, Elapsed Time:    5.0,  Ideal Time:    4.8,  Delay time =   0.04  New State:ENGN_StateRunning_SamplingComplete
2021/01/22 16:15:53.5 Old State:18-ENGN_StateRunning_SamplingComplete New State:ENGN_StateRunning_WaitNewStep
2021/01/22 16:15:53.6 Old State:5-ENGN_StateRunning_WaitNewStep New State:ENGN_StateRunning_StartNewStep
2021/01/22 16:15:53.7 Spectrometer EOS_B0712: AVRG=20.
2021/01/22 16:15:53.7 Spectrometer EOS_B0734: AVRG=20.
2021/01/22 16:15:53.7 Spectrometer EOS_B0814: AVRG=20.
2021/01/22 16:15:53.7 Spectrometer EOS_B0735: AVRG=20.
2021/01/22 16:15:53.7 Spectrometer EOS_B0732: AVRG=20.
2021/01/22 16:15:53.7 Spectrometer EOS_B0808: AVRG=20.
2021/01/22 16:15:53.7 Spectrometer EOS_B0733: AVRG=20.
2021/01/22 16:15:53.7 Spectrometer EOS_B0817: AVRG=20.
2021/01/22 16:15:53.7 StartShaker(500)
2021/01/22 16:15:53.8 Old State:6-ENGN_StateRunning_StartNewStep New State:ENGN_StateRunning_WaitGoingToSample
2021/01/22 16:15:53.8 Sample96_Q(0, 0, 10)
2021/01/22 16:15:53.8 
2021/01/22 16:15:53.8 START SCAN: 
2021/01/22 16:15:53.8    CycleTime (ms)    = 200
2021/01/22 16:15:53.8    CycleCount        = 25
2021/01/22 16:15:53.8    ScansToAverage    = 20
2021/01/22 16:15:53.8    FinishTime (s)    = 5
2021/01/22 16:15:53.8    Temperature       = 25.0
2021/01/22 16:15:53.8    Lamp Power        = 31
2021/01/22 16:15:53.8    Spec PCB temps    = 42.3, 41.9, 42.7, 42.4, 42.2, 41.9, 40.9, 41.1
2021/01/22 16:15:53.8 
2021/01/22 16:15:53.8 GMC_SetPosZ
2021/01/22 16:15:53.9 Temp: 2021/01/22 16:15:53, 25.0000, 25.5000
2021/01/22 16:15:53.9 Old State:12-ENGN_StateRunning_WaitGoingToSample New State:ENGN_StateRunning_GoingToSample
2021/01/22 16:15:59.9 GMC_ServoHere
2021/01/22 16:16:00.0 GMC_SetPosXY
2021/01/22 16:16:00.4 GMC_SetPosZ
2021/01/22 16:16:06.3 Shaker current speed = 494 rpm, set speed = 500
2021/01/22 16:16:06.4 Old State:13-cycle=1, Elapsed Time:    0.0,  Ideal Time:    0.0,  Delay time =   0.00  New State:ENGN_StateRunning_GoingToSampleComplete
2021/01/22 16:16:06.5 Old State:14-ENGN_StateRunning_GoingToSampleComplete New State:ENGN_StateRunning_WaitSampling
2021/01/22 16:16:06.5 Step started at: 16:15:53
2021/01/22 16:16:06.5 XYZ movement time  = 12574 ms + Scan start delay = 12492 ms
2021/01/22 16:16:06.6 Old State:15-ENGN_StateRunning_WaitSampling New State:ENGN_StateRunning_Sampling
2021/01/22 16:16:11.2 
2021/01/22 16:16:11.2 STOP SCAN: 
2021/01/22 16:16:11.2    CycleTime (ms)    = 200
2021/01/22 16:16:11.2    CurrentCycle      = 25
2021/01/22 16:16:11.2    RunTime (ms)      = 4959
2021/01/22 16:16:11.2    AverageCycle (ms) = 198.4
2021/01/22 16:16:11.2    Temperature       = 25.0
2021/01/22 16:16:11.2    DATA OVERRUN %    = 0.00
2021/01/22 16:16:11.2    Spec PCB temps    = 42.3, 41.9, 42.7, 42.4, 42.2, 41.9, 40.8, 41.2
2021/01/22 16:16:11.2 
2021/01/22 16:16:11.3 Old State:16-cycle=25, Elapsed Time:    5.0,  Ideal Time:    4.8,  Delay time =   0.05  New State:ENGN_StateRunning_SamplingComplete
2021/01/22 16:16:11.4 Old State:18-ENGN_StateRunning_SamplingComplete New State:ENGN_StateRunning_WaitNewStep
2021/01/22 16:16:11.7 Old State:5-ENGN_StateRunning_WaitNewStep New State:ENGN_StateRunning_StartNewStep
2021/01/22 16:16:11.8 Spectrometer EOS_B0712: AVRG=20.
2021/01/22 16:16:11.8 Spectrometer EOS_B0734: AVRG=20.
2021/01/22 16:16:11.8 Spectrometer EOS_B0814: AVRG=20.
2021/01/22 16:16:11.8 Spectrometer EOS_B0735: AVRG=20.
2021/01/22 16:16:11.8 Spectrometer EOS_B0732: AVRG=20.
2021/01/22 16:16:11.8 Spectrometer EOS_B0808: AVRG=20.
2021/01/22 16:16:11.8 Spectrometer EOS_B0733: AVRG=20.
2021/01/22 16:16:11.8 Spectrometer EOS_B0817: AVRG=20.
2021/01/22 16:16:11.8 StartShaker(500)
2021/01/22 16:16:11.9 Old State:6-ENGN_StateRunning_StartNewStep New State:ENGN_StateRunning_WaitGoingToSample
2021/01/22 16:16:11.9 Sample96_Q(0, 0, 3)
2021/01/22 16:16:11.9 
2021/01/22 16:16:11.9 START SCAN: 
2021/01/22 16:16:11.9    CycleTime (ms)    = 200
2021/01/22 16:16:11.9    CycleCount        = 1500
2021/01/22 16:16:11.9    ScansToAverage    = 20
2021/01/22 16:16:11.9    FinishTime (s)    = 300
2021/01/22 16:16:11.9    Temperature       = 25.0
2021/01/22 16:16:11.9    Lamp Power        = 31
2021/01/22 16:16:11.9    Spec PCB temps    = 42.3, 41.9, 42.8, 42.4, 42.1, 41.9, 40.8, 41.2
2021/01/22 16:16:11.9 
2021/01/22 16:16:12.0 Old State:12-ENGN_StateRunning_WaitGoingToSample New State:ENGN_StateRunning_GoingToSample
2021/01/22 16:16:12.0 GMC_SetPosZ
2021/01/22 16:16:12.1 Temp: 2021/01/22 16:16:12, 25.0000, 25.5000
2021/01/22 16:16:17.8 GMC_ServoHere
2021/01/22 16:16:17.9 GMC_SetPosXY
2021/01/22 16:16:19.2 GMC_SetPosZ
2021/01/22 16:16:23.2 Shaker current speed = 494 rpm, set speed = 500
2021/01/22 16:16:23.3 Old State:13-ENGN_StateRunning_GoingToSample New State:ENGN_StateRunning_GoingToSampleComplete
2021/01/22 16:16:23.4 Old State:14-cycle=1, Elapsed Time:    0.0,  Ideal Time:    0.0,  Delay time =   0.00  New State:ENGN_StateRunning_WaitSampling
2021/01/22 16:16:23.4 Step started at: 16:16:11
2021/01/22 16:16:23.4 XYZ movement time  = 11280 ms + Scan start delay = 11229 ms
2021/01/22 16:16:23.5 Old State:15-ENGN_StateRunning_WaitSampling New State:ENGN_StateRunning_Sampling
2021/01/22 16:17:12.2 Temp: 2021/01/22 16:17:12, 25.0000, 25.5000
2021/01/22 16:18:12.4 Temp: 2021/01/22 16:18:12, 25.0000, 25.5000
2021/01/22 16:19:12.5 Temp: 2021/01/22 16:19:12, 25.0000, 25.5000
2021/01/22 16:20:12.7 Temp: 2021/01/22 16:20:12, 25.0000, 25.4000
2021/01/22 16:21:12.9 Temp: 2021/01/22 16:21:12, 25.0000, 25.4000
2021/01/22 16:21:23.1 
2021/01/22 16:21:23.1 STOP SCAN: 
2021/01/22 16:21:23.1    CycleTime (ms)    = 200
2021/01/22 16:21:23.1    CurrentCycle      = 1500
2021/01/22 16:21:23.1    RunTime (ms)      = 299957
2021/01/22 16:21:23.1    AverageCycle (ms) = 200.0
2021/01/22 16:21:23.1    Temperature       = 25.0
2021/01/22 16:21:23.1    DATA OVERRUN %    = 0.00
2021/01/22 16:21:23.1    Spec PCB temps    = 42.3, 41.9, 42.8, 42.4, 42.2, 42.0, 40.8, 41.1
2021/01/22 16:21:23.1 
2021/01/22 16:21:23.1 Old State:16-cycle=1500, Elapsed Time:  300.0,  Ideal Time:  299.8,  Delay time =   0.04  New State:ENGN_StateRunning_SamplingComplete
2021/01/22 16:21:23.3 Old State:18-ENGN_StateRunning_SamplingComplete New State:ENGN_StateRunning_WaitNewStep
2021/01/22 16:21:23.6 Old State:5-ENGN_StateRunning_WaitNewStep New State:ENGN_StateRunning_StartNewStep
2021/01/22 16:21:23.6 Spectrometer EOS_B0712: AVRG=20.
2021/01/22 16:21:23.6 Spectrometer EOS_B0734: AVRG=20.
2021/01/22 16:21:23.6 Spectrometer EOS_B0814: AVRG=20.
2021/01/22 16:21:23.6 Spectrometer EOS_B0735: AVRG=20.
2021/01/22 16:21:23.6 Spectrometer EOS_B0732: AVRG=20.
2021/01/22 16:21:23.6 Spectrometer EOS_B0808: AVRG=20.
2021/01/22 16:21:23.6 Spectrometer EOS_B0733: AVRG=20.
2021/01/22 16:21:23.6 Spectrometer EOS_B0817: AVRG=20.
2021/01/22 16:21:23.6 StartShaker(500)
2021/01/22 16:21:23.8 Old State:6-ENGN_StateRunning_StartNewStep New State:ENGN_StateRunning_WaitGoingToSample
2021/01/22 16:21:23.8 Sample96_Q(0, 0, 5)
2021/01/22 16:21:23.8 
2021/01/22 16:21:23.8 START SCAN: 
2021/01/22 16:21:23.8    CycleTime (ms)    = 200
2021/01/22 16:21:23.8    CycleCount        = 250
2021/01/22 16:21:23.8    ScansToAverage    = 20
2021/01/22 16:21:23.8    FinishTime (s)    = 50
2021/01/22 16:21:23.8    Temperature       = 25.0
2021/01/22 16:21:23.8    Lamp Power        = 31
2021/01/22 16:21:23.8    Spec PCB temps    = 42.3, 41.9, 42.8, 42.4, 42.1, 42.0, 40.9, 41.2
2021/01/22 16:21:23.8 
2021/01/22 16:21:23.8 GMC_SetPosZ
2021/01/22 16:21:23.9 Temp: 2021/01/22 16:21:23, 25.0000, 25.5000
2021/01/22 16:21:23.9 Old State:12-ENGN_StateRunning_WaitGoingToSample New State:ENGN_StateRunning_GoingToSample
2021/01/22 16:21:27.6 GMC_ServoHere
2021/01/22 16:21:27.8 GMC_SetPosXY
2021/01/22 16:21:28.2 GMC_SetPosZ
2021/01/22 16:21:32.8 Old State:13-ENGN_StateRunning_GoingToSample New State:ENGN_StateRunning_GoingToSampleComplete
2021/01/22 16:21:32.9 Old State:14-cycle=1, Elapsed Time:    0.0,  Ideal Time:    0.0,  Delay time =   0.00  New State:ENGN_StateRunning_WaitSampling
2021/01/22 16:21:32.9 Step started at: 16:21:23
2021/01/22 16:21:32.9 XYZ movement time  = 8955 ms + Scan start delay = 8905 ms
2021/01/22 16:21:33.0 Old State:15-ENGN_StateRunning_WaitSampling New State:ENGN_StateRunning_Sampling
2021/01/22 16:21:33.7 Shaker current speed = 494 rpm, set speed = 500
2021/01/22 16:22:22.6 
2021/01/22 16:22:22.6 STOP SCAN: 
2021/01/22 16:22:22.6    CycleTime (ms)    = 200
2021/01/22 16:22:22.6    CurrentCycle      = 250
2021/01/22 16:22:22.6    RunTime (ms)      = 49958
2021/01/22 16:22:22.6    AverageCycle (ms) = 199.8
2021/01/22 16:22:22.6    Temperature       = 25.0
2021/01/22 16:22:22.6    DATA OVERRUN %    = 0.00
2021/01/22 16:22:22.6    Spec PCB temps    = 42.3, 41.9, 42.7, 42.4, 42.2, 41.9, 40.9, 41.2
2021/01/22 16:22:22.6 
2021/01/22 16:22:22.7 Old State:16-cycle=250, Elapsed Time:   50.0,  Ideal Time:   49.8,  Delay time =   0.04  New State:ENGN_StateRunning_SamplingComplete
2021/01/22 16:22:22.8 Old State:18-ENGN_StateRunning_SamplingComplete New State:ENGN_StateRunning_WaitNewStep
2021/01/22 16:22:23.1 Old State:5-ENGN_StateRunning_WaitNewStep New State:ENGN_StateRunning_StartNewStep
2021/01/22 16:22:23.1 Spectrometer EOS_B0712: AVRG=20.
2021/01/22 16:22:23.1 Spectrometer EOS_B0734: AVRG=20.
2021/01/22 16:22:23.1 Spectrometer EOS_B0814: AVRG=20.
2021/01/22 16:22:23.1 Spectrometer EOS_B0735: AVRG=20.
2021/01/22 16:22:23.1 Spectrometer EOS_B0732: AVRG=20.
2021/01/22 16:22:23.1 Spectrometer EOS_B0808: AVRG=20.
2021/01/22 16:22:23.1 Spectrometer EOS_B0733: AVRG=20.
2021/01/22 16:22:23.1 Spectrometer EOS_B0817: AVRG=20.
2021/01/22 16:22:23.1 StartShaker(500)
2021/01/22 16:22:23.2 Old State:6-ENGN_StateRunning_StartNewStep New State:ENGN_StateRunning_WaitGoingToSample
2021/01/22 16:22:23.2 Sample96_Q(0, 0, 3)
2021/01/22 16:22:23.2 
2021/01/22 16:22:23.2 START SCAN: 
2021/01/22 16:22:23.2    CycleTime (ms)    = 200
2021/01/22 16:22:23.2    CycleCount        = 1500
2021/01/22 16:22:23.2    ScansToAverage    = 20
2021/01/22 16:22:23.2    FinishTime (s)    = 300
2021/01/22 16:22:23.2    Temperature       = 25.0
2021/01/22 16:22:23.2    Lamp Power        = 31
2021/01/22 16:22:23.2    Spec PCB temps    = 42.3, 41.9, 42.7, 42.4, 42.2, 41.9, 40.9, 41.2
2021/01/22 16:22:23.2 
2021/01/22 16:22:23.3 GMC_SetPosZ
2021/01/22 16:22:23.3 Old State:12-ENGN_StateRunning_WaitGoingToSample New State:ENGN_StateRunning_GoingToSample
2021/01/22 16:22:23.3 Temp: 2021/01/22 16:22:23, 25.0000, 25.5000
2021/01/22 16:22:27.7 GMC_ServoHere
2021/01/22 16:22:27.8 GMC_SetPosXY
2021/01/22 16:22:28.3 GMC_SetPosZ
2021/01/22 16:22:32.4 Old State:13-ENGN_StateRunning_GoingToSample New State:ENGN_StateRunning_GoingToSampleComplete
2021/01/22 16:22:32.5 Old State:14-cycle=1, Elapsed Time:    0.0,  Ideal Time:    0.0,  Delay time =   0.00  New State:ENGN_StateRunning_WaitSampling
2021/01/22 16:22:32.5 Step started at: 16:22:23
2021/01/22 16:22:32.5 XYZ movement time  = 9085 ms + Scan start delay = 9052 ms
2021/01/22 16:22:32.6 Old State:15-ENGN_StateRunning_WaitSampling New State:ENGN_StateRunning_Sampling
2021/01/22 16:22:33.1 Shaker current speed = 494 rpm, set speed = 500
2021/01/22 16:23:23.5 Temp: 2021/01/22 16:23:23, 25.0000, 25.5000
2021/01/22 16:24:23.7 Temp: 2021/01/22 16:24:23, 25.0000, 25.4000
2021/01/22 16:25:23.8 Temp: 2021/01/22 16:25:23, 25.0000, 25.4000
2021/01/22 16:26:24.0 Temp: 2021/01/22 16:26:24, 25.0000, 25.4000
2021/01/22 16:27:24.1 Temp: 2021/01/22 16:27:24, 25.0000, 25.4000
2021/01/22 16:27:32.2 
2021/01/22 16:27:32.2 STOP SCAN: 
2021/01/22 16:27:32.2    CycleTime (ms)    = 200
2021/01/22 16:27:32.2    CurrentCycle      = 1500
2021/01/22 16:27:32.2    RunTime (ms)      = 299959
2021/01/22 16:27:32.2    AverageCycle (ms) = 200.0
2021/01/22 16:27:32.2    Temperature       = 25.0
2021/01/22 16:27:32.2    DATA OVERRUN %    = 0.00
2021/01/22 16:27:32.2    Spec PCB temps    = 42.3, 41.9, 42.7, 42.4, 42.2, 41.9, 40.9, 41.2
2021/01/22 16:27:32.2 
2021/01/22 16:27:32.3 Old State:16-cycle=1500, Elapsed Time:  300.0,  Ideal Time:  299.8,  Delay time =   0.04  New State:ENGN_StateRunning_SamplingComplete
2021/01/22 16:27:32.4 Old State:18-ENGN_StateRunning_SamplingComplete New State:ENGN_StateRunning_WaitNewStep
2021/01/22 16:27:32.7 Old State:5-ENGN_StateRunning_WaitNewStep New State:ENGN_StateRunning_StartNewStep
2021/01/22 16:27:32.8 Spectrometer EOS_B0712: AVRG=20.
2021/01/22 16:27:32.8 Spectrometer EOS_B0734: AVRG=20.
2021/01/22 16:27:32.8 Spectrometer EOS_B0814: AVRG=20.
2021/01/22 16:27:32.8 Spectrometer EOS_B0735: AVRG=20.
2021/01/22 16:27:32.8 Spectrometer EOS_B0732: AVRG=20.
2021/01/22 16:27:32.8 Spectrometer EOS_B0808: AVRG=20.
2021/01/22 16:27:32.8 Spectrometer EOS_B0733: AVRG=20.
2021/01/22 16:27:32.8 Spectrometer EOS_B0817: AVRG=20.
2021/01/22 16:27:32.8 StartShaker(500)
2021/01/22 16:27:32.9 Old State:6-ENGN_StateRunning_StartNewStep New State:ENGN_StateRunning_WaitGoingToSample
2021/01/22 16:27:32.9 Sample96_Q(0, 0, 11)
2021/01/22 16:27:32.9 
2021/01/22 16:27:32.9 START SCAN: 
2021/01/22 16:27:32.9    CycleTime (ms)    = 200
2021/01/22 16:27:32.9    CycleCount        = 25
2021/01/22 16:27:32.9    ScansToAverage    = 20
2021/01/22 16:27:32.9    FinishTime (s)    = 5
2021/01/22 16:27:32.9    Temperature       = 25.0
2021/01/22 16:27:32.9    Lamp Power        = 31
2021/01/22 16:27:32.9    Spec PCB temps    = 42.3, 41.9, 42.7, 42.4, 42.2, 41.9, 40.9, 41.2
2021/01/22 16:27:32.9 
2021/01/22 16:27:33.0 GMC_SetPosZ
2021/01/22 16:27:33.0 Old State:12-ENGN_StateRunning_WaitGoingToSample New State:ENGN_StateRunning_GoingToSample
2021/01/22 16:27:33.1 Temp: 2021/01/22 16:27:33, 25.0000, 25.4000
2021/01/22 16:27:36.7 GMC_ServoHere
2021/01/22 16:27:36.9 GMC_SetPosXY
2021/01/22 16:27:38.3 GMC_SetPosZ
2021/01/22 16:27:44.6 Old State:13-ENGN_StateRunning_GoingToSample New State:ENGN_StateRunning_GoingToSampleComplete
2021/01/22 16:27:44.6 Shaker current speed = 494 rpm, set speed = 500
2021/01/22 16:27:44.7 Old State:14-cycle=1, Elapsed Time:    0.0,  Ideal Time:    0.0,  Delay time =   0.00  New State:ENGN_StateRunning_WaitSampling
2021/01/22 16:27:44.7 Step started at: 16:27:32
2021/01/22 16:27:44.7 XYZ movement time  = 11608 ms + Scan start delay = 11622 ms
2021/01/22 16:27:44.8 Old State:15-ENGN_StateRunning_WaitSampling New State:ENGN_StateRunning_Sampling
2021/01/22 16:27:49.5 
2021/01/22 16:27:49.5 STOP SCAN: 
2021/01/22 16:27:49.5    CycleTime (ms)    = 200
2021/01/22 16:27:49.5    CurrentCycle      = 25
2021/01/22 16:27:49.5    RunTime (ms)      = 4957
2021/01/22 16:27:49.5    AverageCycle (ms) = 198.3
2021/01/22 16:27:49.5    Temperature       = 25.0
2021/01/22 16:27:49.5    DATA OVERRUN %    = 0.00
2021/01/22 16:27:49.5    Spec PCB temps    = 42.3, 41.9, 42.7, 42.4, 42.3, 42.0, 40.9, 41.2
2021/01/22 16:27:49.5 
2021/01/22 16:27:49.5 Old State:16-cycle=25, Elapsed Time:    5.0,  Ideal Time:    4.8,  Delay time =   0.04  New State:ENGN_StateRunning_SamplingComplete
2021/01/22 16:27:49.7 Old State:18-ENGN_StateRunning_SamplingComplete New State:ENGN_StateRunning_WaitNewStep
2021/01/22 16:27:50.0 Old State:5-ENGN_StateRunning_WaitNewStep New State:ENGN_StateRunning_StartNewStep
2021/01/22 16:27:50.0 Spectrometer EOS_B0712: AVRG=20.
2021/01/22 16:27:50.0 Spectrometer EOS_B0734: AVRG=20.
2021/01/22 16:27:50.0 Spectrometer EOS_B0814: AVRG=20.
2021/01/22 16:27:50.0 Spectrometer EOS_B0735: AVRG=20.
2021/01/22 16:27:50.0 Spectrometer EOS_B0732: AVRG=20.
2021/01/22 16:27:50.0 Spectrometer EOS_B0808: AVRG=20.
2021/01/22 16:27:50.0 Spectrometer EOS_B0733: AVRG=20.
2021/01/22 16:27:50.0 Spectrometer EOS_B0817: AVRG=20.
2021/01/22 16:27:50.0 StartShaker(500)
2021/01/22 16:27:50.1 Old State:6-ENGN_StateRunning_StartNewStep New State:ENGN_StateRunning_WaitGoingToSample
2021/01/22 16:27:50.1 Sample96_Q(0, 0, 10)
2021/01/22 16:27:50.2 
2021/01/22 16:27:50.2 START SCAN: 
2021/01/22 16:27:50.2    CycleTime (ms)    = 200
2021/01/22 16:27:50.2    CycleCount        = 25
2021/01/22 16:27:50.2    ScansToAverage    = 20
2021/01/22 16:27:50.2    FinishTime (s)    = 5
2021/01/22 16:27:50.2    Temperature       = 25.0
2021/01/22 16:27:50.2    Lamp Power        = 31
2021/01/22 16:27:50.2    Spec PCB temps    = 42.3, 41.9, 42.7, 42.4, 42.3, 42.0, 40.9, 41.2
2021/01/22 16:27:50.2 
2021/01/22 16:27:50.3 Temp: 2021/01/22 16:27:50, 25.0000, 25.4000
2021/01/22 16:27:50.3 Old State:12-ENGN_StateRunning_WaitGoingToSample New State:ENGN_StateRunning_GoingToSample
2021/01/22 16:27:50.3 GMC_SetPosZ
2021/01/22 16:27:56.3 GMC_ServoHere
2021/01/22 16:27:56.5 GMC_SetPosXY
2021/01/22 16:27:56.8 GMC_SetPosZ
2021/01/22 16:28:02.7 Shaker current speed = 494 rpm, set speed = 500
2021/01/22 16:28:02.8 Old State:13-ENGN_StateRunning_GoingToSample New State:ENGN_StateRunning_GoingToSampleComplete
2021/01/22 16:28:02.9 Old State:14-cycle=1, Elapsed Time:    0.0,  Ideal Time:    0.0,  Delay time =   0.00  New State:ENGN_StateRunning_WaitSampling
2021/01/22 16:28:02.9 Step started at: 16:27:50
2021/01/22 16:28:02.9 XYZ movement time  = 12595 ms + Scan start delay = 12581 ms
2021/01/22 16:28:03.0 Old State:15-ENGN_StateRunning_WaitSampling New State:ENGN_StateRunning_Sampling
2021/01/22 16:28:07.7 
2021/01/22 16:28:07.7 STOP SCAN: 
2021/01/22 16:28:07.7    CycleTime (ms)    = 200
2021/01/22 16:28:07.7    CurrentCycle      = 25
2021/01/22 16:28:07.7    RunTime (ms)      = 4959
2021/01/22 16:28:07.7    AverageCycle (ms) = 198.4
2021/01/22 16:28:07.7    Temperature       = 25.0
2021/01/22 16:28:07.7    DATA OVERRUN %    = 0.00
2021/01/22 16:28:07.7    Spec PCB temps    = 42.3, 41.9, 42.7, 42.4, 42.2, 41.9, 40.9, 41.2
2021/01/22 16:28:07.7 
2021/01/22 16:28:07.8 Old State:16-cycle=25, Elapsed Time:    5.0,  Ideal Time:    4.8,  Delay time =   0.04  New State:ENGN_StateRunning_SamplingComplete
2021/01/22 16:28:07.9 Old State:18-ENGN_StateRunning_SamplingComplete New State:ENGN_StateRunning_WaitNewStep
2021/01/22 16:28:08.2 Old State:5-ENGN_StateRunning_WaitNewStep New State:ENGN_StateRunning_StartNewStep
2021/01/22 16:28:08.3 Spectrometer EOS_B0712: AVRG=20.
2021/01/22 16:28:08.3 Spectrometer EOS_B0734: AVRG=20.
2021/01/22 16:28:08.3 Spectrometer EOS_B0814: AVRG=20.
2021/01/22 16:28:08.3 Spectrometer EOS_B0735: AVRG=20.
2021/01/22 16:28:08.3 Spectrometer EOS_B0732: AVRG=20.
2021/01/22 16:28:08.3 Spectrometer EOS_B0808: AVRG=20.
2021/01/22 16:28:08.3 Spectrometer EOS_B0733: AVRG=20.
2021/01/22 16:28:08.3 Spectrometer EOS_B0817: AVRG=20.
2021/01/22 16:28:08.3 StartShaker(500)
2021/01/22 16:28:08.4 Old State:6-ENGN_StateRunning_StartNewStep New State:ENGN_StateRunning_WaitGoingToSample
2021/01/22 16:28:08.4 Sample96_Q(0, 0, 11)
2021/01/22 16:28:08.4 
2021/01/22 16:28:08.4 START SCAN: 
2021/01/22 16:28:08.4    CycleTime (ms)    = 200
2021/01/22 16:28:08.4    CycleCount        = 25
2021/01/22 16:28:08.4    ScansToAverage    = 20
2021/01/22 16:28:08.4    FinishTime (s)    = 5
2021/01/22 16:28:08.4    Temperature       = 25.0
2021/01/22 16:28:08.4    Lamp Power        = 31
2021/01/22 16:28:08.4    Spec PCB temps    = 42.3, 41.9, 42.7, 42.4, 42.1, 41.9, 40.9, 41.2
2021/01/22 16:28:08.4 
2021/01/22 16:28:08.5 GMC_SetPosZ
2021/01/22 16:28:08.5 Old State:12-ENGN_StateRunning_WaitGoingToSample New State:ENGN_StateRunning_GoingToSample
2021/01/22 16:28:08.5 Temp: 2021/01/22 16:28:08, 25.0000, 25.5000
2021/01/22 16:28:14.2 GMC_ServoHere
2021/01/22 16:28:14.4 GMC_SetPosXY
2021/01/22 16:28:14.7 GMC_SetPosZ
2021/01/22 16:28:20.9 Shaker current speed = 494 rpm, set speed = 500
2021/01/22 16:28:21.0 Old State:13-cycle=1, Elapsed Time:    0.0,  Ideal Time:    0.0,  Delay time =   0.00  New State:ENGN_StateRunning_GoingToSampleComplete
2021/01/22 16:28:21.1 Old State:14-ENGN_StateRunning_GoingToSampleComplete New State:ENGN_StateRunning_WaitSampling
2021/01/22 16:28:21.1 Step started at: 16:28:08
2021/01/22 16:28:21.1 XYZ movement time  = 12478 ms + Scan start delay = 12383 ms
2021/01/22 16:28:21.2 Old State:15-ENGN_StateRunning_WaitSampling New State:ENGN_StateRunning_Sampling
2021/01/22 16:28:25.7 
2021/01/22 16:28:25.7 STOP SCAN: 
2021/01/22 16:28:25.7    CycleTime (ms)    = 200
2021/01/22 16:28:25.7    CurrentCycle      = 25
2021/01/22 16:28:25.7    RunTime (ms)      = 4958
2021/01/22 16:28:25.7    AverageCycle (ms) = 198.3
2021/01/22 16:28:25.7    Temperature       = 25.0
2021/01/22 16:28:25.7    DATA OVERRUN %    = 0.00
2021/01/22 16:28:25.7    Spec PCB temps    = 42.3, 41.9, 42.8, 42.4, 42.2, 41.9, 40.9, 41.2
2021/01/22 16:28:25.7 
2021/01/22 16:28:25.8 Old State:16-cycle=25, Elapsed Time:    5.0,  Ideal Time:    4.8,  Delay time =   0.04  New State:ENGN_StateRunning_SamplingComplete
2021/01/22 16:28:25.9 Old State:18-ENGN_StateRunning_SamplingComplete New State:ENGN_StateRunning_WaitNewStep
2021/01/22 16:28:26.0 Old State:5-ENGN_StateRunning_WaitNewStep New State:ENGN_StateRunning_StartNewStep
2021/01/22 16:28:26.1 Spectrometer EOS_B0712: AVRG=20.
2021/01/22 16:28:26.1 Spectrometer EOS_B0734: AVRG=20.
2021/01/22 16:28:26.1 Spectrometer EOS_B0814: AVRG=20.
2021/01/22 16:28:26.1 Spectrometer EOS_B0735: AVRG=20.
2021/01/22 16:28:26.1 Spectrometer EOS_B0732: AVRG=20.
2021/01/22 16:28:26.1 Spectrometer EOS_B0808: AVRG=20.
2021/01/22 16:28:26.1 Spectrometer EOS_B0733: AVRG=20.
2021/01/22 16:28:26.1 Spectrometer EOS_B0817: AVRG=20.
2021/01/22 16:28:26.1 StartShaker(500)
2021/01/22 16:28:26.2 Old State:6-ENGN_StateRunning_StartNewStep New State:ENGN_StateRunning_WaitGoingToSample
2021/01/22 16:28:26.2 Sample96_Q(0, 0, 10)
2021/01/22 16:28:26.2 
2021/01/22 16:28:26.2 START SCAN: 
2021/01/22 16:28:26.2    CycleTime (ms)    = 200
2021/01/22 16:28:26.2    CycleCount        = 25
2021/01/22 16:28:26.2    ScansToAverage    = 20
2021/01/22 16:28:26.2    FinishTime (s)    = 5
2021/01/22 16:28:26.2    Temperature       = 25.0
2021/01/22 16:28:26.2    Lamp Power        = 31
2021/01/22 16:28:26.2    Spec PCB temps    = 42.3, 41.9, 42.8, 42.4, 42.2, 41.9, 40.9, 41.2
2021/01/22 16:28:26.2 
2021/01/22 16:28:26.3 Old State:12-ENGN_StateRunning_WaitGoingToSample New State:ENGN_StateRunning_GoingToSample
2021/01/22 16:28:26.3 Temp: 2021/01/22 16:28:26, 25.0000, 25.4000
2021/01/22 16:28:26.4 GMC_SetPosZ
2021/01/22 16:28:32.4 GMC_ServoHere
2021/01/22 16:28:32.6 GMC_SetPosXY
2021/01/22 16:28:32.9 GMC_SetPosZ
2021/01/22 16:28:38.9 Shaker current speed = 494 rpm, set speed = 500
2021/01/22 16:28:39.0 Old State:13-cycle=1, Elapsed Time:    0.0,  Ideal Time:    0.0,  Delay time =   0.00  New State:ENGN_StateRunning_GoingToSampleComplete
2021/01/22 16:28:39.1 Old State:14-ENGN_StateRunning_GoingToSampleComplete New State:ENGN_StateRunning_WaitSampling
2021/01/22 16:28:39.1 Step started at: 16:28:26
2021/01/22 16:28:39.1 XYZ movement time  = 12690 ms + Scan start delay = 12602 ms
2021/01/22 16:28:39.2 Old State:15-ENGN_StateRunning_WaitSampling New State:ENGN_StateRunning_Sampling
2021/01/22 16:28:43.8 
2021/01/22 16:28:43.8 STOP SCAN: 
2021/01/22 16:28:43.8    CycleTime (ms)    = 200
2021/01/22 16:28:43.8    CurrentCycle      = 25
2021/01/22 16:28:43.8    RunTime (ms)      = 4958
2021/01/22 16:28:43.8    AverageCycle (ms) = 198.3
2021/01/22 16:28:43.8    Temperature       = 25.0
2021/01/22 16:28:43.8    DATA OVERRUN %    = 0.00
2021/01/22 16:28:43.8    Spec PCB temps    = 42.3, 41.9, 42.8, 42.5, 42.2, 42.0, 40.9, 41.2
2021/01/22 16:28:43.8 
2021/01/22 16:28:43.8 Old State:16-cycle=25, Elapsed Time:    5.0,  Ideal Time:    4.8,  Delay time =   0.04  New State:ENGN_StateRunning_SamplingComplete
2021/01/22 16:28:44.0 Old State:18-ENGN_StateRunning_SamplingComplete New State:ENGN_StateRunning_WaitNewStep
2021/01/22 16:28:44.3 Old State:5-ENGN_StateRunning_WaitNewStep New State:ENGN_StateRunning_StartNewStep
2021/01/22 16:28:44.3 Spectrometer EOS_B0712: AVRG=20.
2021/01/22 16:28:44.3 Spectrometer EOS_B0734: AVRG=20.
2021/01/22 16:28:44.3 Spectrometer EOS_B0814: AVRG=20.
2021/01/22 16:28:44.3 Spectrometer EOS_B0735: AVRG=20.
2021/01/22 16:28:44.3 Spectrometer EOS_B0732: AVRG=20.
2021/01/22 16:28:44.3 Spectrometer EOS_B0808: AVRG=20.
2021/01/22 16:28:44.3 Spectrometer EOS_B0733: AVRG=20.
2021/01/22 16:28:44.3 Spectrometer EOS_B0817: AVRG=20.
2021/01/22 16:28:44.3 StartShaker(500)
2021/01/22 16:28:44.4 Old State:6-ENGN_StateRunning_StartNewStep New State:ENGN_StateRunning_WaitGoingToSample
2021/01/22 16:28:44.4 Sample96_Q(0, 0, 11)
2021/01/22 16:28:44.5 
2021/01/22 16:28:44.5 START SCAN: 
2021/01/22 16:28:44.5    CycleTime (ms)    = 200
2021/01/22 16:28:44.5    CycleCount        = 25
2021/01/22 16:28:44.5    ScansToAverage    = 20
2021/01/22 16:28:44.5    FinishTime (s)    = 5
2021/01/22 16:28:44.5    Temperature       = 25.0
2021/01/22 16:28:44.5    Lamp Power        = 31
2021/01/22 16:28:44.5    Spec PCB temps    = 42.3, 41.9, 42.8, 42.5, 42.1, 42.0, 40.9, 41.2
2021/01/22 16:28:44.5 
2021/01/22 16:28:44.6 Old State:12-ENGN_StateRunning_WaitGoingToSample New State:ENGN_StateRunning_GoingToSample
2021/01/22 16:28:44.6 GMC_SetPosZ
2021/01/22 16:28:44.6 Temp: 2021/01/22 16:28:44, 25.0000, 25.4000
2021/01/22 16:28:50.3 GMC_ServoHere
2021/01/22 16:28:50.5 GMC_SetPosXY
2021/01/22 16:28:50.8 GMC_SetPosZ
2021/01/22 16:28:56.9 Shaker current speed = 494 rpm, set speed = 500
2021/01/22 16:28:57.0 Old State:13-ENGN_StateRunning_GoingToSample New State:ENGN_StateRunning_GoingToSampleComplete
2021/01/22 16:28:57.1 Old State:14-cycle=1, Elapsed Time:    0.0,  Ideal Time:    0.0,  Delay time =   0.00  New State:ENGN_StateRunning_WaitSampling
2021/01/22 16:28:57.1 Step started at: 16:28:44
2021/01/22 16:28:57.1 XYZ movement time  = 12468 ms + Scan start delay = 12425 ms
2021/01/22 16:28:57.2 Old State:15-ENGN_StateRunning_WaitSampling New State:ENGN_StateRunning_Sampling
2021/01/22 16:29:01.8 
2021/01/22 16:29:01.8 STOP SCAN: 
2021/01/22 16:29:01.8    CycleTime (ms)    = 200
2021/01/22 16:29:01.8    CurrentCycle      = 25
2021/01/22 16:29:01.8    RunTime (ms)      = 4957
2021/01/22 16:29:01.8    AverageCycle (ms) = 198.3
2021/01/22 16:29:01.8    Temperature       = 25.0
2021/01/22 16:29:01.8    DATA OVERRUN %    = 0.00
2021/01/22 16:29:01.8    Spec PCB temps    = 42.3, 41.9, 42.7, 42.4, 42.2, 42.0, 40.9, 41.2
2021/01/22 16:29:01.8 
2021/01/22 16:29:01.9 Old State:16-cycle=25, Elapsed Time:    5.0,  Ideal Time:    4.8,  Delay time =   0.04  New State:ENGN_StateRunning_SamplingComplete
2021/01/22 16:29:02.0 Old State:18-ENGN_StateRunning_SamplingComplete New State:ENGN_StateRunning_WaitNewStep
2021/01/22 16:29:02.1 Old State:5-ENGN_StateRunning_WaitNewStep New State:ENGN_StateRunning_StartNewStep
2021/01/22 16:29:02.1 Spectrometer EOS_B0712: AVRG=20.
2021/01/22 16:29:02.1 Spectrometer EOS_B0734: AVRG=20.
2021/01/22 16:29:02.1 Spectrometer EOS_B0814: AVRG=20.
2021/01/22 16:29:02.1 Spectrometer EOS_B0735: AVRG=20.
2021/01/22 16:29:02.1 Spectrometer EOS_B0732: AVRG=20.
2021/01/22 16:29:02.1 Spectrometer EOS_B0808: AVRG=20.
2021/01/22 16:29:02.1 Spectrometer EOS_B0733: AVRG=20.
2021/01/22 16:29:02.1 Spectrometer EOS_B0817: AVRG=20.
2021/01/22 16:29:02.1 StartShaker(500)
2021/01/22 16:29:02.3 Old State:6-ENGN_StateRunning_StartNewStep New State:ENGN_StateRunning_WaitGoingToSample
2021/01/22 16:29:02.3 Sample96_Q(0, 0, 10)
2021/01/22 16:29:02.3 
2021/01/22 16:29:02.3 START SCAN: 
2021/01/22 16:29:02.3    CycleTime (ms)    = 200
2021/01/22 16:29:02.3    CycleCount        = 25
2021/01/22 16:29:02.3    ScansToAverage    = 20
2021/01/22 16:29:02.3    FinishTime (s)    = 5
2021/01/22 16:29:02.3    Temperature       = 25.0
2021/01/22 16:29:02.3    Lamp Power        = 31
2021/01/22 16:29:02.3    Spec PCB temps    = 42.3, 41.9, 42.7, 42.4, 42.2, 42.0, 40.9, 41.2
2021/01/22 16:29:02.3 
2021/01/22 16:29:02.3 GMC_SetPosZ
2021/01/22 16:29:02.4 Old State:12-ENGN_StateRunning_WaitGoingToSample New State:ENGN_StateRunning_GoingToSample
2021/01/22 16:29:02.4 Temp: 2021/01/22 16:29:02, 25.0000, 25.5000
2021/01/22 16:29:08.4 GMC_ServoHere
2021/01/22 16:29:08.5 GMC_SetPosXY
2021/01/22 16:29:08.9 GMC_SetPosZ
2021/01/22 16:29:14.6 Shaker current speed = 494 rpm, set speed = 500
2021/01/22 16:29:14.7 Old State:13-ENGN_StateRunning_GoingToSample New State:ENGN_StateRunning_GoingToSampleComplete
2021/01/22 16:29:14.8 Old State:14-cycle=1, Elapsed Time:    0.0,  Ideal Time:    0.0,  Delay time =   0.00  New State:ENGN_StateRunning_WaitSampling
2021/01/22 16:29:14.8 Step started at: 16:29:02
2021/01/22 16:29:14.8 XYZ movement time  = 12379 ms + Scan start delay = 12359 ms
2021/01/22 16:29:14.9 Old State:15-ENGN_StateRunning_WaitSampling New State:ENGN_StateRunning_Sampling
2021/01/22 16:29:19.6 
2021/01/22 16:29:19.6 STOP SCAN: 
2021/01/22 16:29:19.6    CycleTime (ms)    = 200
2021/01/22 16:29:19.6    CurrentCycle      = 25
2021/01/22 16:29:19.6    RunTime (ms)      = 4958
2021/01/22 16:29:19.6    AverageCycle (ms) = 198.3
2021/01/22 16:29:19.6    Temperature       = 25.0
2021/01/22 16:29:19.6    DATA OVERRUN %    = 0.00
2021/01/22 16:29:19.6    Spec PCB temps    = 42.3, 41.9, 42.7, 42.4, 42.2, 42.0, 40.8, 41.2
2021/01/22 16:29:19.6 
2021/01/22 16:29:19.7 Old State:16-cycle=25, Elapsed Time:    5.0,  Ideal Time:    4.8,  Delay time =   0.04  New State:ENGN_StateRunning_SamplingComplete
2021/01/22 16:29:19.8 Old State:18-ENGN_StateRunning_SamplingComplete New State:ENGN_StateRunning_WaitNewStep
2021/01/22 16:29:20.0 Old State:5-ENGN_StateRunning_WaitNewStep New State:ENGN_StateRunning_StartNewStep
2021/01/22 16:29:20.1 Spectrometer EOS_B0712: AVRG=20.
2021/01/22 16:29:20.1 Spectrometer EOS_B0734: AVRG=20.
2021/01/22 16:29:20.1 Spectrometer EOS_B0814: AVRG=20.
2021/01/22 16:29:20.1 Spectrometer EOS_B0735: AVRG=20.
2021/01/22 16:29:20.1 Spectrometer EOS_B0732: AVRG=20.
2021/01/22 16:29:20.1 Spectrometer EOS_B0808: AVRG=20.
2021/01/22 16:29:20.1 Spectrometer EOS_B0733: AVRG=20.
2021/01/22 16:29:20.1 Spectrometer EOS_B0817: AVRG=20.
2021/01/22 16:29:20.1 StartShaker(500)
2021/01/22 16:29:20.2 Old State:6-ENGN_StateRunning_StartNewStep New State:ENGN_StateRunning_WaitGoingToSample
2021/01/22 16:29:20.2 Sample96_Q(0, 0, 3)
2021/01/22 16:29:20.2 
2021/01/22 16:29:20.2 START SCAN: 
2021/01/22 16:29:20.2    CycleTime (ms)    = 200
2021/01/22 16:29:20.2    CycleCount        = 1500
2021/01/22 16:29:20.2    ScansToAverage    = 20
2021/01/22 16:29:20.2    FinishTime (s)    = 300
2021/01/22 16:29:20.2    Temperature       = 25.0
2021/01/22 16:29:20.2    Lamp Power        = 31
2021/01/22 16:29:20.2    Spec PCB temps    = 42.3, 41.9, 42.7, 42.4, 42.2, 42.0, 40.8, 41.2
2021/01/22 16:29:20.2 
2021/01/22 16:29:20.2 GMC_SetPosZ
2021/01/22 16:29:20.3 Old State:12-ENGN_StateRunning_WaitGoingToSample New State:ENGN_StateRunning_GoingToSample
2021/01/22 16:29:20.3 Temp: 2021/01/22 16:29:20, 25.0000, 25.5000
2021/01/22 16:29:26.0 GMC_ServoHere
2021/01/22 16:29:26.2 GMC_SetPosXY
2021/01/22 16:29:27.4 GMC_SetPosZ
2021/01/22 16:29:31.4 Shaker current speed = 494 rpm, set speed = 500
2021/01/22 16:29:31.5 Old State:13-ENGN_StateRunning_GoingToSample New State:ENGN_StateRunning_GoingToSampleComplete
2021/01/22 16:29:31.6 Old State:14-cycle=1, Elapsed Time:    0.0,  Ideal Time:    0.0,  Delay time =   0.00  New State:ENGN_StateRunning_WaitSampling
2021/01/22 16:29:31.6 Step started at: 16:29:20
2021/01/22 16:29:31.6 XYZ movement time  = 11171 ms + Scan start delay = 11149 ms
2021/01/22 16:29:31.7 Old State:15-ENGN_StateRunning_WaitSampling New State:ENGN_StateRunning_Sampling
2021/01/22 16:30:20.5 Temp: 2021/01/22 16:30:20, 25.0000, 25.5000
2021/01/22 16:31:20.6 Temp: 2021/01/22 16:31:20, 25.0000, 25.5000
2021/01/22 16:32:20.8 Temp: 2021/01/22 16:32:20, 25.0000, 25.5000
2021/01/22 16:33:20.9 Temp: 2021/01/22 16:33:20, 25.0000, 25.5000
2021/01/22 16:34:21.1 Temp: 2021/01/22 16:34:21, 25.0000, 25.5000
2021/01/22 16:34:31.3 
2021/01/22 16:34:31.3 STOP SCAN: 
2021/01/22 16:34:31.3    CycleTime (ms)    = 200
2021/01/22 16:34:31.3    CurrentCycle      = 1500
2021/01/22 16:34:31.3    RunTime (ms)      = 299957
2021/01/22 16:34:31.3    AverageCycle (ms) = 200.0
2021/01/22 16:34:31.3    Temperature       = 25.0
2021/01/22 16:34:31.3    DATA OVERRUN %    = 0.00
2021/01/22 16:34:31.3    Spec PCB temps    = 42.3, 41.9, 42.8, 42.4, 42.2, 42.0, 40.9, 41.2
2021/01/22 16:34:31.3 
2021/01/22 16:34:31.4 Old State:16-cycle=1500, Elapsed Time:  300.0,  Ideal Time:  299.8,  Delay time =   0.04  New State:ENGN_StateRunning_SamplingComplete
2021/01/22 16:34:31.5 Old State:18-ENGN_StateRunning_SamplingComplete New State:ENGN_StateRunning_WaitNewStep
2021/01/22 16:34:31.8 Old State:5-ENGN_StateRunning_WaitNewStep New State:ENGN_StateRunning_StartNewStep
2021/01/22 16:34:31.9 Spectrometer EOS_B0712: AVRG=20.
2021/01/22 16:34:31.9 Spectrometer EOS_B0734: AVRG=20.
2021/01/22 16:34:31.9 Spectrometer EOS_B0814: AVRG=20.
2021/01/22 16:34:31.9 Spectrometer EOS_B0735: AVRG=20.
2021/01/22 16:34:31.9 Spectrometer EOS_B0732: AVRG=20.
2021/01/22 16:34:31.9 Spectrometer EOS_B0808: AVRG=20.
2021/01/22 16:34:31.9 Spectrometer EOS_B0733: AVRG=20.
2021/01/22 16:34:31.9 Spectrometer EOS_B0817: AVRG=20.
2021/01/22 16:34:31.9 StartShaker(500)
2021/01/22 16:34:32.0 Old State:6-ENGN_StateRunning_StartNewStep New State:ENGN_StateRunning_WaitGoingToSample
2021/01/22 16:34:32.0 Sample96_Q(0, 0, 6)
2021/01/22 16:34:32.0 
2021/01/22 16:34:32.0 START SCAN: 
2021/01/22 16:34:32.0    CycleTime (ms)    = 200
2021/01/22 16:34:32.0    CycleCount        = 250
2021/01/22 16:34:32.0    ScansToAverage    = 20
2021/01/22 16:34:32.0    FinishTime (s)    = 50
2021/01/22 16:34:32.0    Temperature       = 25.0
2021/01/22 16:34:32.0    Lamp Power        = 31
2021/01/22 16:34:32.0    Spec PCB temps    = 42.3, 41.9, 42.8, 42.4, 42.2, 42.0, 40.9, 41.2
2021/01/22 16:34:32.0 
2021/01/22 16:34:32.0 GMC_SetPosZ
2021/01/22 16:34:32.1 Temp: 2021/01/22 16:34:32, 25.0000, 25.4000
2021/01/22 16:34:32.1 Old State:12-ENGN_StateRunning_WaitGoingToSample New State:ENGN_StateRunning_GoingToSample
2021/01/22 16:34:35.8 GMC_ServoHere
2021/01/22 16:34:36.0 GMC_SetPosXY
2021/01/22 16:34:36.6 GMC_SetPosZ
2021/01/22 16:34:41.5 Old State:13-cycle=1, Elapsed Time:    0.0,  Ideal Time:    0.0,  Delay time =   0.00  New State:ENGN_StateRunning_GoingToSampleComplete
2021/01/22 16:34:41.6 Old State:14-ENGN_StateRunning_GoingToSampleComplete New State:ENGN_StateRunning_WaitSampling
2021/01/22 16:34:41.6 Step started at: 16:34:32
2021/01/22 16:34:41.6 XYZ movement time  = 9375 ms + Scan start delay = 9303 ms
2021/01/22 16:34:41.7 Old State:15-ENGN_StateRunning_WaitSampling New State:ENGN_StateRunning_Sampling
2021/01/22 16:34:42.0 Shaker current speed = 494 rpm, set speed = 500
2021/01/22 16:35:31.3 
2021/01/22 16:35:31.3 STOP SCAN: 
2021/01/22 16:35:31.3    CycleTime (ms)    = 200
2021/01/22 16:35:31.3    CurrentCycle      = 250
2021/01/22 16:35:31.3    RunTime (ms)      = 49956
2021/01/22 16:35:31.3    AverageCycle (ms) = 199.8
2021/01/22 16:35:31.3    Temperature       = 25.0
2021/01/22 16:35:31.3    DATA OVERRUN %    = 0.00
2021/01/22 16:35:31.3    Spec PCB temps    = 42.3, 41.9, 42.8, 42.5, 42.2, 42.0, 40.9, 41.2
2021/01/22 16:35:31.3 
2021/01/22 16:35:31.4 Old State:16-cycle=250, Elapsed Time:   50.0,  Ideal Time:   49.8,  Delay time =   0.04  New State:ENGN_StateRunning_SamplingComplete
2021/01/22 16:35:31.5 Old State:18-ENGN_StateRunning_SamplingComplete New State:ENGN_StateRunning_WaitNewStep
2021/01/22 16:35:31.6 Old State:5-ENGN_StateRunning_WaitNewStep New State:ENGN_StateRunning_StartNewStep
2021/01/22 16:35:31.6 Spectrometer EOS_B0712: AVRG=20.
2021/01/22 16:35:31.6 Spectrometer EOS_B0734: AVRG=20.
2021/01/22 16:35:31.6 Spectrometer EOS_B0814: AVRG=20.
2021/01/22 16:35:31.6 Spectrometer EOS_B0735: AVRG=20.
2021/01/22 16:35:31.6 Spectrometer EOS_B0732: AVRG=20.
2021/01/22 16:35:31.6 Spectrometer EOS_B0808: AVRG=20.
2021/01/22 16:35:31.6 Spectrometer EOS_B0733: AVRG=20.
2021/01/22 16:35:31.6 Spectrometer EOS_B0817: AVRG=20.
2021/01/22 16:35:31.6 StartShaker(500)
2021/01/22 16:35:31.8 Old State:6-ENGN_StateRunning_StartNewStep New State:ENGN_StateRunning_WaitGoingToSample
2021/01/22 16:35:31.8 Sample96_Q(0, 0, 3)
2021/01/22 16:35:31.8 
2021/01/22 16:35:31.8 START SCAN: 
2021/01/22 16:35:31.8    CycleTime (ms)    = 200
2021/01/22 16:35:31.8    CycleCount        = 1500
2021/01/22 16:35:31.8    ScansToAverage    = 20
2021/01/22 16:35:31.8    FinishTime (s)    = 300
2021/01/22 16:35:31.8    Temperature       = 25.0
2021/01/22 16:35:31.8    Lamp Power        = 31
2021/01/22 16:35:31.8    Spec PCB temps    = 42.3, 41.9, 42.8, 42.4, 42.2, 41.9, 40.9, 41.2
2021/01/22 16:35:31.8 
2021/01/22 16:35:31.9 Old State:12-ENGN_StateRunning_WaitGoingToSample New State:ENGN_StateRunning_GoingToSample
2021/01/22 16:35:31.9 GMC_SetPosZ
2021/01/22 16:35:31.9 Temp: 2021/01/22 16:35:31, 25.0000, 25.5000
2021/01/22 16:35:36.6 GMC_ServoHere
2021/01/22 16:35:36.8 GMC_SetPosXY
2021/01/22 16:35:37.4 GMC_SetPosZ
2021/01/22 16:35:41.5 Old State:13-cycle=1, Elapsed Time:    0.0,  Ideal Time:    0.0,  Delay time =   0.00  New State:ENGN_StateRunning_GoingToSampleComplete
2021/01/22 16:35:41.6 Old State:14-ENGN_StateRunning_GoingToSampleComplete New State:ENGN_StateRunning_WaitSampling
2021/01/22 16:35:41.6 Step started at: 16:35:31
2021/01/22 16:35:41.6 XYZ movement time  = 9670 ms + Scan start delay = 9606 ms
2021/01/22 16:35:41.7 Old State:15-ENGN_StateRunning_WaitSampling New State:ENGN_StateRunning_Sampling
2021/01/22 16:35:41.7 Shaker current speed = 494 rpm, set speed = 500
2021/01/22 16:36:32.1 Temp: 2021/01/22 16:36:32, 25.0000, 25.5000
2021/01/22 16:37:32.2 Temp: 2021/01/22 16:37:32, 25.0000, 25.5000
2021/01/22 16:38:32.4 Temp: 2021/01/22 16:38:32, 25.0000, 25.5000
2021/01/22 16:39:32.5 Temp: 2021/01/22 16:39:32, 25.0000, 25.5000
2021/01/22 16:40:32.7 Temp: 2021/01/22 16:40:32, 25.0000, 25.5000
2021/01/22 16:40:41.3 
2021/01/22 16:40:41.3 STOP SCAN: 
2021/01/22 16:40:41.3    CycleTime (ms)    = 200
2021/01/22 16:40:41.3    CurrentCycle      = 1500
2021/01/22 16:40:41.3    RunTime (ms)      = 299957
2021/01/22 16:40:41.3    AverageCycle (ms) = 200.0
2021/01/22 16:40:41.3    Temperature       = 25.0
2021/01/22 16:40:41.3    DATA OVERRUN %    = 0.00
2021/01/22 16:40:41.3    Spec PCB temps    = 42.3, 42.0, 42.7, 42.4, 42.2, 42.0, 40.9, 41.2
2021/01/22 16:40:41.3 
2021/01/22 16:40:41.4 Old State:16-cycle=1500, Elapsed Time:  300.0,  Ideal Time:  299.8,  Delay time =   0.04  New State:ENGN_StateRunning_SamplingComplete
2021/01/22 16:40:41.6 Old State:18-ENGN_StateRunning_SamplingComplete New State:ENGN_StateRunning_WaitNewStep
2021/01/22 16:40:41.7 Old State:5-ENGN_StateRunning_WaitNewStep New State:ENGN_StateRunning_StartNewStep
2021/01/22 16:40:41.7 Spectrometer EOS_B0712: AVRG=20.
2021/01/22 16:40:41.7 Spectrometer EOS_B0734: AVRG=20.
2021/01/22 16:40:41.7 Spectrometer EOS_B0814: AVRG=20.
2021/01/22 16:40:41.7 Spectrometer EOS_B0735: AVRG=20.
2021/01/22 16:40:41.7 Spectrometer EOS_B0732: AVRG=20.
2021/01/22 16:40:41.7 Spectrometer EOS_B0808: AVRG=20.
2021/01/22 16:40:41.7 Spectrometer EOS_B0733: AVRG=20.
2021/01/22 16:40:41.7 Spectrometer EOS_B0817: AVRG=20.
2021/01/22 16:40:41.7 StartShaker(500)
2021/01/22 16:40:41.8 Old State:6-ENGN_StateRunning_StartNewStep New State:ENGN_StateRunning_WaitGoingToSample
2021/01/22 16:40:41.8 Sample96_Q(0, 0, 11)
2021/01/22 16:40:41.8 
2021/01/22 16:40:41.8 START SCAN: 
2021/01/22 16:40:41.8    CycleTime (ms)    = 200
2021/01/22 16:40:41.8    CycleCount        = 25
2021/01/22 16:40:41.8    ScansToAverage    = 20
2021/01/22 16:40:41.8    FinishTime (s)    = 5
2021/01/22 16:40:41.8    Temperature       = 25.0
2021/01/22 16:40:41.8    Lamp Power        = 31
2021/01/22 16:40:41.8    Spec PCB temps    = 42.3, 42.0, 42.8, 42.4, 42.2, 41.9, 40.9, 41.2
2021/01/22 16:40:41.8 
2021/01/22 16:40:41.9 GMC_SetPosZ
2021/01/22 16:40:41.9 Old State:12-ENGN_StateRunning_WaitGoingToSample New State:ENGN_StateRunning_GoingToSample
2021/01/22 16:40:42.0 Temp: 2021/01/22 16:40:42, 25.0000, 25.6000
2021/01/22 16:40:45.7 GMC_ServoHere
2021/01/22 16:40:45.8 GMC_SetPosXY
2021/01/22 16:40:47.2 GMC_SetPosZ
2021/01/22 16:40:53.3 Shaker current speed = 494 rpm, set speed = 500
2021/01/22 16:40:53.4 Old State:13-ENGN_StateRunning_GoingToSample New State:ENGN_StateRunning_GoingToSampleComplete
2021/01/22 16:40:53.5 Old State:14-cycle=1, Elapsed Time:    0.0,  Ideal Time:    0.0,  Delay time =   0.00  New State:ENGN_StateRunning_WaitSampling
2021/01/22 16:40:53.5 Step started at: 16:40:41
2021/01/22 16:40:53.5 XYZ movement time  = 11502 ms + Scan start delay = 11452 ms
2021/01/22 16:40:53.6 Old State:15-ENGN_StateRunning_WaitSampling New State:ENGN_StateRunning_Sampling
2021/01/22 16:40:58.2 
2021/01/22 16:40:58.2 STOP SCAN: 
2021/01/22 16:40:58.2    CycleTime (ms)    = 200
2021/01/22 16:40:58.2    CurrentCycle      = 25
2021/01/22 16:40:58.2    RunTime (ms)      = 4958
2021/01/22 16:40:58.2    AverageCycle (ms) = 198.3
2021/01/22 16:40:58.2    Temperature       = 25.0
2021/01/22 16:40:58.2    DATA OVERRUN %    = 0.00
2021/01/22 16:40:58.2    Spec PCB temps    = 42.3, 41.9, 42.8, 42.4, 42.2, 42.0, 40.9, 41.2
2021/01/22 16:40:58.2 
2021/01/22 16:40:58.2 Old State:16-cycle=25, Elapsed Time:    5.0,  Ideal Time:    4.8,  Delay time =   0.04  New State:ENGN_StateRunning_SamplingComplete
2021/01/22 16:40:58.4 Old State:18-ENGN_StateRunning_SamplingComplete New State:ENGN_StateRunning_WaitNewStep
2021/01/22 16:40:58.5 Old State:5-ENGN_StateRunning_WaitNewStep New State:ENGN_StateRunning_StartNewStep
2021/01/22 16:40:58.5 Spectrometer EOS_B0712: AVRG=20.
2021/01/22 16:40:58.5 Spectrometer EOS_B0734: AVRG=20.
2021/01/22 16:40:58.5 Spectrometer EOS_B0814: AVRG=20.
2021/01/22 16:40:58.5 Spectrometer EOS_B0735: AVRG=20.
2021/01/22 16:40:58.5 Spectrometer EOS_B0732: AVRG=20.
2021/01/22 16:40:58.5 Spectrometer EOS_B0808: AVRG=20.
2021/01/22 16:40:58.5 Spectrometer EOS_B0733: AVRG=20.
2021/01/22 16:40:58.5 Spectrometer EOS_B0817: AVRG=20.
2021/01/22 16:40:58.5 StartShaker(500)
2021/01/22 16:40:58.6 Old State:6-ENGN_StateRunning_StartNewStep New State:ENGN_StateRunning_WaitGoingToSample
2021/01/22 16:40:58.6 Sample96_Q(0, 0, 10)
2021/01/22 16:40:58.7 
2021/01/22 16:40:58.7 START SCAN: 
2021/01/22 16:40:58.7    CycleTime (ms)    = 200
2021/01/22 16:40:58.7    CycleCount        = 25
2021/01/22 16:40:58.7    ScansToAverage    = 20
2021/01/22 16:40:58.7    FinishTime (s)    = 5
2021/01/22 16:40:58.7    Temperature       = 25.0
2021/01/22 16:40:58.7    Lamp Power        = 31
2021/01/22 16:40:58.7    Spec PCB temps    = 42.3, 41.9, 42.8, 42.4, 42.2, 42.0, 40.9, 41.2
2021/01/22 16:40:58.7 
2021/01/22 16:40:58.7 GMC_SetPosZ
2021/01/22 16:40:58.8 Old State:12-ENGN_StateRunning_WaitGoingToSample New State:ENGN_StateRunning_GoingToSample
2021/01/22 16:40:58.8 Temp: 2021/01/22 16:40:58, 25.0000, 25.5000
2021/01/22 16:41:04.8 GMC_ServoHere
2021/01/22 16:41:04.9 GMC_SetPosXY
2021/01/22 16:41:05.3 GMC_SetPosZ
2021/01/22 16:41:11.2 Shaker current speed = 494 rpm, set speed = 500
2021/01/22 16:41:11.3 Old State:13-cycle=1, Elapsed Time:    0.0,  Ideal Time:    0.0,  Delay time =   0.00  New State:ENGN_StateRunning_GoingToSampleComplete
2021/01/22 16:41:11.4 Old State:14-ENGN_StateRunning_GoingToSampleComplete New State:ENGN_StateRunning_WaitSampling
2021/01/22 16:41:11.4 Step started at: 16:40:58
2021/01/22 16:41:11.4 XYZ movement time  = 12579 ms + Scan start delay = 12519 ms
2021/01/22 16:41:11.5 Old State:15-ENGN_StateRunning_WaitSampling New State:ENGN_StateRunning_Sampling
2021/01/22 16:41:16.1 
2021/01/22 16:41:16.1 STOP SCAN: 
2021/01/22 16:41:16.1    CycleTime (ms)    = 200
2021/01/22 16:41:16.1    CurrentCycle      = 25
2021/01/22 16:41:16.1    RunTime (ms)      = 4956
2021/01/22 16:41:16.1    AverageCycle (ms) = 198.2
2021/01/22 16:41:16.1    Temperature       = 25.0
2021/01/22 16:41:16.1    DATA OVERRUN %    = 0.00
2021/01/22 16:41:16.1    Spec PCB temps    = 42.3, 41.9, 42.7, 42.4, 42.2, 41.9, 40.9, 41.2
2021/01/22 16:41:16.1 
2021/01/22 16:41:16.2 Old State:16-cycle=25, Elapsed Time:    5.0,  Ideal Time:    4.8,  Delay time =   0.04  New State:ENGN_StateRunning_SamplingComplete
2021/01/22 16:41:16.3 Old State:18-ENGN_StateRunning_SamplingComplete New State:ENGN_StateRunning_WaitNewStep
2021/01/22 16:41:16.4 Old State:5-ENGN_StateRunning_WaitNewStep New State:ENGN_StateRunning_StartNewStep
2021/01/22 16:41:16.4 Spectrometer EOS_B0712: AVRG=20.
2021/01/22 16:41:16.4 Spectrometer EOS_B0734: AVRG=20.
2021/01/22 16:41:16.4 Spectrometer EOS_B0814: AVRG=20.
2021/01/22 16:41:16.4 Spectrometer EOS_B0735: AVRG=20.
2021/01/22 16:41:16.4 Spectrometer EOS_B0732: AVRG=20.
2021/01/22 16:41:16.4 Spectrometer EOS_B0808: AVRG=20.
2021/01/22 16:41:16.4 Spectrometer EOS_B0733: AVRG=20.
2021/01/22 16:41:16.4 Spectrometer EOS_B0817: AVRG=20.
2021/01/22 16:41:16.4 StartShaker(500)
2021/01/22 16:41:16.5 Old State:6-ENGN_StateRunning_StartNewStep New State:ENGN_StateRunning_WaitGoingToSample
2021/01/22 16:41:16.5 Sample96_Q(0, 0, 11)
2021/01/22 16:41:16.5 
2021/01/22 16:41:16.5 START SCAN: 
2021/01/22 16:41:16.5    CycleTime (ms)    = 200
2021/01/22 16:41:16.5    CycleCount        = 25
2021/01/22 16:41:16.5    ScansToAverage    = 20
2021/01/22 16:41:16.5    FinishTime (s)    = 5
2021/01/22 16:41:16.5    Temperature       = 25.0
2021/01/22 16:41:16.5    Lamp Power        = 31
2021/01/22 16:41:16.5    Spec PCB temps    = 42.3, 41.9, 42.7, 42.4, 42.2, 41.9, 40.9, 41.2
2021/01/22 16:41:16.5 
2021/01/22 16:41:16.6 GMC_SetPosZ
2021/01/22 16:41:16.6 Old State:12-ENGN_StateRunning_WaitGoingToSample New State:ENGN_StateRunning_GoingToSample
2021/01/22 16:41:16.7 Temp: 2021/01/22 16:41:16, 25.0000, 25.6000
2021/01/22 16:41:22.4 GMC_ServoHere
2021/01/22 16:41:22.5 GMC_SetPosXY
2021/01/22 16:41:22.9 GMC_SetPosZ
2021/01/22 16:41:29.1 Shaker current speed = 494 rpm, set speed = 500
2021/01/22 16:41:29.2 Old State:13-ENGN_StateRunning_GoingToSample New State:ENGN_StateRunning_GoingToSampleComplete
2021/01/22 16:41:29.3 Old State:14-cycle=1, Elapsed Time:    0.0,  Ideal Time:    0.0,  Delay time =   0.00  New State:ENGN_StateRunning_WaitSampling
2021/01/22 16:41:29.3 Step started at: 16:41:16
2021/01/22 16:41:29.3 XYZ movement time  = 12574 ms + Scan start delay = 12538 ms
2021/01/22 16:41:29.4 Old State:15-ENGN_StateRunning_WaitSampling New State:ENGN_StateRunning_Sampling
2021/01/22 16:41:34.0 
2021/01/22 16:41:34.0 STOP SCAN: 
2021/01/22 16:41:34.0    CycleTime (ms)    = 200
2021/01/22 16:41:34.0    CurrentCycle      = 25
2021/01/22 16:41:34.0    RunTime (ms)      = 4958
2021/01/22 16:41:34.0    AverageCycle (ms) = 198.3
2021/01/22 16:41:34.0    Temperature       = 25.0
2021/01/22 16:41:34.0    DATA OVERRUN %    = 0.00
2021/01/22 16:41:34.0    Spec PCB temps    = 42.3, 41.9, 42.8, 42.4, 42.2, 42.0, 40.9, 41.2
2021/01/22 16:41:34.0 
2021/01/22 16:41:34.1 Old State:16-cycle=25, Elapsed Time:    5.0,  Ideal Time:    4.8,  Delay time =   0.04  New State:ENGN_StateRunning_SamplingComplete
2021/01/22 16:41:34.3 Old State:18-ENGN_StateRunning_SamplingComplete New State:ENGN_StateRunning_WaitNewStep
2021/01/22 16:41:34.6 Old State:5-ENGN_StateRunning_WaitNewStep New State:ENGN_StateRunning_StartNewStep
2021/01/22 16:41:34.6 Spectrometer EOS_B0712: AVRG=20.
2021/01/22 16:41:34.6 Spectrometer EOS_B0734: AVRG=20.
2021/01/22 16:41:34.6 Spectrometer EOS_B0814: AVRG=20.
2021/01/22 16:41:34.6 Spectrometer EOS_B0735: AVRG=20.
2021/01/22 16:41:34.6 Spectrometer EOS_B0732: AVRG=20.
2021/01/22 16:41:34.6 Spectrometer EOS_B0808: AVRG=20.
2021/01/22 16:41:34.6 Spectrometer EOS_B0733: AVRG=20.
2021/01/22 16:41:34.6 Spectrometer EOS_B0817: AVRG=20.
2021/01/22 16:41:34.6 StartShaker(500)
2021/01/22 16:41:34.8 Old State:6-ENGN_StateRunning_StartNewStep New State:ENGN_StateRunning_WaitGoingToSample
2021/01/22 16:41:34.8 Sample96_Q(0, 0, 10)
2021/01/22 16:41:34.8 
2021/01/22 16:41:34.8 START SCAN: 
2021/01/22 16:41:34.8    CycleTime (ms)    = 200
2021/01/22 16:41:34.8    CycleCount        = 25
2021/01/22 16:41:34.8    ScansToAverage    = 20
2021/01/22 16:41:34.8    FinishTime (s)    = 5
2021/01/22 16:41:34.8    Temperature       = 25.0
2021/01/22 16:41:34.8    Lamp Power        = 31
2021/01/22 16:41:34.8    Spec PCB temps    = 42.3, 41.9, 42.8, 42.4, 42.2, 42.0, 40.9, 41.2
2021/01/22 16:41:34.8 
2021/01/22 16:41:34.8 GMC_SetPosZ
2021/01/22 16:41:34.9 Old State:12-ENGN_StateRunning_WaitGoingToSample New State:ENGN_StateRunning_GoingToSample
2021/01/22 16:41:34.9 Temp: 2021/01/22 16:41:34, 25.0000, 25.6000
2021/01/22 16:41:40.9 GMC_ServoHere
2021/01/22 16:41:41.0 GMC_SetPosXY
2021/01/22 16:41:41.4 GMC_SetPosZ
2021/01/22 16:41:47.2 Shaker current speed = 494 rpm, set speed = 500
2021/01/22 16:41:47.3 Old State:13-ENGN_StateRunning_GoingToSample New State:ENGN_StateRunning_GoingToSampleComplete
2021/01/22 16:41:47.4 Old State:14-cycle=1, Elapsed Time:    0.0,  Ideal Time:    0.0,  Delay time =   0.00  New State:ENGN_StateRunning_WaitSampling
2021/01/22 16:41:47.4 Step started at: 16:41:34
2021/01/22 16:41:47.4 XYZ movement time  = 12385 ms + Scan start delay = 12375 ms
2021/01/22 16:41:47.5 Old State:15-ENGN_StateRunning_WaitSampling New State:ENGN_StateRunning_Sampling
2021/01/22 16:41:52.1 
2021/01/22 16:41:52.1 STOP SCAN: 
2021/01/22 16:41:52.1    CycleTime (ms)    = 200
2021/01/22 16:41:52.1    CurrentCycle      = 25
2021/01/22 16:41:52.1    RunTime (ms)      = 4959
2021/01/22 16:41:52.1    AverageCycle (ms) = 198.4
2021/01/22 16:41:52.1    Temperature       = 25.0
2021/01/22 16:41:52.1    DATA OVERRUN %    = 0.00
2021/01/22 16:41:52.1    Spec PCB temps    = 42.3, 41.9, 42.8, 42.4, 42.2, 41.9, 40.9, 41.3
2021/01/22 16:41:52.1 
2021/01/22 16:41:52.2 Old State:16-cycle=25, Elapsed Time:    5.0,  Ideal Time:    4.8,  Delay time =   0.04  New State:ENGN_StateRunning_SamplingComplete
2021/01/22 16:41:52.3 Old State:18-ENGN_StateRunning_SamplingComplete New State:ENGN_StateRunning_WaitNewStep
2021/01/22 16:41:52.5 Old State:5-ENGN_StateRunning_WaitNewStep New State:ENGN_StateRunning_StartNewStep
2021/01/22 16:41:52.6 Spectrometer EOS_B0712: AVRG=20.
2021/01/22 16:41:52.6 Spectrometer EOS_B0734: AVRG=20.
2021/01/22 16:41:52.6 Spectrometer EOS_B0814: AVRG=20.
2021/01/22 16:41:52.6 Spectrometer EOS_B0735: AVRG=20.
2021/01/22 16:41:52.6 Spectrometer EOS_B0732: AVRG=20.
2021/01/22 16:41:52.6 Spectrometer EOS_B0808: AVRG=20.
2021/01/22 16:41:52.6 Spectrometer EOS_B0733: AVRG=20.
2021/01/22 16:41:52.6 Spectrometer EOS_B0817: AVRG=20.
2021/01/22 16:41:52.6 StartShaker(500)
2021/01/22 16:41:52.7 Old State:6-ENGN_StateRunning_StartNewStep New State:ENGN_StateRunning_WaitGoingToSample
2021/01/22 16:41:52.7 Sample96_Q(0, 0, 11)
2021/01/22 16:41:52.7 
2021/01/22 16:41:52.7 START SCAN: 
2021/01/22 16:41:52.7    CycleTime (ms)    = 200
2021/01/22 16:41:52.7    CycleCount        = 25
2021/01/22 16:41:52.7    ScansToAverage    = 20
2021/01/22 16:41:52.7    FinishTime (s)    = 5
2021/01/22 16:41:52.7    Temperature       = 25.0
2021/01/22 16:41:52.7    Lamp Power        = 31
2021/01/22 16:41:52.7    Spec PCB temps    = 42.3, 41.9, 42.8, 42.4, 42.2, 41.9, 40.9, 41.3
2021/01/22 16:41:52.7 
2021/01/22 16:41:52.7 GMC_SetPosZ
2021/01/22 16:41:52.8 Temp: 2021/01/22 16:41:52, 25.0000, 25.7000
2021/01/22 16:41:52.8 Old State:12-ENGN_StateRunning_WaitGoingToSample New State:ENGN_StateRunning_GoingToSample
2021/01/22 16:41:58.5 GMC_ServoHere
2021/01/22 16:41:58.7 GMC_SetPosXY
2021/01/22 16:41:59.0 GMC_SetPosZ
2021/01/22 16:42:05.2 Old State:13-ENGN_StateRunning_GoingToSample New State:ENGN_StateRunning_GoingToSampleComplete
2021/01/22 16:42:05.2 Shaker current speed = 494 rpm, set speed = 500
2021/01/22 16:42:05.3 Old State:14-cycle=1, Elapsed Time:    0.0,  Ideal Time:    0.0,  Delay time =   0.00  New State:ENGN_StateRunning_WaitSampling
2021/01/22 16:42:05.3 Step started at: 16:41:52
2021/01/22 16:42:05.3 XYZ movement time  = 12392 ms + Scan start delay = 12392 ms
2021/01/22 16:42:05.4 Old State:15-ENGN_StateRunning_WaitSampling New State:ENGN_StateRunning_Sampling
2021/01/22 16:42:10.0 
2021/01/22 16:42:10.0 STOP SCAN: 
2021/01/22 16:42:10.0    CycleTime (ms)    = 200
2021/01/22 16:42:10.0    CurrentCycle      = 25
2021/01/22 16:42:10.0    RunTime (ms)      = 4958
2021/01/22 16:42:10.0    AverageCycle (ms) = 198.3
2021/01/22 16:42:10.0    Temperature       = 25.0
2021/01/22 16:42:10.0    DATA OVERRUN %    = 0.00
2021/01/22 16:42:10.0    Spec PCB temps    = 42.3, 41.9, 42.8, 42.4, 42.2, 42.0, 40.9, 41.3
2021/01/22 16:42:10.0 
2021/01/22 16:42:10.1 Old State:16-cycle=25, Elapsed Time:    5.0,  Ideal Time:    4.8,  Delay time =   0.04  New State:ENGN_StateRunning_SamplingComplete
2021/01/22 16:42:10.2 Old State:18-ENGN_StateRunning_SamplingComplete New State:ENGN_StateRunning_WaitNewStep
2021/01/22 16:42:10.3 Old State:5-ENGN_StateRunning_WaitNewStep New State:ENGN_StateRunning_StartNewStep
2021/01/22 16:42:10.4 Spectrometer EOS_B0712: AVRG=20.
2021/01/22 16:42:10.4 Spectrometer EOS_B0734: AVRG=20.
2021/01/22 16:42:10.4 Spectrometer EOS_B0814: AVRG=20.
2021/01/22 16:42:10.4 Spectrometer EOS_B0735: AVRG=20.
2021/01/22 16:42:10.4 Spectrometer EOS_B0732: AVRG=20.
2021/01/22 16:42:10.4 Spectrometer EOS_B0808: AVRG=20.
2021/01/22 16:42:10.4 Spectrometer EOS_B0733: AVRG=20.
2021/01/22 16:42:10.4 Spectrometer EOS_B0817: AVRG=20.
2021/01/22 16:42:10.4 StartShaker(500)
2021/01/22 16:42:10.5 Old State:6-ENGN_StateRunning_StartNewStep New State:ENGN_StateRunning_WaitGoingToSample
2021/01/22 16:42:10.5 Sample96_Q(0, 0, 10)
2021/01/22 16:42:10.5 
2021/01/22 16:42:10.5 START SCAN: 
2021/01/22 16:42:10.5    CycleTime (ms)    = 200
2021/01/22 16:42:10.5    CycleCount        = 25
2021/01/22 16:42:10.5    ScansToAverage    = 20
2021/01/22 16:42:10.5    FinishTime (s)    = 5
2021/01/22 16:42:10.5    Temperature       = 25.0
2021/01/22 16:42:10.5    Lamp Power        = 31
2021/01/22 16:42:10.5    Spec PCB temps    = 42.3, 41.9, 42.8, 42.4, 42.2, 42.0, 40.9, 41.3
2021/01/22 16:42:10.5 
2021/01/22 16:42:10.5 GMC_SetPosZ
2021/01/22 16:42:10.6 Old State:12-ENGN_StateRunning_WaitGoingToSample New State:ENGN_StateRunning_GoingToSample
2021/01/22 16:42:10.7 Temp: 2021/01/22 16:42:10, 25.0000, 25.6000
2021/01/22 16:42:16.6 GMC_ServoHere
2021/01/22 16:42:16.7 GMC_SetPosXY
2021/01/22 16:42:17.1 GMC_SetPosZ
2021/01/22 16:42:23.0 Shaker current speed = 494 rpm, set speed = 500
2021/01/22 16:42:23.1 Old State:13-ENGN_StateRunning_GoingToSample New State:ENGN_StateRunning_GoingToSampleComplete
2021/01/22 16:42:23.2 Old State:14-cycle=1, Elapsed Time:    0.0,  Ideal Time:    0.0,  Delay time =   0.00  New State:ENGN_StateRunning_WaitSampling
2021/01/22 16:42:23.2 Step started at: 16:42:10
2021/01/22 16:42:23.2 XYZ movement time  = 12484 ms + Scan start delay = 12481 ms
2021/01/22 16:42:23.3 Old State:15-ENGN_StateRunning_WaitSampling New State:ENGN_StateRunning_Sampling
2021/01/22 16:42:27.9 
2021/01/22 16:42:27.9 STOP SCAN: 
2021/01/22 16:42:27.9    CycleTime (ms)    = 200
2021/01/22 16:42:27.9    CurrentCycle      = 25
2021/01/22 16:42:27.9    RunTime (ms)      = 4955
2021/01/22 16:42:27.9    AverageCycle (ms) = 198.2
2021/01/22 16:42:27.9    Temperature       = 25.0
2021/01/22 16:42:27.9    DATA OVERRUN %    = 0.00
2021/01/22 16:42:27.9    Spec PCB temps    = 42.3, 41.9, 42.8, 42.5, 42.1, 41.9, 40.9, 41.2
2021/01/22 16:42:27.9 
2021/01/22 16:42:28.0 Old State:16-cycle=25, Elapsed Time:    5.0,  Ideal Time:    4.8,  Delay time =   0.04  New State:ENGN_StateRunning_SamplingComplete
2021/01/22 16:42:28.1 Old State:18-ENGN_StateRunning_SamplingComplete New State:ENGN_StateRunning_WaitNewStep
2021/01/22 16:42:28.3 Old State:5-ENGN_StateRunning_WaitNewStep New State:ENGN_StateRunning_StartNewStep
2021/01/22 16:42:28.3 Spectrometer EOS_B0712: AVRG=20.
2021/01/22 16:42:28.3 Spectrometer EOS_B0734: AVRG=20.
2021/01/22 16:42:28.3 Spectrometer EOS_B0814: AVRG=20.
2021/01/22 16:42:28.3 Spectrometer EOS_B0735: AVRG=20.
2021/01/22 16:42:28.3 Spectrometer EOS_B0732: AVRG=20.
2021/01/22 16:42:28.3 Spectrometer EOS_B0808: AVRG=20.
2021/01/22 16:42:28.3 Spectrometer EOS_B0733: AVRG=20.
2021/01/22 16:42:28.3 Spectrometer EOS_B0817: AVRG=20.
2021/01/22 16:42:28.3 StartShaker(500)
2021/01/22 16:42:28.4 Old State:6-ENGN_StateRunning_StartNewStep New State:ENGN_StateRunning_WaitGoingToSample
2021/01/22 16:42:28.4 Sample96_Q(0, 0, 3)
2021/01/22 16:42:28.4 
2021/01/22 16:42:28.4 START SCAN: 
2021/01/22 16:42:28.4    CycleTime (ms)    = 200
2021/01/22 16:42:28.4    CycleCount        = 1500
2021/01/22 16:42:28.4    ScansToAverage    = 20
2021/01/22 16:42:28.4    FinishTime (s)    = 300
2021/01/22 16:42:28.4    Temperature       = 25.0
2021/01/22 16:42:28.4    Lamp Power        = 31
2021/01/22 16:42:28.4    Spec PCB temps    = 42.3, 41.9, 42.8, 42.4, 42.1, 41.9, 40.9, 41.2
2021/01/22 16:42:28.4 
2021/01/22 16:42:28.5 Old State:12-ENGN_StateRunning_WaitGoingToSample New State:ENGN_StateRunning_GoingToSample
2021/01/22 16:42:28.5 Temp: 2021/01/22 16:42:28, 25.0000, 25.6000
2021/01/22 16:42:28.5 GMC_SetPosZ
2021/01/22 16:42:34.3 GMC_ServoHere
2021/01/22 16:42:34.5 GMC_SetPosXY
2021/01/22 16:42:35.7 GMC_SetPosZ
2021/01/22 16:42:39.6 Shaker current speed = 494 rpm, set speed = 500
2021/01/22 16:42:39.7 Old State:13-cycle=1, Elapsed Time:    0.0,  Ideal Time:    0.0,  Delay time =   0.00  New State:ENGN_StateRunning_GoingToSampleComplete
2021/01/22 16:42:39.8 Old State:14-ENGN_StateRunning_GoingToSampleComplete New State:ENGN_StateRunning_WaitSampling
2021/01/22 16:42:39.8 Step started at: 16:42:28
2021/01/22 16:42:39.8 XYZ movement time  = 11171 ms + Scan start delay = 11092 ms
2021/01/22 16:42:39.9 Old State:15-ENGN_StateRunning_WaitSampling New State:ENGN_StateRunning_Sampling
2021/01/22 16:43:28.7 Temp: 2021/01/22 16:43:28, 25.0000, 25.6000
2021/01/22 16:44:28.8 Temp: 2021/01/22 16:44:28, 25.0000, 25.7000
2021/01/22 16:45:29.0 Temp: 2021/01/22 16:45:29, 25.0000, 25.6000
2021/01/22 16:46:29.2 Temp: 2021/01/22 16:46:29, 25.0000, 25.7000
2021/01/22 16:47:29.3 Temp: 2021/01/22 16:47:29, 25.0000, 25.6000
2021/01/22 16:47:39.5 
2021/01/22 16:47:39.5 STOP SCAN: 
2021/01/22 16:47:39.5    CycleTime (ms)    = 200
2021/01/22 16:47:39.5    CurrentCycle      = 1500
2021/01/22 16:47:39.5    RunTime (ms)      = 299958
2021/01/22 16:47:39.5    AverageCycle (ms) = 200.0
2021/01/22 16:47:39.5    Temperature       = 25.0
2021/01/22 16:47:39.5    DATA OVERRUN %    = 0.00
2021/01/22 16:47:39.5    Spec PCB temps    = 42.3, 42.0, 42.8, 42.5, 42.2, 42.0, 40.9, 41.3
2021/01/22 16:47:39.5 
2021/01/22 16:47:39.6 Old State:16-cycle=1500, Elapsed Time:  300.0,  Ideal Time:  299.8,  Delay time =   0.04  New State:ENGN_StateRunning_SamplingComplete
2021/01/22 16:47:39.7 Old State:18-ENGN_StateRunning_SamplingComplete New State:ENGN_StateRunning_WaitNewStep
2021/01/22 16:47:40.0 Old State:5-ENGN_StateRunning_WaitNewStep New State:ENGN_StateRunning_StartNewStep
2021/01/22 16:47:40.1 Spectrometer EOS_B0712: AVRG=20.
2021/01/22 16:47:40.1 Spectrometer EOS_B0734: AVRG=20.
2021/01/22 16:47:40.1 Spectrometer EOS_B0814: AVRG=20.
2021/01/22 16:47:40.1 Spectrometer EOS_B0735: AVRG=20.
2021/01/22 16:47:40.1 Spectrometer EOS_B0732: AVRG=20.
2021/01/22 16:47:40.1 Spectrometer EOS_B0808: AVRG=20.
2021/01/22 16:47:40.1 Spectrometer EOS_B0733: AVRG=20.
2021/01/22 16:47:40.1 Spectrometer EOS_B0817: AVRG=20.
2021/01/22 16:47:40.1 StartShaker(500)
2021/01/22 16:47:40.2 Old State:6-ENGN_StateRunning_StartNewStep New State:ENGN_StateRunning_WaitGoingToSample
2021/01/22 16:47:40.2 Sample96_Q(0, 0, 7)
2021/01/22 16:47:40.2 
2021/01/22 16:47:40.2 START SCAN: 
2021/01/22 16:47:40.2    CycleTime (ms)    = 200
2021/01/22 16:47:40.2    CycleCount        = 250
2021/01/22 16:47:40.2    ScansToAverage    = 20
2021/01/22 16:47:40.2    FinishTime (s)    = 50
2021/01/22 16:47:40.2    Temperature       = 25.0
2021/01/22 16:47:40.2    Lamp Power        = 31
2021/01/22 16:47:40.2    Spec PCB temps    = 42.3, 41.9, 42.8, 42.5, 42.2, 42.0, 40.9, 41.2
2021/01/22 16:47:40.2 
2021/01/22 16:47:40.3 Old State:12-ENGN_StateRunning_WaitGoingToSample New State:ENGN_StateRunning_GoingToSample
2021/01/22 16:47:40.3 Temp: 2021/01/22 16:47:40, 25.0000, 25.6000
2021/01/22 16:47:40.3 GMC_SetPosZ
2021/01/22 16:47:44.1 GMC_ServoHere
2021/01/22 16:47:44.3 GMC_SetPosXY
2021/01/22 16:47:45.1 GMC_SetPosZ
2021/01/22 16:47:50.2 Old State:13-ENGN_StateRunning_GoingToSample New State:ENGN_StateRunning_GoingToSampleComplete
2021/01/22 16:47:50.2 Shaker current speed = 494 rpm, set speed = 500
2021/01/22 16:47:50.3 Old State:14-cycle=1, Elapsed Time:    0.0,  Ideal Time:    0.0,  Delay time =   0.00  New State:ENGN_StateRunning_WaitSampling
2021/01/22 16:47:50.3 Step started at: 16:47:40
2021/01/22 16:47:50.3 XYZ movement time  = 9880 ms + Scan start delay = 9880 ms
2021/01/22 16:47:50.4 Old State:15-ENGN_StateRunning_WaitSampling New State:ENGN_StateRunning_Sampling
2021/01/22 16:48:40.0 
2021/01/22 16:48:40.0 STOP SCAN: 
2021/01/22 16:48:40.0    CycleTime (ms)    = 200
2021/01/22 16:48:40.0    CurrentCycle      = 250
2021/01/22 16:48:40.0    RunTime (ms)      = 49956
2021/01/22 16:48:40.0    AverageCycle (ms) = 199.8
2021/01/22 16:48:40.0    Temperature       = 25.0
2021/01/22 16:48:40.0    DATA OVERRUN %    = 0.00
2021/01/22 16:48:40.0    Spec PCB temps    = 42.3, 42.0, 42.8, 42.4, 42.3, 42.0, 40.9, 41.2
2021/01/22 16:48:40.0 
2021/01/22 16:48:40.0 Old State:16-cycle=250, Elapsed Time:   50.0,  Ideal Time:   49.8,  Delay time =   0.04  New State:ENGN_StateRunning_SamplingComplete
2021/01/22 16:48:40.2 Old State:18-ENGN_StateRunning_SamplingComplete New State:ENGN_StateRunning_WaitNewStep
2021/01/22 16:48:40.5 Temp: 2021/01/22 16:48:40, 25.0000, 25.6000
2021/01/22 16:48:40.5 Old State:5-ENGN_StateRunning_WaitNewStep New State:ENGN_StateRunning_StartNewStep
2021/01/22 16:48:40.5 Spectrometer EOS_B0712: AVRG=20.
2021/01/22 16:48:40.5 Spectrometer EOS_B0734: AVRG=20.
2021/01/22 16:48:40.5 Spectrometer EOS_B0814: AVRG=20.
2021/01/22 16:48:40.5 Spectrometer EOS_B0735: AVRG=20.
2021/01/22 16:48:40.5 Spectrometer EOS_B0732: AVRG=20.
2021/01/22 16:48:40.5 Spectrometer EOS_B0808: AVRG=20.
2021/01/22 16:48:40.5 Spectrometer EOS_B0733: AVRG=20.
2021/01/22 16:48:40.5 Spectrometer EOS_B0817: AVRG=20.
2021/01/22 16:48:40.5 StartShaker(500)
2021/01/22 16:48:40.7 Old State:6-ENGN_StateRunning_StartNewStep New State:ENGN_StateRunning_WaitGoingToSample
2021/01/22 16:48:40.7 Sample96_Q(0, 0, 3)
2021/01/22 16:48:40.7 
2021/01/22 16:48:40.7 START SCAN: 
2021/01/22 16:48:40.7    CycleTime (ms)    = 200
2021/01/22 16:48:40.7    CycleCount        = 1500
2021/01/22 16:48:40.7    ScansToAverage    = 20
2021/01/22 16:48:40.7    FinishTime (s)    = 300
2021/01/22 16:48:40.7    Temperature       = 25.0
2021/01/22 16:48:40.7    Lamp Power        = 31
2021/01/22 16:48:40.7    Spec PCB temps    = 42.3, 42.0, 42.8, 42.5, 42.3, 42.0, 40.9, 41.2
2021/01/22 16:48:40.7 
2021/01/22 16:48:40.7 GMC_SetPosZ
2021/01/22 16:48:40.8 Old State:12-ENGN_StateRunning_WaitGoingToSample New State:ENGN_StateRunning_GoingToSample
2021/01/22 16:48:40.8 Temp: 2021/01/22 16:48:40, 25.0000, 25.7000
2021/01/22 16:48:45.7 GMC_ServoHere
2021/01/22 16:48:45.9 GMC_SetPosXY
2021/01/22 16:48:46.7 GMC_SetPosZ
2021/01/22 16:48:50.6 Shaker current speed = 494 rpm, set speed = 500
2021/01/22 16:48:50.7 Old State:13-ENGN_StateRunning_GoingToSample New State:ENGN_StateRunning_GoingToSampleComplete
2021/01/22 16:48:50.8 Old State:14-cycle=1, Elapsed Time:    0.0,  Ideal Time:    0.0,  Delay time =   0.00  New State:ENGN_StateRunning_WaitSampling
2021/01/22 16:48:50.8 Step started at: 16:48:40
2021/01/22 16:48:50.8 XYZ movement time  = 9967 ms + Scan start delay = 9950 ms
2021/01/22 16:48:50.9 Old State:15-ENGN_StateRunning_WaitSampling New State:ENGN_StateRunning_Sampling
2021/01/22 16:49:41.0 Temp: 2021/01/22 16:49:41, 25.0000, 25.7000
2021/01/22 16:50:41.1 Temp: 2021/01/22 16:50:41, 25.0000, 25.6000
2021/01/22 16:51:41.3 Temp: 2021/01/22 16:51:41, 25.0000, 25.7000
2021/01/22 16:52:41.4 Temp: 2021/01/22 16:52:41, 25.0000, 25.7000
2021/01/22 16:53:41.6 Temp: 2021/01/22 16:53:41, 25.0000, 25.6000
2021/01/22 16:53:50.6 
2021/01/22 16:53:50.6 STOP SCAN: 
2021/01/22 16:53:50.6    CycleTime (ms)    = 200
2021/01/22 16:53:50.6    CurrentCycle      = 1500
2021/01/22 16:53:50.6    RunTime (ms)      = 299959
2021/01/22 16:53:50.6    AverageCycle (ms) = 200.0
2021/01/22 16:53:50.6    Temperature       = 25.0
2021/01/22 16:53:50.6    DATA OVERRUN %    = 0.00
2021/01/22 16:53:50.6    Spec PCB temps    = 42.3, 41.9, 42.7, 42.4, 42.3, 42.0, 40.9, 41.2
2021/01/22 16:53:50.6 
2021/01/22 16:53:50.6 Old State:16-cycle=1500, Elapsed Time:  300.0,  Ideal Time:  299.8,  Delay time =   0.04  New State:ENGN_StateRunning_SamplingComplete
2021/01/22 16:53:50.7 Old State:18-ENGN_StateRunning_SamplingComplete New State:ENGN_StateRunning_WaitNewStep
2021/01/22 16:53:51.0 Old State:5-ENGN_StateRunning_WaitNewStep New State:ENGN_StateRunning_StartNewStep
2021/01/22 16:53:51.0 Spectrometer EOS_B0712: AVRG=20.
2021/01/22 16:53:51.0 Spectrometer EOS_B0734: AVRG=20.
2021/01/22 16:53:51.0 Spectrometer EOS_B0814: AVRG=20.
2021/01/22 16:53:51.0 Spectrometer EOS_B0735: AVRG=20.
2021/01/22 16:53:51.0 Spectrometer EOS_B0732: AVRG=20.
2021/01/22 16:53:51.0 Spectrometer EOS_B0808: AVRG=20.
2021/01/22 16:53:51.0 Spectrometer EOS_B0733: AVRG=20.
2021/01/22 16:53:51.0 Spectrometer EOS_B0817: AVRG=20.
2021/01/22 16:53:51.0 StartShaker(500)
2021/01/22 16:53:51.1 Old State:6-ENGN_StateRunning_StartNewStep New State:ENGN_StateRunning_WaitGoingToSample
2021/01/22 16:53:51.1 Sample96_Q(0, 0, 11)
2021/01/22 16:53:51.1 
2021/01/22 16:53:51.1 START SCAN: 
2021/01/22 16:53:51.1    CycleTime (ms)    = 200
2021/01/22 16:53:51.1    CycleCount        = 25
2021/01/22 16:53:51.1    ScansToAverage    = 20
2021/01/22 16:53:51.1    FinishTime (s)    = 5
2021/01/22 16:53:51.1    Temperature       = 25.0
2021/01/22 16:53:51.1    Lamp Power        = 31
2021/01/22 16:53:51.1    Spec PCB temps    = 42.3, 41.9, 42.7, 42.5, 42.3, 42.0, 40.9, 41.2
2021/01/22 16:53:51.1 
2021/01/22 16:53:51.2 GMC_SetPosZ
2021/01/22 16:53:51.2 Old State:12-ENGN_StateRunning_WaitGoingToSample New State:ENGN_StateRunning_GoingToSample
2021/01/22 16:53:51.2 Temp: 2021/01/22 16:53:51, 25.0000, 25.6000
2021/01/22 16:53:55.0 GMC_ServoHere
2021/01/22 16:53:55.1 GMC_SetPosXY
2021/01/22 16:53:56.5 GMC_SetPosZ
2021/01/22 16:54:02.8 Old State:13-ENGN_StateRunning_GoingToSample New State:ENGN_StateRunning_GoingToSampleComplete
2021/01/22 16:54:02.8 Shaker current speed = 494 rpm, set speed = 500
2021/01/22 16:54:02.9 Old State:14-cycle=1, Elapsed Time:    0.0,  Ideal Time:    0.0,  Delay time =   0.00  New State:ENGN_StateRunning_WaitSampling
2021/01/22 16:54:02.9 Step started at: 16:53:51
2021/01/22 16:54:02.9 XYZ movement time  = 11603 ms + Scan start delay = 11616 ms
2021/01/22 16:54:03.0 Old State:15-ENGN_StateRunning_WaitSampling New State:ENGN_StateRunning_Sampling
2021/01/22 16:54:07.7 
2021/01/22 16:54:07.7 STOP SCAN: 
2021/01/22 16:54:07.7    CycleTime (ms)    = 200
2021/01/22 16:54:07.7    CurrentCycle      = 25
2021/01/22 16:54:07.7    RunTime (ms)      = 4959
2021/01/22 16:54:07.7    AverageCycle (ms) = 198.4
2021/01/22 16:54:07.7    Temperature       = 25.0
2021/01/22 16:54:07.7    DATA OVERRUN %    = 0.00
2021/01/22 16:54:07.7    Spec PCB temps    = 42.3, 42.0, 42.8, 42.5, 42.2, 42.0, 40.9, 41.2
2021/01/22 16:54:07.7 
2021/01/22 16:54:07.7 Old State:16-cycle=25, Elapsed Time:    5.0,  Ideal Time:    4.8,  Delay time =   0.04  New State:ENGN_StateRunning_SamplingComplete
2021/01/22 16:54:07.9 Old State:18-ENGN_StateRunning_SamplingComplete New State:ENGN_StateRunning_WaitNewStep
2021/01/22 16:54:08.1 Old State:5-ENGN_StateRunning_WaitNewStep New State:ENGN_StateRunning_StartNewStep
2021/01/22 16:54:08.1 Spectrometer EOS_B0712: AVRG=20.
2021/01/22 16:54:08.1 Spectrometer EOS_B0734: AVRG=20.
2021/01/22 16:54:08.1 Spectrometer EOS_B0814: AVRG=20.
2021/01/22 16:54:08.1 Spectrometer EOS_B0735: AVRG=20.
2021/01/22 16:54:08.1 Spectrometer EOS_B0732: AVRG=20.
2021/01/22 16:54:08.1 Spectrometer EOS_B0808: AVRG=20.
2021/01/22 16:54:08.1 Spectrometer EOS_B0733: AVRG=20.
2021/01/22 16:54:08.1 Spectrometer EOS_B0817: AVRG=20.
2021/01/22 16:54:08.1 StartShaker(500)
2021/01/22 16:54:08.3 Old State:6-ENGN_StateRunning_StartNewStep New State:ENGN_StateRunning_WaitGoingToSample
2021/01/22 16:54:08.3 Sample96_Q(0, 0, 10)
2021/01/22 16:54:08.3 
2021/01/22 16:54:08.3 START SCAN: 
2021/01/22 16:54:08.3    CycleTime (ms)    = 200
2021/01/22 16:54:08.3    CycleCount        = 25
2021/01/22 16:54:08.3    ScansToAverage    = 20
2021/01/22 16:54:08.3    FinishTime (s)    = 5
2021/01/22 16:54:08.3    Temperature       = 25.0
2021/01/22 16:54:08.3    Lamp Power        = 31
2021/01/22 16:54:08.3    Spec PCB temps    = 42.3, 42.0, 42.8, 42.5, 42.2, 42.0, 40.9, 41.2
2021/01/22 16:54:08.3 
2021/01/22 16:54:08.3 GMC_SetPosZ
2021/01/22 16:54:08.4 Old State:12-ENGN_StateRunning_WaitGoingToSample New State:ENGN_StateRunning_GoingToSample
2021/01/22 16:54:08.4 Temp: 2021/01/22 16:54:08, 25.0000, 25.7000
2021/01/22 16:54:14.4 GMC_ServoHere
2021/01/22 16:54:14.5 GMC_SetPosXY
2021/01/22 16:54:14.9 GMC_SetPosZ
2021/01/22 16:54:20.6 Shaker current speed = 494 rpm, set speed = 500
2021/01/22 16:54:20.7 Old State:13-ENGN_StateRunning_GoingToSample New State:ENGN_StateRunning_GoingToSampleComplete
2021/01/22 16:54:20.8 Old State:14-cycle=1, Elapsed Time:    0.0,  Ideal Time:    0.0,  Delay time =   0.00  New State:ENGN_StateRunning_WaitSampling
2021/01/22 16:54:20.8 Step started at: 16:54:08
2021/01/22 16:54:20.8 XYZ movement time  = 12382 ms + Scan start delay = 12357 ms
2021/01/22 16:54:20.9 Old State:15-ENGN_StateRunning_WaitSampling New State:ENGN_StateRunning_Sampling
2021/01/22 16:54:25.6 
2021/01/22 16:54:25.6 STOP SCAN: 
2021/01/22 16:54:25.6    CycleTime (ms)    = 200
2021/01/22 16:54:25.6    CurrentCycle      = 25
2021/01/22 16:54:25.6    RunTime (ms)      = 4957
2021/01/22 16:54:25.6    AverageCycle (ms) = 198.3
2021/01/22 16:54:25.6    Temperature       = 25.0
2021/01/22 16:54:25.6    DATA OVERRUN %    = 0.00
2021/01/22 16:54:25.6    Spec PCB temps    = 42.3, 42.0, 42.7, 42.5, 42.2, 42.0, 40.9, 41.3
2021/01/22 16:54:25.6 
2021/01/22 16:54:25.7 Old State:16-cycle=25, Elapsed Time:    5.0,  Ideal Time:    4.8,  Delay time =   0.04  New State:ENGN_StateRunning_SamplingComplete
2021/01/22 16:54:25.8 Old State:18-ENGN_StateRunning_SamplingComplete New State:ENGN_StateRunning_WaitNewStep
2021/01/22 16:54:26.0 Old State:5-ENGN_StateRunning_WaitNewStep New State:ENGN_StateRunning_StartNewStep
2021/01/22 16:54:26.1 Spectrometer EOS_B0712: AVRG=20.
2021/01/22 16:54:26.1 Spectrometer EOS_B0734: AVRG=20.
2021/01/22 16:54:26.1 Spectrometer EOS_B0814: AVRG=20.
2021/01/22 16:54:26.1 Spectrometer EOS_B0735: AVRG=20.
2021/01/22 16:54:26.1 Spectrometer EOS_B0732: AVRG=20.
2021/01/22 16:54:26.1 Spectrometer EOS_B0808: AVRG=20.
2021/01/22 16:54:26.1 Spectrometer EOS_B0733: AVRG=20.
2021/01/22 16:54:26.1 Spectrometer EOS_B0817: AVRG=20.
2021/01/22 16:54:26.1 StartShaker(500)
2021/01/22 16:54:26.2 Old State:6-ENGN_StateRunning_StartNewStep New State:ENGN_StateRunning_WaitGoingToSample
2021/01/22 16:54:26.2 Sample96_Q(0, 0, 11)
2021/01/22 16:54:26.2 
2021/01/22 16:54:26.2 START SCAN: 
2021/01/22 16:54:26.2    CycleTime (ms)    = 200
2021/01/22 16:54:26.2    CycleCount        = 25
2021/01/22 16:54:26.2    ScansToAverage    = 20
2021/01/22 16:54:26.2    FinishTime (s)    = 5
2021/01/22 16:54:26.2    Temperature       = 25.0
2021/01/22 16:54:26.2    Lamp Power        = 31
2021/01/22 16:54:26.2    Spec PCB temps    = 42.3, 42.0, 42.7, 42.5, 42.2, 42.0, 40.9, 41.2
2021/01/22 16:54:26.2 
2021/01/22 16:54:26.2 GMC_SetPosZ
2021/01/22 16:54:26.3 Temp: 2021/01/22 16:54:26, 25.0000, 25.7000
2021/01/22 16:54:26.3 Old State:12-ENGN_StateRunning_WaitGoingToSample New State:ENGN_StateRunning_GoingToSample
2021/01/22 16:54:32.0 GMC_ServoHere
2021/01/22 16:54:32.1 GMC_SetPosXY
2021/01/22 16:54:32.5 GMC_SetPosZ
2021/01/22 16:54:38.6 Shaker current speed = 494 rpm, set speed = 500
2021/01/22 16:54:38.7 Old State:13-ENGN_StateRunning_GoingToSample New State:ENGN_StateRunning_GoingToSampleComplete
2021/01/22 16:54:38.8 Old State:14-cycle=1, Elapsed Time:    0.0,  Ideal Time:    0.0,  Delay time =   0.00  New State:ENGN_StateRunning_WaitSampling
2021/01/22 16:54:38.8 Step started at: 16:54:26
2021/01/22 16:54:38.8 XYZ movement time  = 12377 ms + Scan start delay = 12344 ms
2021/01/22 16:54:38.9 Old State:15-ENGN_StateRunning_WaitSampling New State:ENGN_StateRunning_Sampling
2021/01/22 16:54:43.5 
2021/01/22 16:54:43.5 STOP SCAN: 
2021/01/22 16:54:43.5    CycleTime (ms)    = 200
2021/01/22 16:54:43.5    CurrentCycle      = 25
2021/01/22 16:54:43.5    RunTime (ms)      = 4956
2021/01/22 16:54:43.5    AverageCycle (ms) = 198.2
2021/01/22 16:54:43.5    Temperature       = 25.0
2021/01/22 16:54:43.5    DATA OVERRUN %    = 0.00
2021/01/22 16:54:43.5    Spec PCB temps    = 42.3, 42.0, 42.8, 42.5, 42.2, 42.0, 40.9, 41.3
2021/01/22 16:54:43.5 
2021/01/22 16:54:43.6 Old State:16-cycle=25, Elapsed Time:    5.0,  Ideal Time:    4.8,  Delay time =   0.04  New State:ENGN_StateRunning_SamplingComplete
2021/01/22 16:54:43.7 Old State:18-ENGN_StateRunning_SamplingComplete New State:ENGN_StateRunning_WaitNewStep
2021/01/22 16:54:43.8 Old State:5-ENGN_StateRunning_WaitNewStep New State:ENGN_StateRunning_StartNewStep
2021/01/22 16:54:43.8 Spectrometer EOS_B0712: AVRG=20.
2021/01/22 16:54:43.8 Spectrometer EOS_B0734: AVRG=20.
2021/01/22 16:54:43.8 Spectrometer EOS_B0814: AVRG=20.
2021/01/22 16:54:43.8 Spectrometer EOS_B0735: AVRG=20.
2021/01/22 16:54:43.8 Spectrometer EOS_B0732: AVRG=20.
2021/01/22 16:54:43.8 Spectrometer EOS_B0808: AVRG=20.
2021/01/22 16:54:43.8 Spectrometer EOS_B0733: AVRG=20.
2021/01/22 16:54:43.8 Spectrometer EOS_B0817: AVRG=20.
2021/01/22 16:54:43.8 StartShaker(500)
2021/01/22 16:54:44.0 Old State:6-ENGN_StateRunning_StartNewStep New State:ENGN_StateRunning_WaitGoingToSample
2021/01/22 16:54:44.0 Sample96_Q(0, 0, 10)
2021/01/22 16:54:44.0 
2021/01/22 16:54:44.0 START SCAN: 
2021/01/22 16:54:44.0    CycleTime (ms)    = 200
2021/01/22 16:54:44.0    CycleCount        = 25
2021/01/22 16:54:44.0    ScansToAverage    = 20
2021/01/22 16:54:44.0    FinishTime (s)    = 5
2021/01/22 16:54:44.0    Temperature       = 25.0
2021/01/22 16:54:44.0    Lamp Power        = 31
2021/01/22 16:54:44.0    Spec PCB temps    = 42.3, 42.0, 42.8, 42.5, 42.2, 42.0, 40.9, 41.3
2021/01/22 16:54:44.0 
2021/01/22 16:54:44.1 Old State:12-ENGN_StateRunning_WaitGoingToSample New State:ENGN_StateRunning_GoingToSample
2021/01/22 16:54:44.1 GMC_SetPosZ
2021/01/22 16:54:44.2 Temp: 2021/01/22 16:54:44, 25.0000, 25.7000
2021/01/22 16:54:50.2 GMC_ServoHere
2021/01/22 16:54:50.4 GMC_SetPosXY
2021/01/22 16:54:50.7 GMC_SetPosZ
2021/01/22 16:54:56.5 Shaker current speed = 494 rpm, set speed = 500
2021/01/22 16:54:56.6 Old State:13-ENGN_StateRunning_GoingToSample New State:ENGN_StateRunning_GoingToSampleComplete
2021/01/22 16:54:56.7 Old State:14-cycle=1, Elapsed Time:    0.0,  Ideal Time:    0.0,  Delay time =   0.00  New State:ENGN_StateRunning_WaitSampling
2021/01/22 16:54:56.7 Step started at: 16:54:44
2021/01/22 16:54:56.7 XYZ movement time  = 12476 ms + Scan start delay = 12470 ms
2021/01/22 16:54:56.8 Old State:15-ENGN_StateRunning_WaitSampling New State:ENGN_StateRunning_Sampling
2021/01/22 16:55:01.4 
2021/01/22 16:55:01.4 STOP SCAN: 
2021/01/22 16:55:01.4    CycleTime (ms)    = 200
2021/01/22 16:55:01.4    CurrentCycle      = 25
2021/01/22 16:55:01.4    RunTime (ms)      = 4958
2021/01/22 16:55:01.4    AverageCycle (ms) = 198.3
2021/01/22 16:55:01.4    Temperature       = 25.0
2021/01/22 16:55:01.4    DATA OVERRUN %    = 0.00
2021/01/22 16:55:01.4    Spec PCB temps    = 42.3, 41.9, 42.8, 42.5, 42.1, 42.0, 40.9, 41.2
2021/01/22 16:55:01.4 
2021/01/22 16:55:01.5 Old State:16-cycle=25, Elapsed Time:    5.0,  Ideal Time:    4.8,  Delay time =   0.04  New State:ENGN_StateRunning_SamplingComplete
2021/01/22 16:55:01.6 Old State:18-ENGN_StateRunning_SamplingComplete New State:ENGN_StateRunning_WaitNewStep
2021/01/22 16:55:01.7 Old State:5-ENGN_StateRunning_WaitNewStep New State:ENGN_StateRunning_StartNewStep
2021/01/22 16:55:01.8 Spectrometer EOS_B0712: AVRG=20.
2021/01/22 16:55:01.8 Spectrometer EOS_B0734: AVRG=20.
2021/01/22 16:55:01.8 Spectrometer EOS_B0814: AVRG=20.
2021/01/22 16:55:01.8 Spectrometer EOS_B0735: AVRG=20.
2021/01/22 16:55:01.8 Spectrometer EOS_B0732: AVRG=20.
2021/01/22 16:55:01.8 Spectrometer EOS_B0808: AVRG=20.
2021/01/22 16:55:01.8 Spectrometer EOS_B0733: AVRG=20.
2021/01/22 16:55:01.8 Spectrometer EOS_B0817: AVRG=20.
2021/01/22 16:55:01.8 StartShaker(500)
2021/01/22 16:55:01.9 Old State:6-ENGN_StateRunning_StartNewStep New State:ENGN_StateRunning_WaitGoingToSample
2021/01/22 16:55:01.9 Sample96_Q(0, 0, 11)
2021/01/22 16:55:01.9 GMC_SetPosZ
2021/01/22 16:55:01.9 
2021/01/22 16:55:01.9 START SCAN: 
2021/01/22 16:55:01.9    CycleTime (ms)    = 200
2021/01/22 16:55:01.9    CycleCount        = 25
2021/01/22 16:55:01.9    ScansToAverage    = 20
2021/01/22 16:55:01.9    FinishTime (s)    = 5
2021/01/22 16:55:01.9    Temperature       = 25.0
2021/01/22 16:55:01.9    Lamp Power        = 31
2021/01/22 16:55:01.9    Spec PCB temps    = 42.3, 41.9, 42.8, 42.5, 42.1, 42.0, 40.9, 41.2
2021/01/22 16:55:01.9 
2021/01/22 16:55:02.0 Old State:12-ENGN_StateRunning_WaitGoingToSample New State:ENGN_StateRunning_GoingToSample
2021/01/22 16:55:02.0 Temp: 2021/01/22 16:55:02, 25.0000, 25.7000
2021/01/22 16:55:07.7 GMC_ServoHere
2021/01/22 16:55:07.8 GMC_SetPosXY
2021/01/22 16:55:08.1 GMC_SetPosZ
2021/01/22 16:55:14.3 Shaker current speed = 494 rpm, set speed = 500
2021/01/22 16:55:14.4 Old State:13-cycle=1, Elapsed Time:    0.0,  Ideal Time:    0.0,  Delay time =   0.00  New State:ENGN_StateRunning_GoingToSampleComplete
2021/01/22 16:55:14.5 Old State:14-ENGN_StateRunning_GoingToSampleComplete New State:ENGN_StateRunning_WaitSampling
2021/01/22 16:55:14.5 Step started at: 16:55:01
2021/01/22 16:55:14.5 XYZ movement time  = 12417 ms + Scan start delay = 12323 ms
2021/01/22 16:55:14.6 Old State:15-ENGN_StateRunning_WaitSampling New State:ENGN_StateRunning_Sampling
2021/01/22 16:55:19.2 
2021/01/22 16:55:19.2 STOP SCAN: 
2021/01/22 16:55:19.2    CycleTime (ms)    = 200
2021/01/22 16:55:19.2    CurrentCycle      = 25
2021/01/22 16:55:19.2    RunTime (ms)      = 4959
2021/01/22 16:55:19.2    AverageCycle (ms) = 198.4
2021/01/22 16:55:19.2    Temperature       = 25.0
2021/01/22 16:55:19.2    DATA OVERRUN %    = 0.00
2021/01/22 16:55:19.2    Spec PCB temps    = 42.3, 42.0, 42.8, 42.5, 42.2, 42.0, 40.9, 41.2
2021/01/22 16:55:19.2 
2021/01/22 16:55:19.2 Old State:16-cycle=25, Elapsed Time:    5.0,  Ideal Time:    4.8,  Delay time =   0.04  New State:ENGN_StateRunning_SamplingComplete
2021/01/22 16:55:19.4 Old State:18-ENGN_StateRunning_SamplingComplete New State:ENGN_StateRunning_WaitNewStep
2021/01/22 16:55:19.6 Old State:5-ENGN_StateRunning_WaitNewStep New State:ENGN_StateRunning_StartNewStep
2021/01/22 16:55:19.6 Spectrometer EOS_B0712: AVRG=20.
2021/01/22 16:55:19.6 Spectrometer EOS_B0734: AVRG=20.
2021/01/22 16:55:19.6 Spectrometer EOS_B0814: AVRG=20.
2021/01/22 16:55:19.6 Spectrometer EOS_B0735: AVRG=20.
2021/01/22 16:55:19.6 Spectrometer EOS_B0732: AVRG=20.
2021/01/22 16:55:19.6 Spectrometer EOS_B0808: AVRG=20.
2021/01/22 16:55:19.6 Spectrometer EOS_B0733: AVRG=20.
2021/01/22 16:55:19.6 Spectrometer EOS_B0817: AVRG=20.
2021/01/22 16:55:19.6 StartShaker(500)
2021/01/22 16:55:19.7 Old State:6-ENGN_StateRunning_StartNewStep New State:ENGN_StateRunning_WaitGoingToSample
2021/01/22 16:55:19.7 Sample96_Q(0, 0, 10)
2021/01/22 16:55:19.8 
2021/01/22 16:55:19.8 START SCAN: 
2021/01/22 16:55:19.8    CycleTime (ms)    = 200
2021/01/22 16:55:19.8    CycleCount        = 25
2021/01/22 16:55:19.8    ScansToAverage    = 20
2021/01/22 16:55:19.8    FinishTime (s)    = 5
2021/01/22 16:55:19.8    Temperature       = 25.0
2021/01/22 16:55:19.8    Lamp Power        = 31
2021/01/22 16:55:19.8    Spec PCB temps    = 42.3, 42.0, 42.8, 42.5, 42.2, 42.0, 40.9, 41.2
2021/01/22 16:55:19.8 
2021/01/22 16:55:19.8 GMC_SetPosZ
2021/01/22 16:55:19.9 Old State:12-ENGN_StateRunning_WaitGoingToSample New State:ENGN_StateRunning_GoingToSample
2021/01/22 16:55:19.9 Temp: 2021/01/22 16:55:19, 25.0000, 25.6000
2021/01/22 16:55:25.9 GMC_ServoHere
2021/01/22 16:55:26.0 GMC_SetPosXY
2021/01/22 16:55:26.4 GMC_SetPosZ
2021/01/22 16:55:32.1 Shaker current speed = 494 rpm, set speed = 500
2021/01/22 16:55:32.2 Old State:13-ENGN_StateRunning_GoingToSample New State:ENGN_StateRunning_GoingToSampleComplete
2021/01/22 16:55:32.3 Old State:14-cycle=1, Elapsed Time:    0.0,  Ideal Time:    0.0,  Delay time =   0.00  New State:ENGN_StateRunning_WaitSampling
2021/01/22 16:55:32.3 Step started at: 16:55:19
2021/01/22 16:55:32.3 XYZ movement time  = 12379 ms + Scan start delay = 12377 ms
2021/01/22 16:55:32.4 Old State:15-ENGN_StateRunning_WaitSampling New State:ENGN_StateRunning_Sampling
2021/01/22 16:55:37.1 
2021/01/22 16:55:37.1 STOP SCAN: 
2021/01/22 16:55:37.1    CycleTime (ms)    = 200
2021/01/22 16:55:37.1    CurrentCycle      = 25
2021/01/22 16:55:37.1    RunTime (ms)      = 4958
2021/01/22 16:55:37.1    AverageCycle (ms) = 198.3
2021/01/22 16:55:37.1    Temperature       = 25.0
2021/01/22 16:55:37.1    DATA OVERRUN %    = 0.00
2021/01/22 16:55:37.1    Spec PCB temps    = 42.3, 41.9, 42.8, 42.4, 42.2, 42.0, 40.9, 41.2
2021/01/22 16:55:37.1 
2021/01/22 16:55:37.2 Old State:16-cycle=25, Elapsed Time:    5.0,  Ideal Time:    4.8,  Delay time =   0.04  New State:ENGN_StateRunning_SamplingComplete
2021/01/22 16:55:37.3 Old State:18-ENGN_StateRunning_SamplingComplete New State:ENGN_StateRunning_WaitNewStep
2021/01/22 16:55:37.5 Old State:5-ENGN_StateRunning_WaitNewStep New State:ENGN_StateRunning_StartNewStep
2021/01/22 16:55:37.6 Spectrometer EOS_B0712: AVRG=20.
2021/01/22 16:55:37.6 Spectrometer EOS_B0734: AVRG=20.
2021/01/22 16:55:37.6 Spectrometer EOS_B0814: AVRG=20.
2021/01/22 16:55:37.6 Spectrometer EOS_B0735: AVRG=20.
2021/01/22 16:55:37.6 Spectrometer EOS_B0732: AVRG=20.
2021/01/22 16:55:37.6 Spectrometer EOS_B0808: AVRG=20.
2021/01/22 16:55:37.6 Spectrometer EOS_B0733: AVRG=20.
2021/01/22 16:55:37.6 Spectrometer EOS_B0817: AVRG=20.
2021/01/22 16:55:37.6 StartShaker(500)
2021/01/22 16:55:37.7 Old State:6-ENGN_StateRunning_StartNewStep New State:ENGN_StateRunning_WaitGoingToSample
2021/01/22 16:55:37.7 Sample96_Q(0, 0, 3)
2021/01/22 16:55:37.7 
2021/01/22 16:55:37.7 START SCAN: 
2021/01/22 16:55:37.7    CycleTime (ms)    = 200
2021/01/22 16:55:37.7    CycleCount        = 1500
2021/01/22 16:55:37.7    ScansToAverage    = 20
2021/01/22 16:55:37.7    FinishTime (s)    = 300
2021/01/22 16:55:37.7    Temperature       = 25.0
2021/01/22 16:55:37.7    Lamp Power        = 31
2021/01/22 16:55:37.7    Spec PCB temps    = 42.3, 41.9, 42.8, 42.4, 42.2, 42.0, 40.9, 41.2
2021/01/22 16:55:37.7 
2021/01/22 16:55:37.7 GMC_SetPosZ
2021/01/22 16:55:37.8 Old State:12-ENGN_StateRunning_WaitGoingToSample New State:ENGN_StateRunning_GoingToSample
2021/01/22 16:55:37.8 Temp: 2021/01/22 16:55:37, 25.0000, 25.7000
2021/01/22 16:55:43.5 GMC_ServoHere
2021/01/22 16:55:43.6 GMC_SetPosXY
2021/01/22 16:55:44.9 GMC_SetPosZ
2021/01/22 16:55:48.8 Shaker current speed = 494 rpm, set speed = 500
2021/01/22 16:55:48.9 Old State:13-ENGN_StateRunning_GoingToSample New State:ENGN_StateRunning_GoingToSampleComplete
2021/01/22 16:55:49.0 Old State:14-cycle=1, Elapsed Time:    0.0,  Ideal Time:    0.0,  Delay time =   0.00  New State:ENGN_StateRunning_WaitSampling
2021/01/22 16:55:49.0 Step started at: 16:55:37
2021/01/22 16:55:49.0 XYZ movement time  = 11177 ms + Scan start delay = 11140 ms
2021/01/22 16:55:49.1 Old State:15-ENGN_StateRunning_WaitSampling New State:ENGN_StateRunning_Sampling
2021/01/22 16:56:37.9 Temp: 2021/01/22 16:56:37, 25.0000, 25.6000
2021/01/22 16:57:38.1 Temp: 2021/01/22 16:57:38, 25.0000, 25.7000
2021/01/22 16:58:38.2 Temp: 2021/01/22 16:58:38, 25.0000, 25.7000
2021/01/22 16:59:38.4 Temp: 2021/01/22 16:59:38, 25.0000, 25.6000
2021/01/22 17:00:38.6 Temp: 2021/01/22 17:00:38, 25.0000, 25.6000
2021/01/22 17:00:48.8 
2021/01/22 17:00:48.8 STOP SCAN: 
2021/01/22 17:00:48.8    CycleTime (ms)    = 200
2021/01/22 17:00:48.8    CurrentCycle      = 1500
2021/01/22 17:00:48.8    RunTime (ms)      = 299958
2021/01/22 17:00:48.8    AverageCycle (ms) = 200.0
2021/01/22 17:00:48.8    Temperature       = 25.0
2021/01/22 17:00:48.8    DATA OVERRUN %    = 0.00
2021/01/22 17:00:48.8    Spec PCB temps    = 42.3, 41.9, 42.8, 42.5, 42.2, 42.0, 40.9, 41.3
2021/01/22 17:00:48.8 
2021/01/22 17:00:48.8 Old State:16-cycle=1500, Elapsed Time:  300.0,  Ideal Time:  299.8,  Delay time =   0.04  New State:ENGN_StateRunning_SamplingComplete
2021/01/22 17:00:48.9 Old State:18-ENGN_StateRunning_SamplingComplete New State:ENGN_StateRunning_WaitNewStep
2021/01/22 17:00:49.3 Old State:5-ENGN_StateRunning_WaitNewStep New State:ENGN_StateRunning_StartNewStep
2021/01/22 17:00:49.3 Spectrometer EOS_B0712: AVRG=20.
2021/01/22 17:00:49.3 Spectrometer EOS_B0734: AVRG=20.
2021/01/22 17:00:49.3 Spectrometer EOS_B0814: AVRG=20.
2021/01/22 17:00:49.3 Spectrometer EOS_B0735: AVRG=20.
2021/01/22 17:00:49.3 Spectrometer EOS_B0732: AVRG=20.
2021/01/22 17:00:49.3 Spectrometer EOS_B0808: AVRG=20.
2021/01/22 17:00:49.3 Spectrometer EOS_B0733: AVRG=20.
2021/01/22 17:00:49.3 Spectrometer EOS_B0817: AVRG=20.
2021/01/22 17:00:49.3 StartShaker(500)
2021/01/22 17:00:49.4 Old State:6-ENGN_StateRunning_StartNewStep New State:ENGN_StateRunning_WaitGoingToSample
2021/01/22 17:00:49.4 Sample96_Q(0, 0, 8)
2021/01/22 17:00:49.4 
2021/01/22 17:00:49.4 START SCAN: 
2021/01/22 17:00:49.4    CycleTime (ms)    = 200
2021/01/22 17:00:49.4    CycleCount        = 250
2021/01/22 17:00:49.4    ScansToAverage    = 20
2021/01/22 17:00:49.4    FinishTime (s)    = 50
2021/01/22 17:00:49.4    Temperature       = 25.0
2021/01/22 17:00:49.4    Lamp Power        = 31
2021/01/22 17:00:49.4    Spec PCB temps    = 42.3, 41.9, 42.8, 42.5, 42.3, 42.0, 40.9, 41.3
2021/01/22 17:00:49.4 
2021/01/22 17:00:49.5 GMC_SetPosZ
2021/01/22 17:00:49.5 Old State:12-ENGN_StateRunning_WaitGoingToSample New State:ENGN_StateRunning_GoingToSample
2021/01/22 17:00:49.6 Temp: 2021/01/22 17:00:49, 25.0000, 25.7000
2021/01/22 17:00:53.3 GMC_ServoHere
2021/01/22 17:00:53.4 GMC_SetPosXY
2021/01/22 17:00:54.4 GMC_SetPosZ
2021/01/22 17:00:59.8 Shaker current speed = 494 rpm, set speed = 500
2021/01/22 17:00:59.9 Old State:13-cycle=1, Elapsed Time:    0.0,  Ideal Time:    0.0,  Delay time =   0.00  New State:ENGN_StateRunning_GoingToSampleComplete
2021/01/22 17:01:00.0 Old State:14-ENGN_StateRunning_GoingToSampleComplete New State:ENGN_StateRunning_WaitSampling
2021/01/22 17:01:00.0 Step started at: 17:00:49
2021/01/22 17:01:00.0 XYZ movement time  = 10361 ms + Scan start delay = 10285 ms
2021/01/22 17:01:00.1 Old State:15-ENGN_StateRunning_WaitSampling New State:ENGN_StateRunning_Sampling
2021/01/22 17:01:49.7 
2021/01/22 17:01:49.7 STOP SCAN: 
2021/01/22 17:01:49.7    CycleTime (ms)    = 200
2021/01/22 17:01:49.7    CurrentCycle      = 250
2021/01/22 17:01:49.7    RunTime (ms)      = 49958
2021/01/22 17:01:49.7    AverageCycle (ms) = 199.8
2021/01/22 17:01:49.7    Temperature       = 25.0
2021/01/22 17:01:49.7    DATA OVERRUN %    = 0.00
2021/01/22 17:01:49.7    Spec PCB temps    = 42.3, 42.0, 42.8, 42.5, 42.2, 41.9, 40.9, 41.2
2021/01/22 17:01:49.7 
2021/01/22 17:01:49.7 Temp: 2021/01/22 17:01:49, 25.0000, 25.6000
2021/01/22 17:01:49.8 Old State:16-cycle=250, Elapsed Time:   50.0,  Ideal Time:   49.8,  Delay time =   0.04  New State:ENGN_StateRunning_SamplingComplete
2021/01/22 17:01:49.9 Old State:18-ENGN_StateRunning_SamplingComplete New State:ENGN_StateRunning_WaitNewStep
2021/01/22 17:01:50.1 Old State:5-ENGN_StateRunning_WaitNewStep New State:ENGN_StateRunning_StartNewStep
2021/01/22 17:01:50.1 Spectrometer EOS_B0712: AVRG=20.
2021/01/22 17:01:50.1 Spectrometer EOS_B0734: AVRG=20.
2021/01/22 17:01:50.1 Spectrometer EOS_B0814: AVRG=20.
2021/01/22 17:01:50.1 Spectrometer EOS_B0735: AVRG=20.
2021/01/22 17:01:50.1 Spectrometer EOS_B0732: AVRG=20.
2021/01/22 17:01:50.1 Spectrometer EOS_B0808: AVRG=20.
2021/01/22 17:01:50.1 Spectrometer EOS_B0733: AVRG=20.
2021/01/22 17:01:50.1 Spectrometer EOS_B0817: AVRG=20.
2021/01/22 17:01:50.1 StartShaker(500)
2021/01/22 17:01:50.3 Old State:6-ENGN_StateRunning_StartNewStep New State:ENGN_StateRunning_WaitGoingToSample
2021/01/22 17:01:50.3 Sample96_Q(0, 0, 3)
2021/01/22 17:01:50.3 
2021/01/22 17:01:50.3 START SCAN: 
2021/01/22 17:01:50.3    CycleTime (ms)    = 200
2021/01/22 17:01:50.3    CycleCount        = 1500
2021/01/22 17:01:50.3    ScansToAverage    = 20
2021/01/22 17:01:50.3    FinishTime (s)    = 300
2021/01/22 17:01:50.3    Temperature       = 25.0
2021/01/22 17:01:50.3    Lamp Power        = 31
2021/01/22 17:01:50.3    Spec PCB temps    = 42.3, 42.0, 42.8, 42.5, 42.2, 41.9, 40.9, 41.2
2021/01/22 17:01:50.3 
2021/01/22 17:01:50.3 GMC_SetPosZ
2021/01/22 17:01:50.4 Old State:12-ENGN_StateRunning_WaitGoingToSample New State:ENGN_StateRunning_GoingToSample
2021/01/22 17:01:50.4 Temp: 2021/01/22 17:01:50, 25.0000, 25.7000
2021/01/22 17:01:55.6 GMC_ServoHere
2021/01/22 17:01:55.8 GMC_SetPosXY
2021/01/22 17:01:56.7 GMC_SetPosZ
2021/01/22 17:02:00.7 Old State:13-ENGN_StateRunning_GoingToSample New State:ENGN_StateRunning_GoingToSampleComplete
2021/01/22 17:02:00.7 Shaker current speed = 494 rpm, set speed = 500
2021/01/22 17:02:00.8 Old State:14-cycle=1, Elapsed Time:    0.0,  Ideal Time:    0.0,  Delay time =   0.00  New State:ENGN_StateRunning_WaitSampling
2021/01/22 17:02:00.8 Step started at: 17:01:50
2021/01/22 17:02:00.8 XYZ movement time  = 10368 ms + Scan start delay = 10383 ms
2021/01/22 17:02:00.9 Old State:15-ENGN_StateRunning_WaitSampling New State:ENGN_StateRunning_Sampling
2021/01/22 17:02:50.6 Temp: 2021/01/22 17:02:50, 25.0000, 25.7000
2021/01/22 17:03:50.7 Temp: 2021/01/22 17:03:50, 25.0000, 25.6000
2021/01/22 17:04:50.9 Temp: 2021/01/22 17:04:50, 25.0000, 25.7000
2021/01/22 17:05:51.0 Temp: 2021/01/22 17:05:51, 25.0000, 25.6000
2021/01/22 17:06:51.2 Temp: 2021/01/22 17:06:51, 25.0000, 25.7000
2021/01/22 17:07:00.6 
2021/01/22 17:07:00.6 STOP SCAN: 
2021/01/22 17:07:00.6    CycleTime (ms)    = 200
2021/01/22 17:07:00.6    CurrentCycle      = 1500
2021/01/22 17:07:00.6    RunTime (ms)      = 299958
2021/01/22 17:07:00.6    AverageCycle (ms) = 200.0
2021/01/22 17:07:00.6    Temperature       = 25.0
2021/01/22 17:07:00.6    DATA OVERRUN %    = 0.00
2021/01/22 17:07:00.6    Spec PCB temps    = 42.3, 41.9, 42.8, 42.4, 42.2, 42.0, 40.9, 41.3
2021/01/22 17:07:00.6 
2021/01/22 17:07:00.7 Old State:16-cycle=1500, Elapsed Time:  300.0,  Ideal Time:  299.8,  Delay time =   0.04  New State:ENGN_StateRunning_SamplingComplete
2021/01/22 17:07:00.8 Old State:18-ENGN_StateRunning_SamplingComplete New State:ENGN_StateRunning_WaitNewStep
2021/01/22 17:07:01.3 Old State:5-ENGN_StateRunning_WaitNewStep New State:ENGN_StateRunning_StartNewStep
2021/01/22 17:07:01.3 Spectrometer EOS_B0712: AVRG=20.
2021/01/22 17:07:01.3 Spectrometer EOS_B0734: AVRG=20.
2021/01/22 17:07:01.3 Spectrometer EOS_B0814: AVRG=20.
2021/01/22 17:07:01.3 Spectrometer EOS_B0735: AVRG=20.
2021/01/22 17:07:01.3 Spectrometer EOS_B0732: AVRG=20.
2021/01/22 17:07:01.3 Spectrometer EOS_B0808: AVRG=20.
2021/01/22 17:07:01.3 Spectrometer EOS_B0733: AVRG=20.
2021/01/22 17:07:01.3 Spectrometer EOS_B0817: AVRG=20.
2021/01/22 17:07:01.3 StartShaker(500)
2021/01/22 17:07:01.4 Old State:6-ENGN_StateRunning_StartNewStep New State:ENGN_StateRunning_WaitGoingToSample
2021/01/22 17:07:01.4 Sample96_Q(0, 0, 11)
2021/01/22 17:07:01.4 
2021/01/22 17:07:01.4 START SCAN: 
2021/01/22 17:07:01.4    CycleTime (ms)    = 200
2021/01/22 17:07:01.4    CycleCount        = 25
2021/01/22 17:07:01.4    ScansToAverage    = 20
2021/01/22 17:07:01.4    FinishTime (s)    = 5
2021/01/22 17:07:01.4    Temperature       = 25.0
2021/01/22 17:07:01.4    Lamp Power        = 31
2021/01/22 17:07:01.4    Spec PCB temps    = 42.3, 41.9, 42.8, 42.4, 42.2, 42.0, 40.9, 41.2
2021/01/22 17:07:01.4 
2021/01/22 17:07:01.5 GMC_SetPosZ
2021/01/22 17:07:01.5 Old State:12-ENGN_StateRunning_WaitGoingToSample New State:ENGN_StateRunning_GoingToSample
2021/01/22 17:07:01.5 Temp: 2021/01/22 17:07:01, 25.0000, 25.6000
2021/01/22 17:07:05.3 GMC_ServoHere
2021/01/22 17:07:05.4 GMC_SetPosXY
2021/01/22 17:07:06.8 GMC_SetPosZ
2021/01/22 17:07:13.1 Shaker current speed = 494 rpm, set speed = 500
2021/01/22 17:07:13.2 Old State:13-cycle=1, Elapsed Time:    0.0,  Ideal Time:    0.0,  Delay time =   0.00  New State:ENGN_StateRunning_GoingToSampleComplete
2021/01/22 17:07:13.3 Old State:14-ENGN_StateRunning_GoingToSampleComplete New State:ENGN_StateRunning_WaitSampling
2021/01/22 17:07:13.3 Step started at: 17:07:01
2021/01/22 17:07:13.3 XYZ movement time  = 11679 ms + Scan start delay = 11619 ms
2021/01/22 17:07:13.4 Old State:15-ENGN_StateRunning_WaitSampling New State:ENGN_StateRunning_Sampling
2021/01/22 17:07:18.0 
2021/01/22 17:07:18.0 STOP SCAN: 
2021/01/22 17:07:18.0    CycleTime (ms)    = 200
2021/01/22 17:07:18.0    CurrentCycle      = 25
2021/01/22 17:07:18.0    RunTime (ms)      = 4959
2021/01/22 17:07:18.0    AverageCycle (ms) = 198.4
2021/01/22 17:07:18.0    Temperature       = 25.0
2021/01/22 17:07:18.0    DATA OVERRUN %    = 0.00
2021/01/22 17:07:18.0    Spec PCB temps    = 42.3, 41.9, 42.8, 42.4, 42.2, 42.0, 40.9, 41.3
2021/01/22 17:07:18.0 
2021/01/22 17:07:18.0 Old State:16-cycle=25, Elapsed Time:    5.0,  Ideal Time:    4.8,  Delay time =   0.04  New State:ENGN_StateRunning_SamplingComplete
2021/01/22 17:07:18.2 Old State:18-ENGN_StateRunning_SamplingComplete New State:ENGN_StateRunning_WaitNewStep
2021/01/22 17:07:18.4 Old State:5-ENGN_StateRunning_WaitNewStep New State:ENGN_StateRunning_StartNewStep
2021/01/22 17:07:18.4 Spectrometer EOS_B0712: AVRG=20.
2021/01/22 17:07:18.4 Spectrometer EOS_B0734: AVRG=20.
2021/01/22 17:07:18.4 Spectrometer EOS_B0814: AVRG=20.
2021/01/22 17:07:18.4 Spectrometer EOS_B0735: AVRG=20.
2021/01/22 17:07:18.4 Spectrometer EOS_B0732: AVRG=20.
2021/01/22 17:07:18.4 Spectrometer EOS_B0808: AVRG=20.
2021/01/22 17:07:18.4 Spectrometer EOS_B0733: AVRG=20.
2021/01/22 17:07:18.4 Spectrometer EOS_B0817: AVRG=20.
2021/01/22 17:07:18.4 StartShaker(500)
2021/01/22 17:07:18.5 Old State:6-ENGN_StateRunning_StartNewStep New State:ENGN_StateRunning_WaitGoingToSample
2021/01/22 17:07:18.5 Sample96_Q(0, 0, 10)
2021/01/22 17:07:18.5 
2021/01/22 17:07:18.5 START SCAN: 
2021/01/22 17:07:18.5    CycleTime (ms)    = 200
2021/01/22 17:07:18.5    CycleCount        = 25
2021/01/22 17:07:18.5    ScansToAverage    = 20
2021/01/22 17:07:18.5    FinishTime (s)    = 5
2021/01/22 17:07:18.5    Temperature       = 25.0
2021/01/22 17:07:18.5    Lamp Power        = 31
2021/01/22 17:07:18.5    Spec PCB temps    = 42.3, 41.9, 42.8, 42.4, 42.2, 42.0, 40.9, 41.3
2021/01/22 17:07:18.5 
2021/01/22 17:07:18.6 GMC_SetPosZ
2021/01/22 17:07:18.6 Old State:12-ENGN_StateRunning_WaitGoingToSample New State:ENGN_StateRunning_GoingToSample
2021/01/22 17:07:18.7 Temp: 2021/01/22 17:07:18, 25.0000, 25.6000
2021/01/22 17:07:24.7 GMC_ServoHere
2021/01/22 17:07:24.8 GMC_SetPosXY
2021/01/22 17:07:25.1 GMC_SetPosZ
2021/01/22 17:07:31.1 Shaker current speed = 494 rpm, set speed = 500
2021/01/22 17:07:31.2 Old State:13-ENGN_StateRunning_GoingToSample New State:ENGN_StateRunning_GoingToSampleComplete
2021/01/22 17:07:31.3 Old State:14-cycle=1, Elapsed Time:    0.0,  Ideal Time:    0.0,  Delay time =   0.00  New State:ENGN_StateRunning_WaitSampling
2021/01/22 17:07:31.3 Step started at: 17:07:18
2021/01/22 17:07:31.3 XYZ movement time  = 12584 ms + Scan start delay = 12546 ms
2021/01/22 17:07:31.4 Old State:15-ENGN_StateRunning_WaitSampling New State:ENGN_StateRunning_Sampling
2021/01/22 17:07:36.0 
2021/01/22 17:07:36.0 STOP SCAN: 
2021/01/22 17:07:36.0    CycleTime (ms)    = 200
2021/01/22 17:07:36.0    CurrentCycle      = 25
2021/01/22 17:07:36.0    RunTime (ms)      = 4959
2021/01/22 17:07:36.0    AverageCycle (ms) = 198.4
2021/01/22 17:07:36.0    Temperature       = 25.0
2021/01/22 17:07:36.0    DATA OVERRUN %    = 0.00
2021/01/22 17:07:36.0    Spec PCB temps    = 42.3, 41.9, 42.8, 42.5, 42.3, 42.0, 40.9, 41.3
2021/01/22 17:07:36.0 
2021/01/22 17:07:36.0 Old State:16-cycle=25, Elapsed Time:    5.0,  Ideal Time:    4.8,  Delay time =   0.04  New State:ENGN_StateRunning_SamplingComplete
2021/01/22 17:07:36.2 Old State:18-ENGN_StateRunning_SamplingComplete New State:ENGN_StateRunning_WaitNewStep
2021/01/22 17:07:36.3 Old State:5-ENGN_StateRunning_WaitNewStep New State:ENGN_StateRunning_StartNewStep
2021/01/22 17:07:36.3 Spectrometer EOS_B0712: AVRG=20.
2021/01/22 17:07:36.3 Spectrometer EOS_B0734: AVRG=20.
2021/01/22 17:07:36.3 Spectrometer EOS_B0814: AVRG=20.
2021/01/22 17:07:36.3 Spectrometer EOS_B0735: AVRG=20.
2021/01/22 17:07:36.3 Spectrometer EOS_B0732: AVRG=20.
2021/01/22 17:07:36.3 Spectrometer EOS_B0808: AVRG=20.
2021/01/22 17:07:36.3 Spectrometer EOS_B0733: AVRG=20.
2021/01/22 17:07:36.3 Spectrometer EOS_B0817: AVRG=20.
2021/01/22 17:07:36.3 StartShaker(500)
2021/01/22 17:07:36.4 Old State:6-ENGN_StateRunning_StartNewStep New State:ENGN_StateRunning_WaitGoingToSample
2021/01/22 17:07:36.4 Sample96_Q(0, 0, 11)
2021/01/22 17:07:36.4 
2021/01/22 17:07:36.4 START SCAN: 
2021/01/22 17:07:36.4    CycleTime (ms)    = 200
2021/01/22 17:07:36.4    CycleCount        = 25
2021/01/22 17:07:36.4    ScansToAverage    = 20
2021/01/22 17:07:36.4    FinishTime (s)    = 5
2021/01/22 17:07:36.4    Temperature       = 25.0
2021/01/22 17:07:36.4    Lamp Power        = 31
2021/01/22 17:07:36.4    Spec PCB temps    = 42.3, 41.9, 42.8, 42.5, 42.3, 42.0, 40.9, 41.3
2021/01/22 17:07:36.4 
2021/01/22 17:07:36.5 GMC_SetPosZ
2021/01/22 17:07:36.5 Old State:12-ENGN_StateRunning_WaitGoingToSample New State:ENGN_StateRunning_GoingToSample
2021/01/22 17:07:36.6 Temp: 2021/01/22 17:07:36, 25.0000, 25.7000
2021/01/22 17:07:42.2 GMC_ServoHere
2021/01/22 17:07:42.4 GMC_SetPosXY
2021/01/22 17:07:42.7 GMC_SetPosZ
2021/01/22 17:07:49.0 Old State:13-ENGN_StateRunning_GoingToSample New State:ENGN_StateRunning_GoingToSampleComplete
2021/01/22 17:07:49.0 Shaker current speed = 494 rpm, set speed = 500
2021/01/22 17:07:49.1 Old State:14-cycle=1, Elapsed Time:    0.0,  Ideal Time:    0.0,  Delay time =   0.00  New State:ENGN_StateRunning_WaitSampling
2021/01/22 17:07:49.1 Step started at: 17:07:36
2021/01/22 17:07:49.1 XYZ movement time  = 12486 ms + Scan start delay = 12493 ms
2021/01/22 17:07:49.2 Old State:15-ENGN_StateRunning_WaitSampling New State:ENGN_StateRunning_Sampling
2021/01/22 17:07:53.9 
2021/01/22 17:07:53.9 STOP SCAN: 
2021/01/22 17:07:53.9    CycleTime (ms)    = 200
2021/01/22 17:07:53.9    CurrentCycle      = 25
2021/01/22 17:07:53.9    RunTime (ms)      = 4959
2021/01/22 17:07:53.9    AverageCycle (ms) = 198.4
2021/01/22 17:07:53.9    Temperature       = 25.0
2021/01/22 17:07:53.9    DATA OVERRUN %    = 0.00
2021/01/22 17:07:53.9    Spec PCB temps    = 42.3, 42.0, 42.8, 42.5, 42.2, 42.0, 40.9, 41.2
2021/01/22 17:07:53.9 
2021/01/22 17:07:54.0 Old State:16-cycle=25, Elapsed Time:    5.0,  Ideal Time:    4.8,  Delay time =   0.04  New State:ENGN_StateRunning_SamplingComplete
2021/01/22 17:07:54.1 Old State:18-ENGN_StateRunning_SamplingComplete New State:ENGN_StateRunning_WaitNewStep
2021/01/22 17:07:54.5 Old State:5-ENGN_StateRunning_WaitNewStep New State:ENGN_StateRunning_StartNewStep
2021/01/22 17:07:54.6 Spectrometer EOS_B0712: AVRG=20.
2021/01/22 17:07:54.6 Spectrometer EOS_B0734: AVRG=20.
2021/01/22 17:07:54.6 Spectrometer EOS_B0814: AVRG=20.
2021/01/22 17:07:54.6 Spectrometer EOS_B0735: AVRG=20.
2021/01/22 17:07:54.6 Spectrometer EOS_B0732: AVRG=20.
2021/01/22 17:07:54.6 Spectrometer EOS_B0808: AVRG=20.
2021/01/22 17:07:54.6 Spectrometer EOS_B0733: AVRG=20.
2021/01/22 17:07:54.6 Spectrometer EOS_B0817: AVRG=20.
2021/01/22 17:07:54.6 StartShaker(500)
2021/01/22 17:07:54.7 Old State:6-ENGN_StateRunning_StartNewStep New State:ENGN_StateRunning_WaitGoingToSample
2021/01/22 17:07:54.7 Sample96_Q(0, 0, 10)
2021/01/22 17:07:54.7 
2021/01/22 17:07:54.7 START SCAN: 
2021/01/22 17:07:54.7    CycleTime (ms)    = 200
2021/01/22 17:07:54.7    CycleCount        = 25
2021/01/22 17:07:54.7    ScansToAverage    = 20
2021/01/22 17:07:54.7    FinishTime (s)    = 5
2021/01/22 17:07:54.7    Temperature       = 25.0
2021/01/22 17:07:54.7    Lamp Power        = 31
2021/01/22 17:07:54.7    Spec PCB temps    = 42.3, 42.0, 42.8, 42.4, 42.2, 42.0, 40.9, 41.2
2021/01/22 17:07:54.7 
2021/01/22 17:07:54.8 Temp: 2021/01/22 17:07:54, 25.0000, 25.6000
2021/01/22 17:07:54.8 Old State:12-ENGN_StateRunning_WaitGoingToSample New State:ENGN_StateRunning_GoingToSample
2021/01/22 17:07:54.9 GMC_SetPosZ
2021/01/22 17:08:00.9 GMC_ServoHere
2021/01/22 17:08:01.1 GMC_SetPosXY
2021/01/22 17:08:01.4 GMC_SetPosZ
2021/01/22 17:08:07.2 Shaker current speed = 494 rpm, set speed = 500
2021/01/22 17:08:07.3 Old State:13-ENGN_StateRunning_GoingToSample New State:ENGN_StateRunning_GoingToSampleComplete
2021/01/22 17:08:07.4 Old State:14-cycle=1, Elapsed Time:    0.0,  Ideal Time:    0.0,  Delay time =   0.00  New State:ENGN_StateRunning_WaitSampling
2021/01/22 17:08:07.4 Step started at: 17:07:54
2021/01/22 17:08:07.4 XYZ movement time  = 12487 ms + Scan start delay = 12469 ms
2021/01/22 17:08:07.5 Old State:15-ENGN_StateRunning_WaitSampling New State:ENGN_StateRunning_Sampling
2021/01/22 17:08:12.1 
2021/01/22 17:08:12.1 STOP SCAN: 
2021/01/22 17:08:12.1    CycleTime (ms)    = 200
2021/01/22 17:08:12.1    CurrentCycle      = 25
2021/01/22 17:08:12.1    RunTime (ms)      = 4958
2021/01/22 17:08:12.1    AverageCycle (ms) = 198.3
2021/01/22 17:08:12.1    Temperature       = 25.0
2021/01/22 17:08:12.1    DATA OVERRUN %    = 0.00
2021/01/22 17:08:12.1    Spec PCB temps    = 42.3, 41.9, 42.8, 42.4, 42.3, 42.0, 40.9, 41.2
2021/01/22 17:08:12.1 
2021/01/22 17:08:12.2 Old State:16-cycle=25, Elapsed Time:    5.0,  Ideal Time:    4.8,  Delay time =   0.04  New State:ENGN_StateRunning_SamplingComplete
2021/01/22 17:08:12.4 Old State:18-ENGN_StateRunning_SamplingComplete New State:ENGN_StateRunning_WaitNewStep
2021/01/22 17:08:12.7 Old State:5-ENGN_StateRunning_WaitNewStep New State:ENGN_StateRunning_StartNewStep
2021/01/22 17:08:12.7 Spectrometer EOS_B0712: AVRG=20.
2021/01/22 17:08:12.7 Spectrometer EOS_B0734: AVRG=20.
2021/01/22 17:08:12.7 Spectrometer EOS_B0814: AVRG=20.
2021/01/22 17:08:12.7 Spectrometer EOS_B0735: AVRG=20.
2021/01/22 17:08:12.7 Spectrometer EOS_B0732: AVRG=20.
2021/01/22 17:08:12.7 Spectrometer EOS_B0808: AVRG=20.
2021/01/22 17:08:12.7 Spectrometer EOS_B0733: AVRG=20.
2021/01/22 17:08:12.7 Spectrometer EOS_B0817: AVRG=20.
2021/01/22 17:08:12.7 StartShaker(500)
2021/01/22 17:08:12.8 Old State:6-ENGN_StateRunning_StartNewStep New State:ENGN_StateRunning_WaitGoingToSample
2021/01/22 17:08:12.8 Sample96_Q(0, 0, 11)
2021/01/22 17:08:12.9 
2021/01/22 17:08:12.9 START SCAN: 
2021/01/22 17:08:12.9    CycleTime (ms)    = 200
2021/01/22 17:08:12.9    CycleCount        = 25
2021/01/22 17:08:12.9    ScansToAverage    = 20
2021/01/22 17:08:12.9    FinishTime (s)    = 5
2021/01/22 17:08:12.9    Temperature       = 25.0
2021/01/22 17:08:12.9    Lamp Power        = 31
2021/01/22 17:08:12.9    Spec PCB temps    = 42.3, 41.9, 42.8, 42.4, 42.3, 42.0, 40.9, 41.2
2021/01/22 17:08:12.9 
2021/01/22 17:08:12.9 GMC_SetPosZ
2021/01/22 17:08:13.0 Old State:12-ENGN_StateRunning_WaitGoingToSample New State:ENGN_StateRunning_GoingToSample
2021/01/22 17:08:13.0 Temp: 2021/01/22 17:08:13, 25.0000, 25.6000
2021/01/22 17:08:18.7 GMC_ServoHere
2021/01/22 17:08:18.9 GMC_SetPosXY
2021/01/22 17:08:19.2 GMC_SetPosZ
2021/01/22 17:08:25.3 Old State:13-ENGN_StateRunning_GoingToSample New State:ENGN_StateRunning_GoingToSampleComplete
2021/01/22 17:08:25.3 Shaker current speed = 494 rpm, set speed = 500
2021/01/22 17:08:25.4 Old State:14-cycle=1, Elapsed Time:    0.0,  Ideal Time:    0.0,  Delay time =   0.00  New State:ENGN_StateRunning_WaitSampling
2021/01/22 17:08:25.4 Step started at: 17:08:12
2021/01/22 17:08:25.4 XYZ movement time  = 12387 ms + Scan start delay = 12400 ms
2021/01/22 17:08:25.5 Old State:15-ENGN_StateRunning_WaitSampling New State:ENGN_StateRunning_Sampling
2021/01/22 17:08:30.2 
2021/01/22 17:08:30.2 STOP SCAN: 
2021/01/22 17:08:30.2    CycleTime (ms)    = 200
2021/01/22 17:08:30.2    CurrentCycle      = 25
2021/01/22 17:08:30.2    RunTime (ms)      = 4958
2021/01/22 17:08:30.2    AverageCycle (ms) = 198.3
2021/01/22 17:08:30.2    Temperature       = 25.0
2021/01/22 17:08:30.2    DATA OVERRUN %    = 0.00
2021/01/22 17:08:30.2    Spec PCB temps    = 42.3, 42.0, 42.8, 42.4, 42.2, 42.0, 40.9, 41.3
2021/01/22 17:08:30.2 
2021/01/22 17:08:30.3 Old State:16-cycle=25, Elapsed Time:    5.0,  Ideal Time:    4.8,  Delay time =   0.04  New State:ENGN_StateRunning_SamplingComplete
2021/01/22 17:08:30.4 Old State:18-ENGN_StateRunning_SamplingComplete New State:ENGN_StateRunning_WaitNewStep
2021/01/22 17:08:30.6 Old State:5-ENGN_StateRunning_WaitNewStep New State:ENGN_StateRunning_StartNewStep
2021/01/22 17:08:30.6 Spectrometer EOS_B0712: AVRG=20.
2021/01/22 17:08:30.6 Spectrometer EOS_B0734: AVRG=20.
2021/01/22 17:08:30.6 Spectrometer EOS_B0814: AVRG=20.
2021/01/22 17:08:30.6 Spectrometer EOS_B0735: AVRG=20.
2021/01/22 17:08:30.6 Spectrometer EOS_B0732: AVRG=20.
2021/01/22 17:08:30.6 Spectrometer EOS_B0808: AVRG=20.
2021/01/22 17:08:30.6 Spectrometer EOS_B0733: AVRG=20.
2021/01/22 17:08:30.6 Spectrometer EOS_B0817: AVRG=20.
2021/01/22 17:08:30.6 StartShaker(500)
2021/01/22 17:08:30.8 Old State:6-ENGN_StateRunning_StartNewStep New State:ENGN_StateRunning_WaitGoingToSample
2021/01/22 17:08:30.8 Sample96_Q(0, 0, 10)
2021/01/22 17:08:30.8 
2021/01/22 17:08:30.8 START SCAN: 
2021/01/22 17:08:30.8    CycleTime (ms)    = 200
2021/01/22 17:08:30.8    CycleCount        = 25
2021/01/22 17:08:30.8    ScansToAverage    = 20
2021/01/22 17:08:30.8    FinishTime (s)    = 5
2021/01/22 17:08:30.8    Temperature       = 25.0
2021/01/22 17:08:30.8    Lamp Power        = 31
2021/01/22 17:08:30.8    Spec PCB temps    = 42.3, 42.0, 42.8, 42.4, 42.2, 42.0, 40.9, 41.3
2021/01/22 17:08:30.8 
2021/01/22 17:08:30.8 GMC_SetPosZ
2021/01/22 17:08:30.9 Old State:12-ENGN_StateRunning_WaitGoingToSample New State:ENGN_StateRunning_GoingToSample
2021/01/22 17:08:30.9 Temp: 2021/01/22 17:08:30, 25.0000, 25.6000
2021/01/22 17:08:36.9 GMC_ServoHere
2021/01/22 17:08:37.1 GMC_SetPosXY
2021/01/22 17:08:37.4 GMC_SetPosZ
2021/01/22 17:08:43.4 Shaker current speed = 494 rpm, set speed = 500
2021/01/22 17:08:43.5 Old State:13-ENGN_StateRunning_GoingToSample New State:ENGN_StateRunning_GoingToSampleComplete
2021/01/22 17:08:43.6 Old State:14-cycle=1, Elapsed Time:    0.0,  Ideal Time:    0.0,  Delay time =   0.00  New State:ENGN_StateRunning_WaitSampling
2021/01/22 17:08:43.6 Step started at: 17:08:30
2021/01/22 17:08:43.6 XYZ movement time  = 12588 ms + Scan start delay = 12533 ms
2021/01/22 17:08:43.7 Old State:15-ENGN_StateRunning_WaitSampling New State:ENGN_StateRunning_Sampling
2021/01/22 17:08:48.3 
2021/01/22 17:08:48.3 STOP SCAN: 
2021/01/22 17:08:48.3    CycleTime (ms)    = 200
2021/01/22 17:08:48.3    CurrentCycle      = 25
2021/01/22 17:08:48.3    RunTime (ms)      = 4957
2021/01/22 17:08:48.3    AverageCycle (ms) = 198.3
2021/01/22 17:08:48.3    Temperature       = 25.0
2021/01/22 17:08:48.3    DATA OVERRUN %    = 0.00
2021/01/22 17:08:48.3    Spec PCB temps    = 42.3, 41.9, 42.7, 42.4, 42.2, 42.0, 40.9, 41.3
2021/01/22 17:08:48.3 
2021/01/22 17:08:48.3 Old State:16-cycle=25, Elapsed Time:    5.0,  Ideal Time:    4.8,  Delay time =   0.04  New State:ENGN_StateRunning_SamplingComplete
2021/01/22 17:08:48.4 Old State:18-ENGN_StateRunning_SamplingComplete New State:ENGN_StateRunning_WaitNewStep
2021/01/22 17:08:48.8 Old State:5-ENGN_StateRunning_WaitNewStep New State:ENGN_StateRunning_StartNewStep
2021/01/22 17:08:48.8 Spectrometer EOS_B0712: AVRG=20.
2021/01/22 17:08:48.8 Spectrometer EOS_B0734: AVRG=20.
2021/01/22 17:08:48.8 Spectrometer EOS_B0814: AVRG=20.
2021/01/22 17:08:48.8 Spectrometer EOS_B0735: AVRG=20.
2021/01/22 17:08:48.8 Spectrometer EOS_B0732: AVRG=20.
2021/01/22 17:08:48.8 Spectrometer EOS_B0808: AVRG=20.
2021/01/22 17:08:48.8 Spectrometer EOS_B0733: AVRG=20.
2021/01/22 17:08:48.8 Spectrometer EOS_B0817: AVRG=20.
2021/01/22 17:08:48.8 StartShaker(500)
2021/01/22 17:08:48.9 Old State:6-ENGN_StateRunning_StartNewStep New State:ENGN_StateRunning_WaitGoingToSample
2021/01/22 17:08:48.9 Sample96_Q(0, 0, 3)
2021/01/22 17:08:48.9 
2021/01/22 17:08:48.9 START SCAN: 
2021/01/22 17:08:48.9    CycleTime (ms)    = 200
2021/01/22 17:08:48.9    CycleCount        = 1500
2021/01/22 17:08:48.9    ScansToAverage    = 20
2021/01/22 17:08:48.9    FinishTime (s)    = 300
2021/01/22 17:08:48.9    Temperature       = 25.0
2021/01/22 17:08:48.9    Lamp Power        = 31
2021/01/22 17:08:48.9    Spec PCB temps    = 42.3, 41.9, 42.7, 42.4, 42.2, 42.0, 40.9, 41.3
2021/01/22 17:08:48.9 
2021/01/22 17:08:49.0 Old State:12-ENGN_StateRunning_WaitGoingToSample New State:ENGN_StateRunning_GoingToSample
2021/01/22 17:08:49.0 GMC_SetPosZ
2021/01/22 17:08:49.1 Temp: 2021/01/22 17:08:49, 25.0000, 25.6000
2021/01/22 17:08:54.8 GMC_ServoHere
2021/01/22 17:08:55.0 GMC_SetPosXY
2021/01/22 17:08:56.2 GMC_SetPosZ
2021/01/22 17:09:00.2 Shaker current speed = 494 rpm, set speed = 500
2021/01/22 17:09:00.3 Old State:13-ENGN_StateRunning_GoingToSample New State:ENGN_StateRunning_GoingToSampleComplete
2021/01/22 17:09:00.4 Old State:14-cycle=1, Elapsed Time:    0.0,  Ideal Time:    0.0,  Delay time =   0.00  New State:ENGN_StateRunning_WaitSampling
2021/01/22 17:09:00.4 Step started at: 17:08:48
2021/01/22 17:09:00.4 XYZ movement time  = 11274 ms + Scan start delay = 11228 ms
2021/01/22 17:09:00.5 Old State:15-ENGN_StateRunning_WaitSampling New State:ENGN_StateRunning_Sampling
2021/01/22 17:09:49.3 Temp: 2021/01/22 17:09:49, 25.0000, 25.6000
2021/01/22 17:10:49.4 Temp: 2021/01/22 17:10:49, 25.0000, 25.7000
2021/01/22 17:11:49.6 Temp: 2021/01/22 17:11:49, 25.0000, 25.7000
2021/01/22 17:12:49.7 Temp: 2021/01/22 17:12:49, 25.0000, 25.7000
2021/01/22 17:13:49.9 Temp: 2021/01/22 17:13:49, 25.0000, 25.7000
2021/01/22 17:14:00.1 
2021/01/22 17:14:00.1 STOP SCAN: 
2021/01/22 17:14:00.1    CycleTime (ms)    = 200
2021/01/22 17:14:00.1    CurrentCycle      = 1500
2021/01/22 17:14:00.1    RunTime (ms)      = 299958
2021/01/22 17:14:00.1    AverageCycle (ms) = 200.0
2021/01/22 17:14:00.1    Temperature       = 25.0
2021/01/22 17:14:00.1    DATA OVERRUN %    = 0.00
2021/01/22 17:14:00.1    Spec PCB temps    = 42.3, 42.0, 42.8, 42.5, 42.2, 42.0, 40.9, 41.2
2021/01/22 17:14:00.1 
2021/01/22 17:14:00.1 Old State:16-cycle=1500, Elapsed Time:  300.0,  Ideal Time:  299.8,  Delay time =   0.04  New State:ENGN_StateRunning_SamplingComplete
2021/01/22 17:14:00.3 Old State:18-ENGN_StateRunning_SamplingComplete New State:ENGN_StateRunning_WaitNewStep
2021/01/22 17:14:00.6 Old State:5-ENGN_StateRunning_WaitNewStep New State:ENGN_StateRunning_StartNewStep
2021/01/22 17:14:00.6 Spectrometer EOS_B0712: AVRG=20.
2021/01/22 17:14:00.6 Spectrometer EOS_B0734: AVRG=20.
2021/01/22 17:14:00.6 Spectrometer EOS_B0814: AVRG=20.
2021/01/22 17:14:00.6 Spectrometer EOS_B0735: AVRG=20.
2021/01/22 17:14:00.6 Spectrometer EOS_B0732: AVRG=20.
2021/01/22 17:14:00.6 Spectrometer EOS_B0808: AVRG=20.
2021/01/22 17:14:00.6 Spectrometer EOS_B0733: AVRG=20.
2021/01/22 17:14:00.6 Spectrometer EOS_B0817: AVRG=20.
2021/01/22 17:14:00.6 StartShaker(500)
2021/01/22 17:14:00.8 Old State:6-ENGN_StateRunning_StartNewStep New State:ENGN_StateRunning_WaitGoingToSample
2021/01/22 17:14:00.8 Sample96_Q(0, 0, 9)
2021/01/22 17:14:00.8 
2021/01/22 17:14:00.8 START SCAN: 
2021/01/22 17:14:00.8    CycleTime (ms)    = 200
2021/01/22 17:14:00.8    CycleCount        = 250
2021/01/22 17:14:00.8    ScansToAverage    = 20
2021/01/22 17:14:00.8    FinishTime (s)    = 50
2021/01/22 17:14:00.8    Temperature       = 25.0
2021/01/22 17:14:00.8    Lamp Power        = 31
2021/01/22 17:14:00.8    Spec PCB temps    = 42.3, 42.0, 42.8, 42.4, 42.2, 42.0, 40.9, 41.2
2021/01/22 17:14:00.8 
2021/01/22 17:14:00.8 GMC_SetPosZ
2021/01/22 17:14:00.9 Old State:12-ENGN_StateRunning_WaitGoingToSample New State:ENGN_StateRunning_GoingToSample
2021/01/22 17:14:00.9 Temp: 2021/01/22 17:14:00, 25.0000, 25.7000
2021/01/22 17:14:04.6 GMC_ServoHere
2021/01/22 17:14:04.8 GMC_SetPosXY
2021/01/22 17:14:05.9 GMC_SetPosZ
2021/01/22 17:14:11.6 Old State:13-ENGN_StateRunning_GoingToSample New State:ENGN_StateRunning_GoingToSampleComplete
2021/01/22 17:14:11.6 Shaker current speed = 494 rpm, set speed = 500
2021/01/22 17:14:11.7 Old State:14-cycle=1, Elapsed Time:    0.0,  Ideal Time:    0.0,  Delay time =   0.00  New State:ENGN_StateRunning_WaitSampling
2021/01/22 17:14:11.7 Step started at: 17:14:00
2021/01/22 17:14:11.7 XYZ movement time  = 10683 ms + Scan start delay = 10697 ms
2021/01/22 17:14:11.8 Old State:15-ENGN_StateRunning_WaitSampling New State:ENGN_StateRunning_Sampling
2021/01/22 17:15:01.0 Temp: 2021/01/22 17:15:01, 25.0000, 25.7000
2021/01/22 17:15:01.4 
2021/01/22 17:15:01.4 STOP SCAN: 
2021/01/22 17:15:01.4    CycleTime (ms)    = 200
2021/01/22 17:15:01.4    CurrentCycle      = 250
2021/01/22 17:15:01.4    RunTime (ms)      = 49957
2021/01/22 17:15:01.4    AverageCycle (ms) = 199.8
2021/01/22 17:15:01.4    Temperature       = 25.0
2021/01/22 17:15:01.4    DATA OVERRUN %    = 0.00
2021/01/22 17:15:01.4    Spec PCB temps    = 42.3, 42.0, 42.7, 42.4, 42.2, 42.0, 40.9, 41.3
2021/01/22 17:15:01.4 
2021/01/22 17:15:01.5 Old State:16-cycle=250, Elapsed Time:   50.0,  Ideal Time:   49.8,  Delay time =   0.04  New State:ENGN_StateRunning_SamplingComplete
2021/01/22 17:15:01.6 Old State:18-ENGN_StateRunning_SamplingComplete New State:ENGN_StateRunning_WaitNewStep
2021/01/22 17:15:01.8 Old State:5-ENGN_StateRunning_WaitNewStep New State:ENGN_StateRunning_StartNewStep
2021/01/22 17:15:01.8 Spectrometer EOS_B0712: AVRG=20.
2021/01/22 17:15:01.8 Spectrometer EOS_B0734: AVRG=20.
2021/01/22 17:15:01.8 Spectrometer EOS_B0814: AVRG=20.
2021/01/22 17:15:01.8 Spectrometer EOS_B0735: AVRG=20.
2021/01/22 17:15:01.8 Spectrometer EOS_B0732: AVRG=20.
2021/01/22 17:15:01.8 Spectrometer EOS_B0808: AVRG=20.
2021/01/22 17:15:01.8 Spectrometer EOS_B0733: AVRG=20.
2021/01/22 17:15:01.8 Spectrometer EOS_B0817: AVRG=20.
2021/01/22 17:15:01.8 StartShaker(500)
2021/01/22 17:15:02.0 Old State:6-ENGN_StateRunning_StartNewStep New State:ENGN_StateRunning_WaitGoingToSample
2021/01/22 17:15:02.0 Sample96_Q(0, 0, 3)
2021/01/22 17:15:02.0 
2021/01/22 17:15:02.0 START SCAN: 
2021/01/22 17:15:02.0    CycleTime (ms)    = 200
2021/01/22 17:15:02.0    CycleCount        = 1500
2021/01/22 17:15:02.0    ScansToAverage    = 20
2021/01/22 17:15:02.0    FinishTime (s)    = 300
2021/01/22 17:15:02.0    Temperature       = 25.0
2021/01/22 17:15:02.0    Lamp Power        = 31
2021/01/22 17:15:02.0    Spec PCB temps    = 42.3, 42.0, 42.7, 42.4, 42.2, 42.0, 40.9, 41.3
2021/01/22 17:15:02.0 
2021/01/22 17:15:02.1 Old State:12-ENGN_StateRunning_WaitGoingToSample New State:ENGN_StateRunning_GoingToSample
2021/01/22 17:15:02.1 Temp: 2021/01/22 17:15:02, 25.0000, 25.7000
2021/01/22 17:15:02.1 GMC_SetPosZ
2021/01/22 17:15:07.5 GMC_ServoHere
2021/01/22 17:15:07.7 GMC_SetPosXY
2021/01/22 17:15:08.8 GMC_SetPosZ
2021/01/22 17:15:12.7 Shaker current speed = 494 rpm, set speed = 500
2021/01/22 17:15:12.8 Old State:13-ENGN_StateRunning_GoingToSample New State:ENGN_StateRunning_GoingToSampleComplete
2021/01/22 17:15:12.9 Old State:14-cycle=1, Elapsed Time:    0.0,  Ideal Time:    0.0,  Delay time =   0.00  New State:ENGN_StateRunning_WaitSampling
2021/01/22 17:15:12.9 Step started at: 17:15:02
2021/01/22 17:15:12.9 XYZ movement time  = 10786 ms + Scan start delay = 10765 ms
2021/01/22 17:15:13.0 Old State:15-ENGN_StateRunning_WaitSampling New State:ENGN_StateRunning_Sampling
2021/01/22 17:16:02.2 Temp: 2021/01/22 17:16:02, 25.0000, 25.7000
2021/01/22 17:17:02.4 Temp: 2021/01/22 17:17:02, 25.0000, 25.7000
2021/01/22 17:18:02.5 Temp: 2021/01/22 17:18:02, 25.0000, 25.7000
2021/01/22 17:19:02.7 Temp: 2021/01/22 17:19:02, 25.0000, 25.7000
2021/01/22 17:20:02.8 Temp: 2021/01/22 17:20:02, 25.0000, 25.7000
2021/01/22 17:20:12.7 
2021/01/22 17:20:12.7 STOP SCAN: 
2021/01/22 17:20:12.7    CycleTime (ms)    = 200
2021/01/22 17:20:12.7    CurrentCycle      = 1500
2021/01/22 17:20:12.7    RunTime (ms)      = 299957
2021/01/22 17:20:12.7    AverageCycle (ms) = 200.0
2021/01/22 17:20:12.7    Temperature       = 25.0
2021/01/22 17:20:12.7    DATA OVERRUN %    = 0.00
2021/01/22 17:20:12.7    Spec PCB temps    = 42.3, 42.0, 42.8, 42.5, 42.2, 42.0, 40.9, 41.3
2021/01/22 17:20:12.7 
2021/01/22 17:20:12.7 Old State:16-cycle=1500, Elapsed Time:  300.0,  Ideal Time:  299.8,  Delay time =   0.04  New State:ENGN_StateRunning_SamplingComplete
2021/01/22 17:20:12.9 Old State:18-ENGN_StateRunning_SamplingComplete New State:ENGN_StateRunning_WaitNewStep
2021/01/22 17:20:13.2 Old State:5-ENGN_StateRunning_WaitNewStep New State:ENGN_StateRunning_StartNewStep
2021/01/22 17:20:13.2 Spectrometer EOS_B0712: AVRG=20.
2021/01/22 17:20:13.2 Spectrometer EOS_B0734: AVRG=20.
2021/01/22 17:20:13.2 Spectrometer EOS_B0814: AVRG=20.
2021/01/22 17:20:13.2 Spectrometer EOS_B0735: AVRG=20.
2021/01/22 17:20:13.2 Spectrometer EOS_B0732: AVRG=20.
2021/01/22 17:20:13.2 Spectrometer EOS_B0808: AVRG=20.
2021/01/22 17:20:13.2 Spectrometer EOS_B0733: AVRG=20.
2021/01/22 17:20:13.2 Spectrometer EOS_B0817: AVRG=20.
2021/01/22 17:20:13.2 StartShaker(500)
2021/01/22 17:20:13.3 Old State:6-ENGN_StateRunning_StartNewStep New State:ENGN_StateRunning_WaitGoingToSample
2021/01/22 17:20:13.3 Sample96_Q(0, 0, 11)
2021/01/22 17:20:13.3 
2021/01/22 17:20:13.3 START SCAN: 
2021/01/22 17:20:13.3    CycleTime (ms)    = 200
2021/01/22 17:20:13.3    CycleCount        = 25
2021/01/22 17:20:13.3    ScansToAverage    = 20
2021/01/22 17:20:13.3    FinishTime (s)    = 5
2021/01/22 17:20:13.3    Temperature       = 25.0
2021/01/22 17:20:13.3    Lamp Power        = 31
2021/01/22 17:20:13.3    Spec PCB temps    = 42.3, 42.0, 42.8, 42.4, 42.2, 41.9, 40.9, 41.3
2021/01/22 17:20:13.3 
2021/01/22 17:20:13.4 GMC_SetPosZ
2021/01/22 17:20:13.4 Old State:12-ENGN_StateRunning_WaitGoingToSample New State:ENGN_StateRunning_GoingToSample
2021/01/22 17:20:13.5 Temp: 2021/01/22 17:20:13, 25.0000, 25.7000
2021/01/22 17:20:17.2 GMC_ServoHere
2021/01/22 17:20:17.4 GMC_SetPosXY
2021/01/22 17:20:18.7 GMC_SetPosZ
2021/01/22 17:20:25.0 Shaker current speed = 494 rpm, set speed = 500
2021/01/22 17:20:25.1 Old State:13-ENGN_StateRunning_GoingToSample New State:ENGN_StateRunning_GoingToSampleComplete
2021/01/22 17:20:25.2 Old State:14-cycle=1, Elapsed Time:    0.0,  Ideal Time:    0.0,  Delay time =   0.00  New State:ENGN_StateRunning_WaitSampling
2021/01/22 17:20:25.2 Step started at: 17:20:13
2021/01/22 17:20:25.2 XYZ movement time  = 11665 ms + Scan start delay = 11621 ms
2021/01/22 17:20:25.3 Old State:15-ENGN_StateRunning_WaitSampling New State:ENGN_StateRunning_Sampling
2021/01/22 17:20:29.9 
2021/01/22 17:20:29.9 STOP SCAN: 
2021/01/22 17:20:29.9    CycleTime (ms)    = 200
2021/01/22 17:20:29.9    CurrentCycle      = 25
2021/01/22 17:20:29.9    RunTime (ms)      = 4954
2021/01/22 17:20:29.9    AverageCycle (ms) = 198.2
2021/01/22 17:20:29.9    Temperature       = 25.0
2021/01/22 17:20:29.9    DATA OVERRUN %    = 0.00
2021/01/22 17:20:29.9    Spec PCB temps    = 42.3, 42.0, 42.8, 42.5, 42.3, 42.0, 40.9, 41.3
2021/01/22 17:20:29.9 
2021/01/22 17:20:29.9 Old State:16-cycle=25, Elapsed Time:    5.0,  Ideal Time:    4.8,  Delay time =   0.04  New State:ENGN_StateRunning_SamplingComplete
2021/01/22 17:20:30.1 Old State:18-ENGN_StateRunning_SamplingComplete New State:ENGN_StateRunning_WaitNewStep
2021/01/22 17:20:30.4 Old State:5-ENGN_StateRunning_WaitNewStep New State:ENGN_StateRunning_StartNewStep
2021/01/22 17:20:30.4 Spectrometer EOS_B0712: AVRG=20.
2021/01/22 17:20:30.4 Spectrometer EOS_B0734: AVRG=20.
2021/01/22 17:20:30.4 Spectrometer EOS_B0814: AVRG=20.
2021/01/22 17:20:30.4 Spectrometer EOS_B0735: AVRG=20.
2021/01/22 17:20:30.4 Spectrometer EOS_B0732: AVRG=20.
2021/01/22 17:20:30.4 Spectrometer EOS_B0808: AVRG=20.
2021/01/22 17:20:30.4 Spectrometer EOS_B0733: AVRG=20.
2021/01/22 17:20:30.4 Spectrometer EOS_B0817: AVRG=20.
2021/01/22 17:20:30.4 StartShaker(500)
2021/01/22 17:20:30.5 Old State:6-ENGN_StateRunning_StartNewStep New State:ENGN_StateRunning_WaitGoingToSample
2021/01/22 17:20:30.5 Sample96_Q(0, 0, 10)
2021/01/22 17:20:30.6 
2021/01/22 17:20:30.6 START SCAN: 
2021/01/22 17:20:30.6    CycleTime (ms)    = 200
2021/01/22 17:20:30.6    CycleCount        = 25
2021/01/22 17:20:30.6    ScansToAverage    = 20
2021/01/22 17:20:30.6    FinishTime (s)    = 5
2021/01/22 17:20:30.6    Temperature       = 25.0
2021/01/22 17:20:30.6    Lamp Power        = 31
2021/01/22 17:20:30.6    Spec PCB temps    = 42.3, 42.0, 42.8, 42.5, 42.3, 42.0, 40.9, 41.2
2021/01/22 17:20:30.6 
2021/01/22 17:20:30.7 Old State:12-ENGN_StateRunning_WaitGoingToSample New State:ENGN_StateRunning_GoingToSample
2021/01/22 17:20:30.7 GMC_SetPosZ
2021/01/22 17:20:30.7 Temp: 2021/01/22 17:20:30, 25.0000, 25.7000
2021/01/22 17:20:36.8 GMC_ServoHere
2021/01/22 17:20:36.9 GMC_SetPosXY
2021/01/22 17:20:37.3 GMC_SetPosZ
2021/01/22 17:20:43.2 Shaker current speed = 494 rpm, set speed = 500
2021/01/22 17:20:43.3 Old State:13-cycle=1, Elapsed Time:    0.0,  Ideal Time:    0.0,  Delay time =   0.00  New State:ENGN_StateRunning_GoingToSampleComplete
2021/01/22 17:20:43.4 Old State:14-ENGN_StateRunning_GoingToSampleComplete New State:ENGN_StateRunning_WaitSampling
2021/01/22 17:20:43.4 Step started at: 17:20:30
2021/01/22 17:20:43.4 XYZ movement time  = 12675 ms + Scan start delay = 12607 ms
2021/01/22 17:20:43.5 Old State:15-ENGN_StateRunning_WaitSampling New State:ENGN_StateRunning_Sampling
2021/01/22 17:20:48.1 
2021/01/22 17:20:48.1 STOP SCAN: 
2021/01/22 17:20:48.1    CycleTime (ms)    = 200
2021/01/22 17:20:48.1    CurrentCycle      = 25
2021/01/22 17:20:48.1    RunTime (ms)      = 4958
2021/01/22 17:20:48.1    AverageCycle (ms) = 198.3
2021/01/22 17:20:48.1    Temperature       = 25.0
2021/01/22 17:20:48.1    DATA OVERRUN %    = 0.00
2021/01/22 17:20:48.1    Spec PCB temps    = 42.3, 41.9, 42.8, 42.5, 42.2, 42.0, 40.9, 41.2
2021/01/22 17:20:48.1 
2021/01/22 17:20:48.2 Old State:16-cycle=25, Elapsed Time:    5.0,  Ideal Time:    4.8,  Delay time =   0.04  New State:ENGN_StateRunning_SamplingComplete
2021/01/22 17:20:48.3 Old State:18-ENGN_StateRunning_SamplingComplete New State:ENGN_StateRunning_WaitNewStep
2021/01/22 17:20:48.6 Old State:5-ENGN_StateRunning_WaitNewStep New State:ENGN_StateRunning_StartNewStep
2021/01/22 17:20:48.7 Spectrometer EOS_B0712: AVRG=20.
2021/01/22 17:20:48.7 Spectrometer EOS_B0734: AVRG=20.
2021/01/22 17:20:48.7 Spectrometer EOS_B0814: AVRG=20.
2021/01/22 17:20:48.7 Spectrometer EOS_B0735: AVRG=20.
2021/01/22 17:20:48.7 Spectrometer EOS_B0732: AVRG=20.
2021/01/22 17:20:48.7 Spectrometer EOS_B0808: AVRG=20.
2021/01/22 17:20:48.7 Spectrometer EOS_B0733: AVRG=20.
2021/01/22 17:20:48.7 Spectrometer EOS_B0817: AVRG=20.
2021/01/22 17:20:48.7 StartShaker(500)
2021/01/22 17:20:48.8 Old State:6-ENGN_StateRunning_StartNewStep New State:ENGN_StateRunning_WaitGoingToSample
2021/01/22 17:20:48.8 Sample96_Q(0, 0, 11)
2021/01/22 17:20:48.8 
2021/01/22 17:20:48.8 START SCAN: 
2021/01/22 17:20:48.8    CycleTime (ms)    = 200
2021/01/22 17:20:48.8    CycleCount        = 25
2021/01/22 17:20:48.8    ScansToAverage    = 20
2021/01/22 17:20:48.8    FinishTime (s)    = 5
2021/01/22 17:20:48.8    Temperature       = 25.0
2021/01/22 17:20:48.8    Lamp Power        = 31
2021/01/22 17:20:48.8    Spec PCB temps    = 42.3, 41.9, 42.8, 42.5, 42.2, 42.0, 40.9, 41.2
2021/01/22 17:20:48.8 
2021/01/22 17:20:48.9 Old State:12-ENGN_StateRunning_WaitGoingToSample New State:ENGN_StateRunning_GoingToSample
2021/01/22 17:20:48.9 GMC_SetPosZ
2021/01/22 17:20:48.9 Temp: 2021/01/22 17:20:48, 25.0000, 25.7000
2021/01/22 17:20:54.7 GMC_ServoHere
2021/01/22 17:20:54.8 GMC_SetPosXY
2021/01/22 17:20:55.1 GMC_SetPosZ
2021/01/22 17:21:01.4 Shaker current speed = 494 rpm, set speed = 500
2021/01/22 17:21:01.5 Old State:13-ENGN_StateRunning_GoingToSample New State:ENGN_StateRunning_GoingToSampleComplete
2021/01/22 17:21:01.6 Old State:14-cycle=1, Elapsed Time:    0.0,  Ideal Time:    0.0,  Delay time =   0.00  New State:ENGN_StateRunning_WaitSampling
2021/01/22 17:21:01.6 Step started at: 17:20:48
2021/01/22 17:21:01.6 XYZ movement time  = 12604 ms + Scan start delay = 12576 ms
2021/01/22 17:21:01.7 Old State:15-ENGN_StateRunning_WaitSampling New State:ENGN_StateRunning_Sampling
2021/01/22 17:21:06.3 
2021/01/22 17:21:06.3 STOP SCAN: 
2021/01/22 17:21:06.3    CycleTime (ms)    = 200
2021/01/22 17:21:06.3    CurrentCycle      = 25
2021/01/22 17:21:06.3    RunTime (ms)      = 4959
2021/01/22 17:21:06.3    AverageCycle (ms) = 198.4
2021/01/22 17:21:06.3    Temperature       = 25.0
2021/01/22 17:21:06.3    DATA OVERRUN %    = 0.00
2021/01/22 17:21:06.3    Spec PCB temps    = 42.3, 42.0, 42.8, 42.5, 42.2, 42.0, 40.9, 41.3
2021/01/22 17:21:06.3 
2021/01/22 17:21:06.4 Old State:16-cycle=25, Elapsed Time:    5.0,  Ideal Time:    4.8,  Delay time =   0.04  New State:ENGN_StateRunning_SamplingComplete
2021/01/22 17:21:06.6 Old State:18-ENGN_StateRunning_SamplingComplete New State:ENGN_StateRunning_WaitNewStep
2021/01/22 17:21:06.9 Old State:5-ENGN_StateRunning_WaitNewStep New State:ENGN_StateRunning_StartNewStep
2021/01/22 17:21:06.9 Spectrometer EOS_B0712: AVRG=20.
2021/01/22 17:21:06.9 Spectrometer EOS_B0734: AVRG=20.
2021/01/22 17:21:06.9 Spectrometer EOS_B0814: AVRG=20.
2021/01/22 17:21:06.9 Spectrometer EOS_B0735: AVRG=20.
2021/01/22 17:21:06.9 Spectrometer EOS_B0732: AVRG=20.
2021/01/22 17:21:06.9 Spectrometer EOS_B0808: AVRG=20.
2021/01/22 17:21:06.9 Spectrometer EOS_B0733: AVRG=20.
2021/01/22 17:21:06.9 Spectrometer EOS_B0817: AVRG=20.
2021/01/22 17:21:06.9 StartShaker(500)
2021/01/22 17:21:07.0 Old State:6-ENGN_StateRunning_StartNewStep New State:ENGN_StateRunning_WaitGoingToSample
2021/01/22 17:21:07.0 Sample96_Q(0, 0, 10)
2021/01/22 17:21:07.0 
2021/01/22 17:21:07.0 START SCAN: 
2021/01/22 17:21:07.0    CycleTime (ms)    = 200
2021/01/22 17:21:07.0    CycleCount        = 25
2021/01/22 17:21:07.0    ScansToAverage    = 20
2021/01/22 17:21:07.0    FinishTime (s)    = 5
2021/01/22 17:21:07.0    Temperature       = 25.0
2021/01/22 17:21:07.0    Lamp Power        = 31
2021/01/22 17:21:07.0    Spec PCB temps    = 42.3, 42.0, 42.8, 42.4, 42.2, 42.0, 40.9, 41.3
2021/01/22 17:21:07.0 
2021/01/22 17:21:07.1 GMC_SetPosZ
2021/01/22 17:21:07.1 Temp: 2021/01/22 17:21:07, 25.0000, 25.7000
2021/01/22 17:21:07.1 Old State:12-ENGN_StateRunning_WaitGoingToSample New State:ENGN_StateRunning_GoingToSample
2021/01/22 17:21:13.1 GMC_ServoHere
2021/01/22 17:21:13.3 GMC_SetPosXY
2021/01/22 17:21:13.6 GMC_SetPosZ
2021/01/22 17:21:19.6 Shaker current speed = 494 rpm, set speed = 500
2021/01/22 17:21:19.7 Old State:13-cycle=1, Elapsed Time:    0.0,  Ideal Time:    0.0,  Delay time =   0.00  New State:ENGN_StateRunning_GoingToSampleComplete
2021/01/22 17:21:19.8 Old State:14-ENGN_StateRunning_GoingToSampleComplete New State:ENGN_StateRunning_WaitSampling
2021/01/22 17:21:19.8 Step started at: 17:21:07
2021/01/22 17:21:19.8 XYZ movement time  = 12587 ms + Scan start delay = 12522 ms
2021/01/22 17:21:19.9 Old State:15-ENGN_StateRunning_WaitSampling New State:ENGN_StateRunning_Sampling
2021/01/22 17:21:24.5 
2021/01/22 17:21:24.5 STOP SCAN: 
2021/01/22 17:21:24.5    CycleTime (ms)    = 200
2021/01/22 17:21:24.5    CurrentCycle      = 25
2021/01/22 17:21:24.5    RunTime (ms)      = 4959
2021/01/22 17:21:24.5    AverageCycle (ms) = 198.4
2021/01/22 17:21:24.5    Temperature       = 25.0
2021/01/22 17:21:24.5    DATA OVERRUN %    = 0.00
2021/01/22 17:21:24.5    Spec PCB temps    = 42.3, 42.0, 42.8, 42.5, 42.2, 42.0, 40.9, 41.2
2021/01/22 17:21:24.5 
2021/01/22 17:21:24.5 Old State:16-cycle=25, Elapsed Time:    5.0,  Ideal Time:    4.8,  Delay time =   0.04  New State:ENGN_StateRunning_SamplingComplete
2021/01/22 17:21:24.7 Old State:18-ENGN_StateRunning_SamplingComplete New State:ENGN_StateRunning_WaitNewStep
2021/01/22 17:21:25.1 Old State:5-ENGN_StateRunning_WaitNewStep New State:ENGN_StateRunning_StartNewStep
2021/01/22 17:21:25.1 Spectrometer EOS_B0712: AVRG=20.
2021/01/22 17:21:25.1 Spectrometer EOS_B0734: AVRG=20.
2021/01/22 17:21:25.1 Spectrometer EOS_B0814: AVRG=20.
2021/01/22 17:21:25.1 Spectrometer EOS_B0735: AVRG=20.
2021/01/22 17:21:25.1 Spectrometer EOS_B0732: AVRG=20.
2021/01/22 17:21:25.1 Spectrometer EOS_B0808: AVRG=20.
2021/01/22 17:21:25.1 Spectrometer EOS_B0733: AVRG=20.
2021/01/22 17:21:25.1 Spectrometer EOS_B0817: AVRG=20.
2021/01/22 17:21:25.1 StartShaker(500)
2021/01/22 17:21:25.3 Old State:6-ENGN_StateRunning_StartNewStep New State:ENGN_StateRunning_WaitGoingToSample
2021/01/22 17:21:25.3 Sample96_Q(0, 0, 11)
2021/01/22 17:21:25.3 
2021/01/22 17:21:25.3 START SCAN: 
2021/01/22 17:21:25.3    CycleTime (ms)    = 200
2021/01/22 17:21:25.3    CycleCount        = 25
2021/01/22 17:21:25.3    ScansToAverage    = 20
2021/01/22 17:21:25.3    FinishTime (s)    = 5
2021/01/22 17:21:25.3    Temperature       = 25.0
2021/01/22 17:21:25.3    Lamp Power        = 31
2021/01/22 17:21:25.3    Spec PCB temps    = 42.3, 42.0, 42.8, 42.4, 42.2, 42.0, 40.9, 41.2
2021/01/22 17:21:25.3 
2021/01/22 17:21:25.3 GMC_SetPosZ
2021/01/22 17:21:25.3 Temp: 2021/01/22 17:21:25, 25.0000, 25.7000
2021/01/22 17:21:25.4 Old State:12-ENGN_StateRunning_WaitGoingToSample New State:ENGN_StateRunning_GoingToSample
2021/01/22 17:21:31.1 GMC_ServoHere
2021/01/22 17:21:31.2 GMC_SetPosXY
2021/01/22 17:21:31.5 GMC_SetPosZ
2021/01/22 17:21:37.8 Old State:13-ENGN_StateRunning_GoingToSample New State:ENGN_StateRunning_GoingToSampleComplete
2021/01/22 17:21:37.8 Shaker current speed = 494 rpm, set speed = 500
2021/01/22 17:21:37.9 Old State:14-cycle=1, Elapsed Time:    0.0,  Ideal Time:    0.0,  Delay time =   0.00  New State:ENGN_StateRunning_WaitSampling
2021/01/22 17:21:37.9 Step started at: 17:21:25
2021/01/22 17:21:37.9 XYZ movement time  = 12473 ms + Scan start delay = 12480 ms
2021/01/22 17:21:38.0 Old State:15-ENGN_StateRunning_WaitSampling New State:ENGN_StateRunning_Sampling
2021/01/22 17:21:42.7 
2021/01/22 17:21:42.7 STOP SCAN: 
2021/01/22 17:21:42.7    CycleTime (ms)    = 200
2021/01/22 17:21:42.7    CurrentCycle      = 25
2021/01/22 17:21:42.7    RunTime (ms)      = 4956
2021/01/22 17:21:42.7    AverageCycle (ms) = 198.2
2021/01/22 17:21:42.7    Temperature       = 25.0
2021/01/22 17:21:42.7    DATA OVERRUN %    = 0.00
2021/01/22 17:21:42.7    Spec PCB temps    = 42.3, 41.9, 42.8, 42.4, 42.3, 42.0, 40.9, 41.3
2021/01/22 17:21:42.7 
2021/01/22 17:21:42.8 Old State:16-cycle=25, Elapsed Time:    5.0,  Ideal Time:    4.8,  Delay time =   0.04  New State:ENGN_StateRunning_SamplingComplete
2021/01/22 17:21:42.9 Old State:18-ENGN_StateRunning_SamplingComplete New State:ENGN_StateRunning_WaitNewStep
2021/01/22 17:21:43.2 Old State:5-ENGN_StateRunning_WaitNewStep New State:ENGN_StateRunning_StartNewStep
2021/01/22 17:21:43.3 Spectrometer EOS_B0712: AVRG=20.
2021/01/22 17:21:43.3 Spectrometer EOS_B0734: AVRG=20.
2021/01/22 17:21:43.3 Spectrometer EOS_B0814: AVRG=20.
2021/01/22 17:21:43.3 Spectrometer EOS_B0735: AVRG=20.
2021/01/22 17:21:43.3 Spectrometer EOS_B0732: AVRG=20.
2021/01/22 17:21:43.3 Spectrometer EOS_B0808: AVRG=20.
2021/01/22 17:21:43.3 Spectrometer EOS_B0733: AVRG=20.
2021/01/22 17:21:43.3 Spectrometer EOS_B0817: AVRG=20.
2021/01/22 17:21:43.3 StartShaker(500)
2021/01/22 17:21:43.4 Old State:6-ENGN_StateRunning_StartNewStep New State:ENGN_StateRunning_WaitGoingToSample
2021/01/22 17:21:43.4 Sample96_Q(0, 0, 10)
2021/01/22 17:21:43.4 
2021/01/22 17:21:43.4 START SCAN: 
2021/01/22 17:21:43.4    CycleTime (ms)    = 200
2021/01/22 17:21:43.4    CycleCount        = 25
2021/01/22 17:21:43.4    ScansToAverage    = 20
2021/01/22 17:21:43.4    FinishTime (s)    = 5
2021/01/22 17:21:43.4    Temperature       = 25.0
2021/01/22 17:21:43.4    Lamp Power        = 31
2021/01/22 17:21:43.4    Spec PCB temps    = 42.3, 41.9, 42.8, 42.4, 42.3, 41.9, 40.9, 41.3
2021/01/22 17:21:43.4 
2021/01/22 17:21:43.5 GMC_SetPosZ
2021/01/22 17:21:43.5 Old State:12-ENGN_StateRunning_WaitGoingToSample New State:ENGN_StateRunning_GoingToSample
2021/01/22 17:21:43.6 Temp: 2021/01/22 17:21:43, 25.0000, 25.7000
2021/01/22 17:21:49.5 GMC_ServoHere
2021/01/22 17:21:49.7 GMC_SetPosXY
2021/01/22 17:21:50.0 GMC_SetPosZ
2021/01/22 17:21:55.9 Shaker current speed = 494 rpm, set speed = 500
2021/01/22 17:21:56.0 Old State:13-cycle=1, Elapsed Time:    0.0,  Ideal Time:    0.0,  Delay time =   0.00  New State:ENGN_StateRunning_GoingToSampleComplete
2021/01/22 17:21:56.1 Old State:14-ENGN_StateRunning_GoingToSampleComplete New State:ENGN_StateRunning_WaitSampling
2021/01/22 17:21:56.1 Step started at: 17:21:43
2021/01/22 17:21:56.1 XYZ movement time  = 12505 ms + Scan start delay = 12415 ms
2021/01/22 17:21:56.2 Old State:15-ENGN_StateRunning_WaitSampling New State:ENGN_StateRunning_Sampling
2021/01/22 17:22:00.8 
2021/01/22 17:22:00.8 STOP SCAN: 
2021/01/22 17:22:00.8    CycleTime (ms)    = 200
2021/01/22 17:22:00.8    CurrentCycle      = 25
2021/01/22 17:22:00.8    RunTime (ms)      = 4959
2021/01/22 17:22:00.8    AverageCycle (ms) = 198.4
2021/01/22 17:22:00.8    Temperature       = 25.0
2021/01/22 17:22:00.8    DATA OVERRUN %    = 0.00
2021/01/22 17:22:00.8    Spec PCB temps    = 42.3, 41.9, 42.8, 42.4, 42.2, 42.0, 40.9, 41.2
2021/01/22 17:22:00.8 
2021/01/22 17:22:00.8 Old State:16-cycle=25, Elapsed Time:    5.0,  Ideal Time:    4.8,  Delay time =   0.04  New State:ENGN_StateRunning_SamplingComplete
2021/01/22 17:22:01.0 Old State:18-ENGN_StateRunning_SamplingComplete New State:ENGN_StateRunning_WaitEjecting
2021/01/22 17:22:01.0 EJECT: To Tray, position=0,0
2021/01/22 17:22:01.0    EjectOffset = 10.0
2021/01/22 17:22:01.1 Old State:19-ENGN_StateRunning_WaitEjecting New State:ENGN_StateRunning_Ejecting
2021/01/22 17:22:01.1 GMC_SetPosZ
2021/01/22 17:22:06.9 GMC_ServoHere
2021/01/22 17:22:07.1 GMC_SetPosXY
2021/01/22 17:22:09.3 GMC_SetPosZ
2021/01/22 17:22:13.4 GMC_Eject
2021/01/22 17:22:13.4 EjectDone => EjectEject
2021/01/22 17:22:13.4 SetEjectorPosn(0)
2021/01/22 17:22:16.6 GMC_HomeZ
2021/01/22 17:22:16.6 HomeDevice[0005]
2021/01/22 17:22:22.1 GMC_SetPosZ
2021/01/22 17:22:22.5 GMC_ServoHere
2021/01/22 17:22:22.7 GMC_SetPosZ
2021/01/22 17:22:22.9 GMC_ServoHere
2021/01/22 17:22:23.0 GMC_SetPosXY
2021/01/22 17:22:25.3 GMC_SetPosZ
2021/01/22 17:22:25.6 Old State:20-ENGN_StateRunning_Ejecting New State:ENGN_StateRunning_EjectingComplete
2021/01/22 17:22:25.7 Old State:21-ENGN_StateRunning_EjectingComplete New State:ENGN_StateRunning_Finished
2021/01/22 17:22:25.7 StopShaker
2021/01/22 17:22:25.8 Old State:22-ENGN_StateRunning_Finished New State:ENGN_State_SystemReady
2021/01/22 17:22:25.9 Temp: 2021/01/22 17:22:25, 25.0000, 25.7000
2021/01/22 17:22:25.9 Temperature monitor DISABLE
2021/01/22 17:22:25.9 
2021/01/22 17:22:25.9 ASSAY SUMMARY:
2021/01/22 17:22:25.9 ==============
2021/01/22 17:22:25.9   Step #   Duration   Cycles  Overrun %
2021/01/22 17:22:25.9        1      59     300    0.00
2021/01/22 17:22:25.9        2     259    1300    0.00
2021/01/22 17:22:25.9        3      29     150    0.00
2021/01/22 17:22:25.9        4     299    1500    0.00
2021/01/22 17:22:25.9        5       4      25    0.00
2021/01/22 17:22:25.9        6       4      25    0.00
2021/01/22 17:22:25.9        7       4      25    0.00
2021/01/22 17:22:25.9        8       4      25    0.00
2021/01/22 17:22:25.9        9       4      25    0.00
2021/01/22 17:22:25.9       10       4      25    0.00
2021/01/22 17:22:25.9       11     299    1500    0.00
2021/01/22 17:22:25.9       12      49     250    0.00
2021/01/22 17:22:25.9       13     299    1500    0.00
2021/01/22 17:22:25.9       14       4      25    0.00
2021/01/22 17:22:25.9       15       4      25    0.00
2021/01/22 17:22:25.9       16       4      25    0.00
2021/01/22 17:22:25.9       17       4      25    0.00
2021/01/22 17:22:25.9       18       4      25    0.00
2021/01/22 17:22:25.9       19       4      25    0.00
2021/01/22 17:22:25.9       20     299    1500    0.00
2021/01/22 17:22:25.9       21      49     250    0.00
2021/01/22 17:22:25.9       22     299    1500    0.00
2021/01/22 17:22:25.9       23       4      25    0.00
2021/01/22 17:22:25.9       24       4      25    0.00
2021/01/22 17:22:25.9       25       4      25    0.00
2021/01/22 17:22:25.9       26       4      25    0.00
2021/01/22 17:22:25.9       27       4      25    0.00
2021/01/22 17:22:25.9       28       4      25    0.00
2021/01/22 17:22:25.9       29     299    1500    0.00
2021/01/22 17:22:25.9       30      49     250    0.00
2021/01/22 17:22:25.9       31     299    1500    0.00
2021/01/22 17:22:25.9       32       4      25    0.00
2021/01/22 17:22:25.9       33       4      25    0.00
2021/01/22 17:22:25.9       34       4      25    0.00
2021/01/22 17:22:25.9       35       4      25    0.00
2021/01/22 17:22:25.9       36       4      25    0.00
2021/01/22 17:22:25.9       37       4      25    0.00
2021/01/22 17:22:25.9       38     299    1500    0.00
2021/01/22 17:22:25.9       39      49     250    0.00
2021/01/22 17:22:25.9       40     299    1500    0.00
2021/01/22 17:22:25.9       41       4      25    0.00
2021/01/22 17:22:25.9       42       4      25    0.00
2021/01/22 17:22:25.9       43       4      25    0.00
2021/01/22 17:22:25.9       44       4      25    0.00
2021/01/22 17:22:25.9       45       4      25    0.00
2021/01/22 17:22:25.9       46       4      25    0.00
2021/01/22 17:22:25.9       47     299    1500    0.00
2021/01/22 17:22:25.9       48      49     250    0.00
2021/01/22 17:22:25.9       49     299    1500    0.00
2021/01/22 17:22:25.9       50       4      25    0.00
2021/01/22 17:22:25.9       51       4      25    0.00
2021/01/22 17:22:25.9       52       4      25    0.00
2021/01/22 17:22:25.9       53       4      25    0.00
2021/01/22 17:22:25.9       54       4      25    0.00
2021/01/22 17:22:25.9       55       4      25    0.00
2021/01/22 17:22:25.9       56     299    1500    0.00
2021/01/22 17:22:25.9       57      49     250    0.00
2021/01/22 17:22:25.9       58     299    1500    0.00
2021/01/22 17:22:25.9       59       4      25    0.00
2021/01/22 17:22:25.9       60       4      25    0.00
2021/01/22 17:22:25.9       61       4      25    0.00
2021/01/22 17:22:25.9       62       4      25    0.00
2021/01/22 17:22:25.9       63       4      25    0.00
2021/01/22 17:22:25.9       64       4      25    0.00
2021/01/22 17:22:25.9 -------------------------------------------
2021/01/22 17:22:25.9    Total    4757   23800    0.00
2021/01/22 17:22:25.9 Temperature monitor DISABLE
2021/01/22 17:22:26.1 DFIO: Start writing C:\Users\Labuser\Documents\Octet Data\Mesfin\SARS_CoV2\VHH72_trimer\Experiment_1\210122_001.frd
2021/01/22 17:22:26.2 DFIO: Finished writing C:\Users\Labuser\Documents\Octet Data\Mesfin\SARS_CoV2\VHH72_trimer\Experiment_1\210122_001.frd
2021/01/22 17:22:26.2 DFIO: Start writing C:\Users\Labuser\Documents\Octet Data\Mesfin\SARS_CoV2\VHH72_trimer\Experiment_1\210122_002.frd
2021/01/22 17:22:26.2 DFIO: Finished writing C:\Users\Labuser\Documents\Octet Data\Mesfin\SARS_CoV2\VHH72_trimer\Experiment_1\210122_002.frd
2021/01/22 17:22:26.2 DFIO: Start writing C:\Users\Labuser\Documents\Octet Data\Mesfin\SARS_CoV2\VHH72_trimer\Experiment_1\210122_003.frd
2021/01/22 17:22:26.2 DFIO: Finished writing C:\Users\Labuser\Documents\Octet Data\Mesfin\SARS_CoV2\VHH72_trimer\Experiment_1\210122_003.frd
2021/01/22 17:22:26.2 DFIO: Start writing C:\Users\Labuser\Documents\Octet Data\Mesfin\SARS_CoV2\VHH72_trimer\Experiment_1\210122_004.frd
